text:
|Publication number
|Publication type
|Application number
|Publication date
|14 Aug 2001
|Filing date
|26 Jun 2000
|Priority date
|10 Feb 1998
|Fee status
|Also published as
|Publication number
|Timothy H. Fox, Ian R. Crocker
|Original Assignee
|Emory University
|Export Citation
|BiBTeX, EndNote, RefMan
|Patent Citations (42), Non-Patent Citations (14), Referenced by (64), Classifications (14), Legal Events (4)
|External Links: USPTO, USPTO Assignment, Espacenet
This application is a divisional of U.S. Ser. No. 09/021,198 entitled "Systems and Methods for Providing Radiation Therapy and Catheter Guides" filed on Feb. 10, 1998 by Timothy H. Fox and Ian R. Crocker, now U.S. Pat. No. 6,083,167.
The present invention relates generally to systems and methods devices for providing radiation therapy and, more particularly, to systems and methods for providing radiation therapy for the prevention of restenosis. According to another aspect, the invention generally relates to catheter guides for enabling clinicians to determine location or orientation of catheters.
A leading cause of death in the western world is atherosclerosis. More than thirteen million people in the United States have been diagnosed with this disease with a large number of patients having arteries or veins that have become narrowed and need to be enlarged. Angioplasty, a common technique to enlarge an artery or vein, is an interventional radiologic technique in which a narrowed artery or vein is enlarged with the use of a balloon angioplasty catheter.
To perform an angioplasty, an angiogram is used to obtain a precise image of the narrowed artery or vein. A catheter is inserted into a blood vessel and is guided to the site of the narrowing with an X-ray monitor. A contrast medium is injected through the catheter and a series of X-ray images are obtained to outline the blood vessel. The images obtained from the X-ray are used to identify and measure the abnormal narrowing of the blood vessel. The initial catheter used for diagnostic purposes is then removed and a catheter having an inflatable balloon around its shaft is inserted. The balloon is inflated in the narrowed portion of the vessel to widen the artery or vein. In some cases, a metallic stent is placed within the blood vessel in order to widen the blood vessel.
Although the initial success rate is high, the long-term success rate of angioplasty is unfortunately rather low and a subsequent procedure is often necessary. It has been estimated that nearly 25-45% of the 450,000 coronary angioplasties done in the United States each year fail within the first few months of the operation. Restenosis is defined to be the reclosing of the artery or vein to less than 50% of its original size. Because of restenosis, an additional angioplasty or another procedure, such as bypass surgery, is necessary to reopen the blood vessel. The need for subsequent procedures has a traumatic effect on the patient and on those concerned about the patient's health and additionally greatly increases medical costs. In fact, the cost to the health care system in the United States for subsequent procedures has been estimated to be 2 billion dollars annually.
The problems associated with restenosis are not limited only with an angioplasty procedure but also occur with other coronary revascularization procedures. Restenosis, for instance, is also associated with rotoblator, atherectomy catheters, hot and cold laser catheters, transluminal extraction catheters, and ultrasonic ablation. The magnitude of the problem caused by restenosis is therefore quite extensive.
Restenosis is difficult to prevent since its cause is not completely understood. Restenosis, however, is believed to be primarily caused by intimal hyperplasia and negative remodeling, and, to a lesser extent, elastic recoil. Angioplasty, as well as other revascularization techniques, expose the medial smooth muscle of the blood vessel to circulating mitogenic factors when the artery is dilated. As a result of platelet degranulation, growth factors are released which induce mitogenesis and promote proliferation of cells. Vascular smooth muscle cells consequently migrate into the intima, where they synthesize collagen and elastin, which comprise the bulk of the restenotic lesion.
One approach to combating restenosis is through drug therapy. Some agents tested in restenosis trials include antiplatelet agents, anticoagulants, thromboxane antagonists, prostanoids, calcium channel blockers, ACE inhibitors, antiproliferative growth factor inhibitors, lipid lowering agents, vitamins and antioxidants, and corticosteroids and non-steriodial A. Another agent that has been used to prevent restenosis is hyaluaronic acid. The use of hyaluaronic acid is shown and described in U.S. Pat. No. 5,614,506 to Falk et al., the disclosure of which is incorporated herein. The use of agents to block restenosis has generally been met with failure or at most limited success. The drug trials involving these agents have been fraught with problems such as incomplete angiographic follow-up, variable definitions of restenosis, small sample sizes, and varying drug dosage.
Another approach used to combat is the use of a stent. A polymer stent, for instance, is placed within the treated blood vessel and delivers medicine to the damaged blood vessel. The polymer stent need not be removed since it naturally dissolves after the blood vessel has been repaired. Stents have produced promising results in that they appear to significantly reduce the rate of restenosis by approximately 10%. Studies of stent usage, however, have revealed undesirable side effects, such as an increased incidence of bleeding complications associated with stent implantation. Stents unfortunately are unable to completely eliminate restenosis and a need exists for a treatment method that even further reduces the rate of restenosis.
An emerging and promising treatment for coronary restenosis is intracoronary radiation therapy (ICRT). In general, intracoronary radiation therapy delivers radiation to a damaged area of the blood vessel to prevent restenosis. A delivery catheter allows a radiation source to be delivered to the angioplasty site where it remains for a number of minutes before being withdrawn. The radiation source may be a line or "train" of several miniature cylindrical sealed sources containing a radioactive material, such as Sr-90, Ir-192, I-125, or Re-186. Rather than a train of radiation sources, other irradiation delivery techniques include radioactive seeds or pellets, radioactive wires, intravascular x-ray sources, and liquid-filled balloons emitting particulate or electromagnetic radiation. A radiation source never comes in contact with the patient's tissue or blood and a transfer device shields the radiation from health care workers during its handling. One advantage of intracoronary radiation therapy is that it typically adds less than ten minutes to the total procedure time and is easily incorporated within the cath lab. An example of an intracoronary radiation therapy is shown and described in U.S. Pat. No. 5,683,345 to Waksman et al., the disclosure of which is incorporated herein.
Trials of intracoronary radiation therapy have produced promising results. In one trial, for instance, one-half of a group of patients received gamma radiation and the other half received placebo treatment while all of the patients in this trial had a coronary stent. The preliminary results of the study showed that the treated group had a restenosis rate of 17%, compared with a restenosis rate of 54% in the non-treated group. Intracoronary radiation therapy has also shown to be preferable over the use of a stent. For instance, coronary stents typically produce a late loss index of 25 to 30%, meaning that on average 70 to 75% of the initial improvement in lumen diameter achieved by angioplasty was still present six months later. In contrast, a study involving intracoronary radiation therapy reduced the late loss index to only 5%, meaning that on average 95% of the initial improvement in lumen diameter was still present six months later. When only those patients receiving a higher dosage of radiation were evaluated, the late loss index dropped to zero. This study therefore suggests that with proper dosing level, additional devices or therapies should not be necessary.
The success of the intracoronary radiation therapy trials also point to shortcomings of the therapy. As discussed above, the success of intracoronary radiation therapy depends upon the proper dosage level with dosimetry measured in the sub-millimeter range being essential to optimally prescribing and delivering the radiation to the lesion. Proper dosing is difficult to achieve since it is dependent, among other things, upon the distance between the source of radiation and the lesion and upon the exposure time. Presently, X-ray imaging is used to determine the size of the vessel. The calculation or measurement of the dose-rate at a certain distance from the source is next made to determine the dwell-time required to deliver a certain dose given the vessel size.
Achieving the proper dosing is difficult with conventional methods. Since the imaging techniques used to estimate distances produce two-dimensional images, an accurate distance between the radiation source and the lesion cannot easily be obtained due to possible undetectable differences in location along the third dimension. As a result of errors in estimating the distance, the lesion may not be exposed to the radiation for the correct period of time. The proper dose, moreover, depends upon the lumen size and the wall thickness of the blood vessel, both of which cannot be accurately measured through conventional imaging techniques. Because of inaccuracies in measuring the distance between the source and lesion, the lumen size, and the wall thickness, optimal results with intracoronary radiation therapy are difficult to achieve. Intracoronary radiation therapy therefore cannot be successfully performed in a consistent manner.
Furthermore, conventional radiation treatment planning is less than optimal for this irradiation. A typical radiation treatment plan involves data entry and digitization of anatomical regions and images. The time needed to determine the distance and the dwell-time would delay the immediate treatment of a patient and may be detrimental to the patient's health. A conventional radiation treatment planning system would prevent medical personnel from adjusting or altering their treatment plan before treatment which may be necessary due to unforeseen circumstances discovered during the revascularization technique, such as the lumen size, wall thickness, and other patient specific information. A need therefore exists for a radiation treatment planning system which addresses the shortcomings of conventional treatment planning.
Another difficulty encountered with conventional treatment methods is identifying the position of a catheter within the blood vessel. This difficulty relates both to the position of the catheter along the length of a blood vessel and also pertains to the rotational position of the catheter. In other words, with conventional techniques, it is difficult and at times impossible to know the precise position and orientation of a catheter within the patient's body. This problem of locating a catheter is problematic not only in the treatment of restenosis but is encountered in virtually all uses of a catheter. Furthermore, this problem occurs when assessing the magnitude of a patient's illness, when arriving at a treatment plan, during the treatment of the patient, and also during the verification of a particular treatment.
To illustrate the problem in locating a catheter with conventional techniques, consider a situation in which a patient has a lesion along a particular length of an artery and intracoronary radiation is to be used to treat the lesion. The position of the catheter is first approximately placed at a desired location within the patient and this desired location may be verified with fluoroscopy. U.S. Pat. No. 5,054,492 to Scribner et al., the disclosure of which is incorporated herein by reference, provides one example of how a marker may be placed on a catheter guide and viewed through fluoroscopy in order to verity the location of catheter. The catheter is placed at the approximate position of the distal end of the lesion through the use of an automatic pull-back mechanism. The precise location of the catheter cannot be obtained but the approximate location of the catheter is estimated by calculating the speed of the catheter with the time that it has been moved in order to arrive at the traveled distance. With the catheter at the approximate location of a distal end of the lesion, the lesion is exposed to the radiation source and the dwell-time and dwell-positions of the radiation source or sources is again controlled with the automatic pull-back mechanism by estimating distance from the catheter speed and time of travel. Because of the inherent inaccuracies in indirectly estimating the position of a catheter through the parameters of speed and time, a need exists for systems and methods that allow the position of a catheter to determined more accurately.
Another limitation in catheter technology is that the rotational orientation of the catheter is typically not monitored. As a result, even though a lesion may only be located along one segment of an artery and not along an entire circumference of the lumen wall, the radiation is delivered in all directions in order to ensure that the lesion receives the radiation. Because conventional techniques do not allow the radiation to be directed only to the lesion, the radiation is often needlessly exposed to healthy portions of the artery. A need therefore exists for systems and methods that allow radiation to be focused to only certain segments of the arterial wall.
Even if radiation should be applied to the entire surface of an artery, it is difficult if not impossible to deliver a uniform dose of radiation. In delivering the radiation, assumptions are made that the artery is linear and that the radiation is distributed in a uniform manner. These assumptions, however, are often incorrect due to various factors, such as a curve in the artery which may result in a higher concentration of radiation being delivered at surfaces of the artery closest to the turn and lower concentrations being delivered at arterial surfaces farther away from the turn. Because of difficulty in controlling the rotational orientation of a catheter, conventional techniques are unable to optimally deliver a dose to a lesion.
One obstacle to the optimal delivery of a dose is the inability to detect the rotational position of the catheter. With many catheters, it is impossible to ascertain the rotational position of the catheter within the patient. U.S. Pat. No. 5,054,492 to Scribner et al. and U.S. Pat. No. 5,596,990 to Yock et al., the disclosures of which are both incorporated herein by reference, are directed to this problem but offer incomplete solutions. Scribner and Yock disclose catheter guides having a marker that serves as a landmark. With Scribner, an ultrasonic catheter is equipped with an ultrasonically opaque marker which appears as an artifact on resultant images taken with the transducer. The catheter also has a fluoroscopic marker that is used to identify the rotational position of the catheter from which the orientation of the ultrasonic marker may be deduced. Yock relates to a catheter having an ultrasonic catheter having a marker that is placed at a predetermined orientation and an image of the marker is produced.
Although Scribner and Yock offer improvements in that they provide some reference to estimate orientation, these references provide an initial orientation but do not allow the precise orientation of the catheter to be determined at all locations along a treatment volume. Since the catheter may twist or turn as it is being moved along the treatment volume, the initial indication of orientation provided by the marker may lead clinicians to an incorrect assumption about the orientation of the catheter. Scribner and Yock, moreover, do not assist clinicians in their estimate of the catheter's position along the length of a treatment volume. A need therefore still exists for devices and methods that enable the precise determination of a catheter's location and orientation.
The present invention addresses the problems described above by providing systems and methods that may substantially reduce the rate of restenosis. An imaging system acquires in-vivo images within a blood vessel or other area to be treated with radiation at each position along the length of the treatment volume and these images are passed along to a treatment planner. The imaging system includes an automatic pull back mechanism for moving an ultrasonic transducer along the length of a treatment volume. Alternatively, the system may incorporate stereotactic localization in order to track the position and orientation of the transducer. The imaging system acquires images at the same position of the radiation source whereby accurate distances from the radiation source to a lesion or other tissue to be treated may be obtained.
The treatment planner receives the images from the imaging system, either through a frame grabber or more preferably through a network interface that delivers the digital image data. The treatment planner assembles the plurality of transverse two-dimensional images and reconstructs a three-dimensional image of the treatment volume with each pixel having a unique x and y set of coordinates and each image being assigned a unique z coordinate. The treatment planner also has a dose calculator for determining a dose from a dose prescription and an optimizer for deriving an optimal dose for a particular treatment volume. A dose and image registration unit within the treatment planner overlays the dose upon the treatment volume and displays this information to medical personnel on a display, video recorder, or on some other output device. The treatment planner outputs the treatment plan and provides this plan to medical personnel, such as through a printer or monitor.
With a method according to the invention, an ultrasonic transducer is inserted within a treatment volume and is pulled back until it reaches a distal end of a treatment volume. An automatic pull back mechanism within the imaging system is then initiated and images are acquired at each position along the length of the blood vessel or other tissue to be treated. With each image acquired, the treatment planner assigns each pixel an x and y coordinate and also assigns each image plane a z coordinate. If optimization is enabled, the dose is set based on segmented anatomy and the dwell-time and positions of the radiation source are optimized. If optimization is not enabled, then the prescription point and source position are set. The dose is calculated and the dose is registered with the treatment volume images. After the dose has been registered with the images, both quantitative and qualitative plan evaluations are performed. If the plan is unacceptable, then the treatment planner performs another iteration of optimization or alters the values for the prescription point and source position.
A catheter guide, according to another aspect of the invention, allows for the precise tracking of a catheter's location and orientation. The catheter guide includes a first set of markers that produce image artifacts allowing clinicians to accurately arrive at a reference axis and a second set of markers allowing clinicians to accurately determine the position of a catheter along a length of a treatment volume. In one example of the invention, the first set of marker comprise straight lines formed parallel to each other along the length of the catheter guide and the second set of markers comprise straight lines that intersect the first set of markers. With the reference axis provided by the first set of markers, the orientation of the catheter and thus the orientation of resultant images can be accurately determined. Through knowledge of the orientation and position of the catheter, radiation can be focused or directed to only desired surfaces of an arterial wall along select portions of the artery. Other advantages and uses for the catheter guide will become apparent from the detailed description below.
Accordingly, it is an object of the present invention to provide systems and methods that substantially reduces the rate of restenosis.
It is a further object of the present invention to provide systems and methods that may be used to accurately and consistently deliver radiation.
It is yet another object of the present invention to provide systems and methods for allowing treatment plans to be iteratively changed, such as in response to unforeseen patient-specific information.
It is yet a further object of the present invention to provide systems and methods for the prevention of restenosis that reduce delays in the determination of a treatment plan.
It is a further object of this invention to allow determination of the rotational and longitudinal correlation of a catheter.
Other objects, features, and advantages of the present invention will become apparent with respect to the remainder of this document.
The accompanying drawings, which are incorporated in and form a part of the specification, illustrate preferred embodiments of the present invention and, together with the description, disclose the principles of the invention. In the drawings:
FIG. 1 is a block diagram of a system according to a preferred embodiment of the invention;
FIGS. 2A and 2B are flow charts of a method for determining the optimal dosimetry according to a preferred embodiment of the invention;
FIGS. 3A to 3C are flow charts of a method for obtaining a treatment plan according to a preferred embodiment of the invention;
FIG. 4 is an exemplary operating room layout incorporating the system of FIG. 1;
FIG. 5 is a flow chart illustrating a method of operation for a voice recognition unit of FIG. 1;
FIG. 6A is a perspective view of a guide catheter and FIG. 6B is an exemplary ultrasound image;
FIGS. 7A and 7B are graphs of coordinate systems for radiation source and for an image acquired of a treatment volume, respectively;
FIG. 8A is a top plan view of a catheter guide according to a preferred embodiment of the invention; and
FIGS. 8B and 8C are ultrasound images acquired at two different positions along a length of the catheter guide.
Reference will now be made in detail to preferred embodiments of the invention, non-limiting examples of which are illustrated in the accompanying drawings. With reference to FIG. 1, a radiation treatment planning and verification system 10 according to a preferred embodiment of the invention includes a treatment planner 20 and an imaging system 12. The imaging system 12 is an in-vivo imaging system in that it generates images of a treatment volume from within the blood vessel or other area to be treated. The imaging system 12 is preferably an ultrasonic system having a transducer 14 and an ultrasound device 16 for controlling operations of the transducer 14. The preferred imaging system 12 is commonly referred to as an intravascular ultrasound (IVUS) system. One exemplary IVUS system that may be used is the ORACLE® InVision™ intravascular ultrasound imaging system manufactured by Endosonics Corporation of Rancho Cordovo, Calif., although other IVUS systems may be used.
An ultrasonic imaging catheter comprises a catheter body having a distal end and a proximal end. The ultrasonic imaging transducer 14 is located within the distal end and is arranged to produce an image in an image plane which is generally normal to the axial direction of the catheter. An imaging coordinate system is established to define the image planes relative to the position of the imaging transducer 14 within the catheter. The x-y plane coordinates are associated with the axial direction of the catheter on the ultrasound image plane and the z coordinate is used for the value of the image plane, which is normal to the axial direction of the catheter.
The IVUS 12 is capable of imaging the vessel lumen, the thickness of the arterial wall, breaks or dissections in the arterial wall following intervention, and distribution of plaque and calcium deposits in the vessel wall. The ultrasound device 16 of the system 12 includes, in one embodiment, an automated motorized pull-back mechanism for moving the transducer 14 along the length of the treatment volume The IVUS system 12 produces transverse images of sections of the arterial lumen and wall as the transducer 14 is moved along the length of the treatment volume.
The treatment planner 20 acquires the images from the IVUS system 12 and combines the transverse images to generate a three-dimensional image of the treatment volume. The treatment planner 20 may include a video frame grabber 22 for acquiring each transverse image and for supplying each frame to an image processor 26. Instead of or in addition to the video frame grabber 22, the treatment planner 20 preferably includes a network interface 24 for transferring the digital image data produced by the IVUS system 12 directly to the image processor 26. With the network interface 24, the image processor 26 has available the precise digital data from which it can reconstruct the images of the treatment volume and can also perform other processing of the digital image data The image processor 26, as alluded to above, acquires data from the IVUS system 12 and produces three-dimensional images of the treatment volume, which are supplied to a dose and image registration unit 34.
In order to determine the proper dose, a dose prescription 28 must first be supplied to the treatment planner 20. The dose prescription 28 indicates the type and magnitude of radiation. The dose prescription 28, for instance, may specify radioactive seeds or pellets, a radioactive wire, an intravascular x-ray source, a liquid-filled balloon emitting radiation, or other techniques for delivering radiation to the treatment volume. The dose prescription 28, moreover, may specify different levels of radiation at different locations in the image data set and may be supplied to the treatment planner 20 in any suitable manner, such as through manual keying of the data with a keyboard 42, through the use of a mouse 44, or through use of a microphone 46, speakers 48, and voice recognition unit 36.
The dose prescription 28 is received by a dose calculator 30 within the treatment planner 20. In general, the dose calculator 30 determines the dose from the prescribed source of radiation at each point within the treatment volume. The dose calculator 30 provides the calculated dose information to an optimizer 32 which, in general, provides the optimal dose for a particular treatment volume.
The dose calculator 30 may use various methods for determining the absorbed radiation dose to human tissue. These methods include but are not limited to Monte Carlo, semi-empirical methods, and experimentally measured data from a radiation detector with adequate spatial resolution.
The dose and image registration unit 34 receives the dose information from the optimizer 32 and the image information from the image processor 26 and registers the dose with the images. The dose and image registration unit 34 indicates the dose deposited throughout the target volume and displays this information on a monitor 40. The dose and image registered information from the unit 34 may also be supplied to a video recorder 38 and to other units not shown.
The dose and image registration unit 34 visualizes the dose deposited throughout the treatment volume by using both qualitative and quantitative dose evaluation methods. A qualitative method allows the clinician to subjectively judge a two-dimensional (2-D) or three-dimensional (3-D) picture of isodose surfaces superimposed with imaging data. Joining points of common dose in a calculated dose distribution creates isodose distributions. The visualization of dose with respect to the treatment volume can be shown not only on the axial images, but also on reformatted saggital and coronal images. Surface or volume rendering techniques can be used for displaying the 3-D views of the isodose distribution with the treatment volume or imaging data. In addition to qualitative tools for judging the merit of a computed dose distribution, quantitative tools such as dose volume histograms (DVH) or dose surface histograms (DSH) and minimum/maximum dose for structure contours are methods used to further evaluate these dose distributions. The use of DVHs summarizes the 3-D dose distribution data into a graphical representation of lesion and other structures of interest, such as the lumen wall, media, and adventia. Also, besides using DVHs on structure contours, the concept can also be applied to treatment volumes and isodose levels. The use of minimum/maximum dose to the lesion and other structures of interest gives a quantitative assessment of the dose range. Also, the average dose and median dose to a structure contour can be computed. Other ways of registering and evaluating the dose on images of the treatment volume will be apparent to those skilled in the art.
After the treatment planner 20 has completed its processing, a finalized treatment plan 52 is generated with a printer 50. The treatment plan 52 is then used by medical personnel in delivering the proper dose to the treatment volume. The treatment plan, for instance, specifies the dwell-time of the radiation source for each position along the length of a blood vessel or other treatment volume.
The treatment planner 20 further includes a controller 54 for controlling the operations of the various elements within the treatment planner 20. The controller 54, for instance, receives commands or data supplied through the voice recognition unit 36 or through other devices, such as the keyboard 42 or mouse 44. The controller 54 also controls operations of the printer 50, video recorder 38, and monitor 40. The data and commands received at the controller 54 are used in the operations of the other elements of the planner 20, such as the image processor 26, dose calculator 30, optimizer 32, and dose and image registration unit 34.
The treatment planner 20 preferably comprises a computer having at least a 200 MHz processor. The treatment planner 20 includes a sound card which is preferably a Sound Blaster 32 bit sound card manufactured by Creative Labs of Melpitas, Calif. The video frame grabber 22 is a ComputerEyes frame grabber manufactured by Digital Vision of Dedham, Mass., and the network interface is a Ethernet twisted pair/thinnet interface card manufactured by 3COM of Santa Clara, Calif. The planner 20 runs on Windows 95 operating system and preferably comprises at least 32 MB of RAM and at least a 6 GB hard drive.
A method for calculating and registering dose with ultrasound images will be described with reference to FIGS. 2A and 2B. At step 62, the ultrasound transducer 14 is inserted within the blood vessel or other tissue to be treated. At a step 64, the imaging system 12 acquires a transverse image at the position of the transducer 14 and determines, at step 66, whether the image is at a distal marker end. A distal end of the treatment volume may be indicated by a marker band at an end on a receptacle for the ultrasound transducer 14 which is visible on images displayed on the imaging system 12. With the marker bands, the proximal and distal ends of a treatment path may be used to define the treatment volume. Alternatively, the proximal and distal ends of the treatment volume may be defined by imaging anatomical landmarks or by viewing the position of the transducer 14 at the beginning of the treatment volume by some external device, such as fluoroscopy, diagnostic x-rays, CT, or MRI scans. The length along the treatment volume may therefore be determined from information generated by the automatic pull-back transducer 14 of imaging system 12 or by calibrated measurements on the external device.
The transducer 14 is moved along the length of the blood vessel or tissue to be treated until the transducer 14 is at a distal end of the treatment volume. At step 68, the imaging system 12 acquires an image at a discrete transverse slice through the blood vessel. The imaging system 12 assigns the image to a coordinate system at step 70 and determines, at step 72, whether the present image is at the proximal marker end. If the image is not at the proximal marker end, then at step 74 the imaging system 12 acquires the next image at the next position along the length of the treatment volume and at step 70 assigns this image to the established imaging coordinate system. In this manner, transverse images along the entire length of the treatment volume are acquired and assigned to the imaging coordinate system.
After images have been acquired for each position along the treatment volume, the treatment planner 20 receives the dose prescription 28 and the dose calculator 30 calculates the desired dose at step 76. The dose is calculated such that each radiation source has an independent coordinate system describing its location within the blood vessel. Each image acquired with the imaging system 12 in contrast, is placed on a Cartesian coordinate system with each pixel having a unique x and y coordinate value and each axial image having a unique z coordinate based on its distance from the distal end of the treatment volume. At step 78, a coordinate system transformation is executed by the dose and image registration unit 34 so that the image coordinate system may be registered with the radiation source coordinate system. In order to register the images and radiation source, internal landmarks are identified for establishing the location and position of the radiation source with respect to the imaging system 12. These landmarks are visible from an imaging device, such as fluoroscopy and intravascular ultrasound. As discussed above the landmarks may comprise metal marker bands at proximal and distal ends with these bands being placed on the transducer 14. The invention is not limited to the use of these metal marker bands and the landmarks may be identified in other ways, such as with a metal prosthesis, through identification of internal branching arteries, or through other ultrasonically visible marks that appear as an artifact on the image.
At step 80, the dose and image registration unit 34 overlays the dose distribution on each image of the treatment volume. The dose and image registration unit 34 at step 80 furthermore provides the superimposition of dose and images to the monitor 40 or other output device, such as the video recorder 38. In this manner, visualization of isodose distributions on each image or reformatted image occurs immediately and this information may be used to optimize subsequent therapy. At step 82, these images are evaluated to optimize subsequent therapy.
A method for performing vascular radio therapy to arrive at an optimal treatment plan will now be described with reference to FIGS. 3A to 3C. Vascular radio therapy is initiated at step 102 and at step 104 intravascular ultrasound is performed with the imaging system 12 using automated pullback. At step 106 the images acquired from the imaging system 12 are transferred to the treatment planner, as described in FIGS. 2A and 2B, at step 108 the treatment planner 20 selects those images within the treatment volume, and at step 110 the treatment planner 20 segments anatomical regions or an automated segmentation method may be used.
At step 112, an inquiry is made at the treatment planner 20 as to whether computer-aided optimization should be performed. If optimization should be performed, then at step 114 the optimizer 32 sets the desired dose to the segmented anatomy and at step 116 the dwell time and positions of the radiation source or sources are optimized. Any suitable optimization method may be used, such as the Simplex method or other linear programming, least squares, or simulated annealing. At step 122, the dose rate matrix is then calculated. If optimization is not to be performed, then at step 118 a prescription point is selected and at step 120 a source position is set. The treatment planner 20 then calculates the dose rate matrix based upon the selected prescription point and set source position. At step 124, the dose and image registration unit 34 registers the dose with image data.
At step 126, the treatment planner 20 computes DVH and statistics for display on monitor 40. At step 128, 2-D and 3-D superimposed isodose surfaces with imaging data are calculated and displayed on the monitor 40. Next, at step 130, a clinician determines whether the treatment plan is acceptable. If the plan is not acceptable, then processing returns to step 112 for another iteration of optimization or for the selection of different prescription points or source positions. When the treatment plan is acceptable, processing terminates at step 132 and the patient is treated.
An example of a layout of the imaging system 12 and treatment planner 20 for vascular radiotherapy is shown is FIG. 4. As shown in the layout 150, a patient 158 is placed upon an operating table 156 and monitors 154 for fluoroscopy are placed to one side. A cardiologist or interventional radiologist 160 is positioned near the patient 158 and controls the positioning of the ultrasound transducer 14 and other catheters. A radiation delivery device 152 is positioned near the treatment planner 20, which is also in close proximity to the imaging system 12. A radiation oncologist 162 is advantageously positioned proximate the imaging system 12 and treatment planner 20.
The voice recognition unit 34 of the treatment planner 20 allows the physician administering the treatment, who is dressed in sterile garb, to provide verbal commands to the treatment planner 20 in order to control operations of the treatment planner 20. The treatment planner 20 allows the patient 158 to remain at one location while the dose is calculated and while the optimal treatment plan is derived. The voice recognition unit uses a command and control engine for converting speech to text. One engine commercially available is the Microsoft Speech Engine. This engine can be used to incorporate voice commands for controlling software operations.
FIG. 5 shows a general method of operation for the speech voice recognition unit 34. At step 172, the voice recognition unit 34 listens for speech and detects speech at step 174. At step 176, the voice recognition unit 34 determines whether the phrase detected is a verbal command. If the speech is a verbal command, then at step 178 the voice and recognition unit 34 performs the command and returns to step 172 in order to continue to listen for speech. If the detected speech is not a command, as determined at step 176, then the voice recognition unit 34 returns directly to step 172 and listens for speech.
Because the imaging is performed within the cath lab, the patient 158 need not be moved subsequent to derivation of the treatment plan but rather remains at the same location for the actual surgical procedure. Because the imaging system 12 and treatment planner 20 remain within the surgical area, the imaging system 12 and treatment planner 20 may be used during a procedure to alter the treatment plan in the event that an unforeseen circumstance arises.
The imaging system 12 and treatment planner 20 provide a number of advantages over conventional treatment methods. For instance, whereas conventional techniques measure distances from a radiation source to a treated surface via an external device, the invention uses an intravascular ultrasound system 12 to acquire in-vivo images at the position of the radiation source. In other words, the ultrasound transducer 14 is placed at the position of the radiation source thereby allowing extremely accurate measurements to be obtained. The images obtained from the imaging system 12, furthermore, are not limited to two-dimensional images which have inherent inaccuracies due to variations in the third dimension but instead has the treatment planner 20 for formulating three-dimensional images of the treatment volume.
The step 76 of transforming between the IVUS coordinate system and radiation source coordinate system will be described in more detail with reference to FIGS. 6A and 6B. FIG. 6A is a side view of an ultrasound imaging wire 202 and FIG. 6B is a transverse ultrasound image. FIGS. 6A and 6B illustrate the imaging coordinate system 12 associated with the ultrasound imaging wire 202 inside a guide catheter 204 and vessel anatomy used by the treatment planner 20. The z-axis is directed from a distal end 204 a to proximal end 204 b along the direction of the guide catheter 203. An image plane is generally normal to the axial direction of the catheter 202 and establishes the z coordinate of the transverse ultrasound image. The x-y plane contains the coordinates associated with the axial direction of the catheter 202 on ultrasound image plane. This coordinate system defined by the x, y, and z axes is referred to as the IVUS coordinate system.
The radiation source is defined in the IVUS coordinate system and the longitudinal axis of the radiation source is parallel with the z-axis of the IVUS coordinate system. Each radiation source's center position is assigned a unique set of x, y, and z coordinates in the IVUS coordinate system. The dose calculation points are also defined in the IVUS coordinate system. At step 78, the dose is calculated by the dose calculator 30 from each radioactive source's position in the radiation source coordinate system. In order to calculate the dose to any point in the IVUS coordinate system, a transformation between the IVUS coordinate system and the radiation source coordinate system must be established. To transform a point (xI, yI, zI) from the IVUS coordinate system for the calculation in the radiation source coordinate system (x, y, z), the following equation is used:
where (tx, ty, tz) is the seed center position for the radiation source of interest defined in the IVUS coordinate system.
The step 78 of calculating dose to points in the radiation source coordinate system will be described below in more detail. The treatment planner 20 uses more than one dose calculation model depending on the type of radiation source with these models including semi-empirical methods, Monte Carlo methods, and experimentally measured data using a radiation detector with an adequate spatial resolution. A description of the model used for catheter-based radioactive line sources, including seeds, will be provided as an example for step 76. The dose is calculated by the dose calculator 20 within the treatment volume at any 3-D point for a radioactive line source. The dose calculator 20 uses the coordinates of the seed center (xo, yo, zo) and calculates the dose at a point in space, P(x, y, z). The longitudinal axis of the seed or line source is parallel with the z-axis of the coordinate system shown in FIG. 7A. A coordinate system transformation is first performed from P (x, y, z) to P′ (x′, y′, z′) located on a y-z plane. To perform this transformation, P (x, y, z) is rotated around z-axis in the clockwise direction by θz to obtain P′ (x′, y′, z′), according to the following equation:
A 2-D coordinate system shown in FIG. 7B is then used for dose calculation with x=x0. The dose calculator 20 next calculates the dose to the point P′(xo, y′, z′) and P′(y′, z′) is transformed to P″(r, θ) according to the following relationship:
The American Associate of Physicists in Medicine Task Group 43 ("AAPM TG43") in "Dosimetry of interstitial brachytherapy sources: Recommendations of the AAPM Radiation Therapy Committee Task Group No. 43," Med. Phys. 22 (2), pp. 209-234, February 1995, recommends a dose calculation formalism if the distance r from the source and angle θ is known according to FIG. 7B. The dose calculator 20 preferably calculates the dose to P″(r, θ) using TG43 experimentally determined factors according to the following relationships:
This step is performed for each dose calculation point and each seed position. The treatment planner 20 sums the contributions from each source position to each calculation point to provide a composite dose distribution.
A catheter 220 according to another aspect of the invention is shown in FIGS. 8A to 8C. The catheter 220 is a stereotactic localizing guiding catheter allowing clinicians to precisely determine the position and orientation of the ultrasonic transducer 14. To simplify the illustration of the invention, the ultrasonic transducer 14 has not been depicted in FIGS. 8A to 8C.
The catheter 220 includes a catheter body 222 generally formed in a cylindrical shape having proximal and distal ends. The catheter 220 is provided with a set of markers 224, 226, and 228 which allow clinicians to determine the location and orientation of the ultrasonic transducer 14. The first marker 224 extends along a length of the catheter 220 in parallel to the longitudinal axis of the catheter 220. The second marker 226 also extends along the length of the catheter 220 and is parallel to both the longitudinal axis of the catheter 220 and to the first maker 224. In this example, marker 224 is formed to have a width larger than that of the second marker 226 for reasons that will be apparent from the description below. The third marker 228 is formed along a straight line and wraps around the body 222 of the catheter 220 in a spiral fashion. The catheter 220 may also include one or more 20 regularly spaced markers 225, each of which forms a band around the circumference of the catheter body 222.
As described in greater detail above, one problem with conventional imaging techniques is that it is difficult to determine the precise location and orientation of a catheter within a patient. This difficulty is associated with not only imaging catheters but also with delivery catheters. Catheter 220 according to the invention addresses these problems by allowing clinicians to determine the precise location and orientation of a catheter. Each of the markers 224, 226 228, and 225 produces image artifacts in ultrasound images acquired with the transducer 14. Depending upon the relative positions of the markers 224, 226, 228, and 225, a clinician can determine the position of a catheter along the length of the catheter guide 220.
The manner of determining the precise location and orientation of a catheter within the catheter guide 220 will be described with reference to the examples shown in FIGS. 8B and 8C. FIGS. 8B and 8C illustrate ultrasound images acquired at positions indicated by the arrows 8B and 8C in FIG. 8A, respectively. Each ultrasound image, such as the images shown in FIGS. 8B and 8C, includes image artifacts for image markers 224 and 226. These two markers 224 and 226 are used to provide a reference axis from which an orientation of the catheter may be deduced. The image marker 224 forms an image artifact which is distinct from the image artifact produced by image marker 226. Consequently, clinicians can easily align the image artifacts produced from markers 224 and 226 with, for instance, a z axis extending along a length of a patient. Although a single marker, such as marker 224, may provide a reference point from which clinicians can eslimate a reference axis, the use of two markers 224 and 226 enable clinicians to accurately determine a reference axis and hence accurately determine the orientation of the catheter.
Whereas markers 224 and 226 are used to determine the orientation of a catheter, the markers 225 and 228 are used to determine the position of the catheter along the length of the catheter guide 220. The marker 228 wraps around the catheter body 222 in a regular manner traversing a distance along a length of the catheter body 222 proportionate to the marker's 228 angular position. In this example, the marker 228 extends from one marker 225 to the next marker 225 with the markers 225 being equally spaced from each other, such as every 10 cm. The clinician would therefore view an image artifact produced by an image marker 225 every 10 cm as the transducer 14 is being moved along the catheter guide 220.
To accurately determine the location of the transducer 14 or other catheter between the markers 225, the clinician can calculate the catheter's position based on the angular position of marker 228. The ultrasound image produced by the transducer 14 produces image artifacts for markers 224, 226, and 228 with the artifact for marker 228 being at an angle θ relative to a reference axis extending from marker 224 to marker 226. In general, the distance d of a catheter along the length of the catheter guide 220 is determined according to the following equation:
where θ is an angular position of the marker 228 relative to marker 224 and L is the wrapping distance, which in this example is the length between markers 225.
For example, with reference to FIG. 8B, an artifact for image marker 228 is at an angular position of θ1 where θ1 is equal to 45 degrees. The distance d of the catheter is therefore determined according to the following equation:
The clinician can therefore easily determine that the ultrasound image was taken at a distance of 1.25 cm from the previous marker band 225. As another example, when the transducer 14 is at the position 8C along the length of the catheter guide 220, the artifact for marker 228 is at an angle θ2 where θ2 is equal to 270 degrees. The position of the transducer 14 at this location is determined according to the following equation:
The catheter guide 220 eliminates much of the guesswork presently associated with the use of catheters. Rather than estimating the location of the transducer 14 or other catheter using a pull-back mechanism, the precise location of the catheter may be determined from artifacts produced from markers 224, 228 and 225. The ability to accurate locate a catheter provides clinicians with numerous advantages, such as the ability to accurately determine the location of a lesion, the ability to accurately deliver radiation to the precise location of the lesion, and the ability to acquire follow-up images precisely at the location of the lesion in order to assess the effectiveness of the treatment. Through the use of artifacts produced by markers 224 and 226, clinicians can accurately determine the precise orientation of a catheter. As a result, clinicians may accurately control the delivery of radiation to only certain angular segments of an artery thereby avoiding the delivery of radiation to healthy tissue. Furthermore, by imaging the treatment volume simultaneously with delivering radiation, clinicians can ensure that the correct dosage of radiation is delivered to the treatment volume.
The catheter body 222 may be comprised of any suitable material usually formed from any one of a wide variety of biologically compatible materials, typically being made from natural or synthetic polymers, such as silicone, rubber, natural rubber, polyethylene, polyvinylchloride, polyurethanes, polyesters, polytetrafluoroethylenes (PTFE). The catheter body 222 is often formed as a composite having a reinforcement material incorporated within the polymeric body in order to enhance its strength, flexibility, and toughness. Suitable enforcement layers include but are not limited to wire mesh layers. The flexible tubular members of the catheter body 222 will normally be formed by extrusion with one or more integral lumens being provided. If desired, a diameter the catheter can be modified by heat expansion and shrinkage using conventional techniques. The techniques for forming vascular catheters are well known to those skilled in the art.
The overall dimensions of the catheter 220 will depend on the particular application of the catheter, with the length varying widely but typically between about 40 cm and 150 cm and usually being between about 40 cm and 120 cm, for peripheral catheters and being between about 110 cm and 150 cm for coronary catheters. The diameter of the catheter body 222 may also valy widely, with the diameter typically being between about 3 to 6 French (F), wherein one French equals 0.33 mm.
The markers 224, 226, 228, and 225 may be applied to the catheter body 222 by any suitable technique, including conventional techniques used for forming fluoroscopic rings on intravascular catheters. A wide variety of ultrasonically visible marks can be provided including a hole formed in the body 222, an ultrasonically opaque material secured on or in the body 222 (such as paint, a flat piece of metal ribbon, or other low profile material capable of reflecting acoustic energy), or other ultrasound absorbing material such as urethanes, silicone, epoxies. The markers 224, 226, 228, and 225 may be formed using a heavy metal foil, such as gold or platinum, bonded to the exterior of the catheter body 222 using heat or a suitable adhesive, typically by embedding the foil into the catheter body 222 using heat. The markers are preferably encapsulated within a thin sheath, typically by heat shrinking polyethylene or other suitable thermoplastic, over the exterior of the catheter body 222.
The forgoing description of the preferred embodiments of the invention has been presented only for the purpose of illustration and description and is not intended to be exhaustive or to limit the invention to the precise form disclosed. Many modifications and variations are possible in light of the above teaching.
For example, although the invention has been described with the prevention of restenosis within a blood vessel, the invention may be used to treat other tissues. Moreover, the invention is not limited to the prevention of restenosis but may used to accurately deliver a dose of radiation for reasons other than the prevention of restenosis. The invention, for instance, may be used in the delivery of radiation to cancerous tissues, such as cervical cancer.
Additionally, the invention has been described with reference to an imaging transducer 14 and a separate radiation delivery catheter. It should be understood that the imaging transducer 14 and the radiation catheter may be combined into a single unit.
Further, the catheter guide according to the invention may be formed with any number of markers 225 or may be formed without markers 225. Additionally, although the markers 224 and 226 are distinguishable based on their widths, the markers 224 and 226 may be distinguishable in other ways. For instance, markers 224 and 226 may be formed at different heights or markers 224 and 226 may be formed with different materials resulting in different image artifacts. As should be apparent to those skilled in the art, maker 228 may wrap entirely around the body 222 or may only partially wrap around the body 222. The angle at which the marker 228 wraps around the body 222 may also be varied in order to establish a desired relationship between the angular difference θ and the length L. For instance, for an angular difference of 90 degrees, the marker 228 may traverse 1 cm, 5 cm, 10 cm or any desired distance along the length of the body 222. As another example, marker 224 may be formed as a band of two markers while marker 226 remains a single marker. Also, although the catheter guide 220 has been described with reference to the ultrasound transducer 14, the catheter guide 220 may be designed to operate with other types of imaging transducers.
The guide according to the invention is not restricted to catheters used to treat restenosis may be used with any suitable device and for any suitable purpose. Thus, the guide may be used for diagnosis, treatment, or verification. The guide, moreover, may be used with an endobronclial catheter, an intra-uterine device, with metal cylinders which are either straight or curved, or with other interstitial or intra-cavity devices. Other uses of the guide will become apparent to those skilled in the art.
The guide according to the invention is furthermore not limited to use with ultrasound imaging transducers but may be used with other imaging techniques. The guide, for instance, may be used with magnetic resonance (MR), computer tomographic (CT), or other tomographic techniques.
The embodiments were chosen and described in order to explain the principles of the invention and their practical application so as to enable others skilled in the art to utilize the invention and various embodiments and with various modifications as are suited to the particular use contemplated.
|Cited Patent
|Filing date
|Publication date
|29 Jul 1974
|19 Oct 1976
|The United States Of America As Represented By The Department Of Health, Education And Welfare
|Method of radiation therapy treatment planning
|17 Jun 1987
|10 Jan 1989
|Victory Engineering Company
|Radiopaque marker for stereotaxic catheter
|26 Mar 1987
|7 Feb 1989
|Washington Research Foundation
|Endoscopically deliverable ultrasound imaging system
|22 Aug 1988
|27 Mar 1990
|General Electric Company
|High frequency focused ultrasonic transducer for invasive tissue characterization
|27 Oct 1987
|17 Apr 1990
|Endosonics Corporation
|Apparatus and method for imaging small cavities
|3 Apr 1989
|28 Aug 1990
|B. Braun-Ssc Ag
|Color marks on catheter guide wire
|18 May 1989
|2 Jul 1991
|University Of Florida
|Dosimetric technique for stereotactic radiosurgery same
|17 Dec 1990
|8 Oct 1991
|Cardiovascular Imaging Systems, Inc.
|Ultrasonic imaging catheter having rotational image correlation
|19 Dec 1989
|11 May 1993
|Scimed Life Systems, Inc.
|Method for placement of a balloon dilatation catheter across a stenosis and apparatus therefor
|27 Dec 1990
|9 Nov 1993
|Wormer Mark E Van
|Acoustically enhanced catheter
|16 Feb 1993
|12 Jul 1994
|Griffith James M
|Combination catheter for invasive probe delivery and balloon dilation
|14 Dec 1992
|23 Aug 1994
|Medical Instrumentation And Diagnostic Corporation
|Three-dimensional computer graphics simulation and computerized numerical optimization for dose delivery and treatment planning
|28 Sep 1992
|13 Sep 1994
|Kabushiki Kaisha Toshiba
|Diagnostic apparatus for circulatory systems
|9 Sep 1992
|18 Oct 1994
|Kabushiki Kaisha Toshiba
|Apparatus for diagnosing vascular systems in organism
|30 Jun 1992
|8 Nov 1994
|Cardiovascular Imaging Systems, Inc.
|Automated longitudinal position translator for ultrasonic imaging probes, and methods of using same
|23 Feb 1994
|21 Feb 1995
|William Beaumont Hospital
|Real time radiation treatment planning system
|3 Aug 1994
|21 Mar 1995
|Batten; Bobby G.
|Slaved biopsy device, analysis apparatus, and process
|14 Sep 1993
|25 Apr 1995
|Cardiac Pathways Corporation
|Endocardial mapping and ablation system utilizing separately controlled steerable ablation catheter with ultrasonic imaging capabilities and method
|13 Dec 1993
|4 Jul 1995
|The Spectranetics Corporation
|Radiopaque tip marker for alignment of a catheter within a body
|14 Sep 1993
|10 Oct 1995
|Spectranetics Corp
|Fiber optic catheter with shortened guide wire lumen
|4 Aug 1994
|23 Jan 1996
|Cardiovascular Imaging Systems, Inc.
|Automated longitudinal position translator for ultrasonic imaging probes, and methods of using same
|4 Aug 1994
|12 Mar 1996
|Olympus Optical Co., Ltd.
|Ultrasonic image diagnosing apparatus for displaying three-dimensional image
|8 Apr 1994
|7 May 1996
|Spectranetics Corporation
|Fiber optic guide wire and support catheter therefor
|12 Dec 1995
|14 Jan 1997
|Cardiovascular Imaging Systems, Inc.
|Automated longitudinal position translator for ultrasonic imaging probes, and methods of using same
|6 Jun 1995
|28 Jan 1997
|Yock; Paul
|Rotational correlation of intravascular ultrasound image with guide catheter position
|3 Aug 1994
|25 Mar 1997
|Hyal Pharmaceutical Corporation
|Use of hyaluronic acid and forms to prevent arterial restenosis
|30 Sep 1994
|8 Apr 1997
|Liprie; Samuel F.
|Catheter for maneuvering radioactive source wire to site of treatment
|15 Nov 1994
|1 Jul 1997
|Neocardia, Llc
|Method and apparatus for uniform radiation treatment of vascular lumens
|4 Oct 1995
|1 Jul 1997
|Spectranetics Corporation
|Fibert optic guide wire and support catheter therefor
|6 Jun 1995
|29 Jul 1997
|Cardiovascular Imaging Systems, Inc.
|Method and apparatus for intravascular
|28 May 1996
|26 Aug 1997
|Coraje, Inc.
|Intravascular ultrasound imaging guidewire
|4 Jun 1996
|23 Sep 1997
|Intravascular Research Limited
|Guide wire for a catheter with position indicating means
|3 Dec 1993
|14 Oct 1997
|Cardiovascular Imaging Systems, Inc.
|Method and apparatus for intravascular ultrasonography
|12 Sep 1996
|4 Nov 1997
|Cardiovascular Imaging Systems, Inc.
|Guidewire with imaging capability
|27 Oct 1994
|4 Nov 1997
|Novoste Corporation
|Method and apparatus for treating a desired area in the vascular system of a patient
|1 May 1995
|30 Dec 1997
|Cedars-Sinai Medical Center
|Ultrasonic transducer orientation sensing and display apparatus and method
|30 Nov 1995
|13 Jan 1998
|The Trustees Of Columbia University In The City Of New York
|Apparatus and method to reduce restenosis after arterial intervention
|3 Apr 1996
|17 Mar 1998
|Saad; Saad A.
|Catheter with integral electromagnetic location identification device
|7 Mar 1997
|12 Jan 1999
|Hibbard; Lyn
|Autosegmentation/autocontouring system and method for use with three-dimensional radiation therapy treatment planning
|20 Oct 1997
|9 Mar 1999
|Cardiovascular Imaging Systems, Inc.
|Rotational correlation of intravascular ultrasound image with guide catheter position
|10 Dec 1996
|16 Mar 1999
|Neocardia, Llc
|Device and method for controlling dose rate during intravascular radiotherapy
|7 Jun 1995
|21 Dec 1999
|Boston Scientific Corporation
|Catheters for imaging, sensing electrical potentials, and ablating tissue
|"Hyal Receives First U.S. Patent Allowance for Prevention of Restensois," Canada News Wire, (Dec. 1996).
|EndoSonics, "EndoSonics Announces Software for 3-Dimensional Display of Intravascular Ultrasound Images," pp. 1-2 (website date Dec. 10, 1997).
|EndoSonics, "In-Vision(TM) The Future in IntraCoronary Ultrasound Now . . . " pp. 1-6 (website date Dec. 10, 1997).
|EndoSonics, "Oracle(R) Imaging System IntraCoronary Ultrasound Imaging," pp. 1-3 (website date Dec. 10, 1997).
|EndoSonics, "In-Vision™ The Future in IntraCoronary Ultrasound Now . . . " pp. 1-6 (website date Dec. 10, 1997).
|EndoSonics, "Oracle® Imaging System IntraCoronary Ultrasound Imaging," pp. 1-3 (website date Dec. 10, 1997).
|EndSonics, "Solid-State Intracoronary Ultrasound: A Technology Whose Time Has Come," pp. 1-3 (website date Dec. 10, 1997).
|Fox, et al., "Calculated Dose Distributions of Beta-Particle Sources Used for Intravascular Brachytherapy," American Society of Therapeutic Radiology and Oncology (ASTRO) Annual Conference, Orlando, Florida (Oct. 19-23, 1997).
|Mintz, Gary S., et al., "Intravascular Ultrasound Predictors of Restenosis After Percutaneous Transcather Coronary Revascularization" J. Am. Coll. Cardiol. 27:1678-1687 (1996).
|Moran, John F., "Management of Restenosis: The Challange for Angioplasty in the 1990's," Midwest Cardiovascular Institute, Good Samaritan Hospital, Viewpoint (Winter 1996).
|Nath, et al., "Dosimetry of interstitial brachytherapy sources: Recommendations of the AAPM Radiation Therapy Committee Task Group No. 43," Med. Phys. 22(2):209-234 (Feb. 1995).
|Novoste(TM) 1996 Annual Report.
|Novoste™ 1996 Annual Report.
|Soares, et al., "Calibration and characterization of beta-particle sources for intravascular brachytherapy," Med. Phys., 25(3):339-346 (1998).
|Citing Patent
|Filing date
|Publication date
|11 Aug 2000
|20 Aug 2002
|Alpha Intervention Technology, Inc.
|Indentification and quantification of needle displacement departures from treatment plan
|1 Feb 2002
|22 Nov 2005
|Ge Medical Systems Global Technology Company, Llc
|System and method for wireless voice control of an interventional or diagnostic medical device
|30 Sep 2002
|21 Aug 2007
|Smiths Group Plc
|Medico-surgical devices
|13 Jan 2004
|21 Aug 2007
|Cheetah Omni, Llc
|System and method for voice control of medical devices
|30 Mar 2004
|7 Oct 2008
|Cheetah Omni, Llc
|Infra-red light source including a raman shifter
|8 Sep 2008
|15 Dec 2009
|Cheetah Omni, Llc
|System and method for generating infrared light for use in medical procedures
|16 Sep 2008
|11 May 2010
|Innovatech, Llc
|Marked precoated strings and method of manufacturing same
|23 Oct 2003
|14 Sep 2010
|Medtronic, Inc.
|Medical paddle lead and method for spinal cord stimulation
|11 Mar 2009
|12 Oct 2010
|Innovatech, Llc
|Marked precoated medical device and method of manufacturing same
|23 Apr 2010
|12 Apr 2011
|Innovatech Llc
|Marked precoated strings and method of manufacturing same
|19 Apr 2011
|Volcano Corporation
|Vascular image co-registration
|29 Sep 2006
|14 Jun 2011
|Aptus Endosystems, Inc.
|Endovascular aneurysm repair method
|21 Dec 2007
|1 Nov 2011
|Innovatech, Llc
|Marked precoated medical device and method of manufacturing same
|5 Feb 2009
|8 Nov 2011
|Cheetah Omni, L.L.C.
|Broadband or mid-infrared fiber light sources
|17 Jul 2007
|15 Nov 2011
|West Corporation
|System, method, and computer-readable medium for verbal control of a conference call
|30 Oct 2007
|27 Dec 2011
|Aptus Endosystems, Inc.
|Endovascular aneurysm repair systems and methods
|24 Jun 2005
|10 Jan 2012
|Aptus Endosystems, Inc.
|Endovascular aneurysm repair system
|24 Nov 2009
|17 Jan 2012
|Cheetah Omni, Llc
|System and method for voice control of medical devices
|25 Feb 2004
|31 Jul 2012
|Aptus Endosystems, Inc.
|Systems and methods for attaching a prosthesis within a body lumen or hollow organ
|11 Jul 2008
|31 Jul 2012
|Innovatech, Llc
|Marked precoated medical device and method of manufacturing same
|9 Feb 2009
|31 Jul 2012
|Innovatech, Llc
|Marked precoated medical device and method of manufacturing same
|23 Jun 2006
|30 Oct 2012
|Volcano Corporation
|Three dimensional co-registration for intravascular diagnosis and therapy
|25 Jan 2011
|30 Oct 2012
|Nucletron Operations B.V.
|Real time radiation treatment planning system
|4 Mar 2011
|29 Jan 2013
|Innovatech, Llc
|Marked precoated strings and method of manufacturing same
|27 Sep 2011
|19 Feb 2013
|West Corporation
|System, method and computer-readable medium for verbal control of a conference call
|23 Sep 2011
|5 Mar 2013
|Cheetah Omni, L.L.C.
|Broadband or mid-infrared fiber light sources
|12 Jan 2012
|25 Jun 2013
|Cheetah Omni, Llc
|System and method for voice control of medical devices
|3 Jul 2012
|5 Nov 2013
|Innovatech, Llc
|Marked precoated medical device and method of manufacturing same
|25 Jan 2013
|11 Mar 2014
|Omni Medsci, Inc.
|Broadband or mid-infrared fiber light sources
|25 Jun 2012
|25 Mar 2014
|Omni Medsci, Inc.
|System and method for voice control of medical devices
|13 Jun 2012
|1 Apr 2014
|Aptus Endosystems, Inc.
|Systems and methods for attaching a prosthesis with a body lumen or hollow organ
|2 Nov 2010
|8 Apr 2014
|Aptus Endosystems, Inc.
|Devices, systems, and methods for prosthesis delivery and implantation, including the use of a fastener tool
|24 Sep 2012
|3 Jun 2014
|Nucletron Operations B.V.
|Real-time radiation treatment planning system
|16 Jan 2013
|8 Jul 2014
|Innovatech, Llc
|Marked precoated strings and method of manufacturing same
|31 Mar 2010
|29 Jul 2014
|Matthew R. Witten
|System and method for radiation therapy treatment planning using a memetic optimization algorithm
|10 Jun 2013
|30 Sep 2014
|Omni Medsci, Inc.
|System and method for voice control of medical devices
|9 Mar 2012
|2 Dec 2014
|Innovatech, Llc
|Marked fluoropolymer surfaces and method of manufacturing same
|27 Jun 2012
|27 Jan 2015
|Innovatech Llc
|Marked precoated medical device and method of manufacturing same
|30 Nov 2007
|3 Feb 2015
|The John Hopkins University
|Image registration of multiple medical imaging modalities using a multiple degree-of-freedom-encoded fiducial device
|5 Nov 2013
|3 Mar 2015
|Omni Medsci, Inc.
|Broadband or mid-infrared fiber light sources
|9 Jun 2011
|5 May 2015
|Aptus Endosystems, Inc.
|Devices, systems, and methods for supporting tissue and/or structures within a hollow body organ
|3 Sep 2014
|16 Jun 2015
|Omni Medsci, Inc.
|System and method for voice control of medical devices
|10 May 2011
|16 Jun 2015
|Siemens Aktiengesellschaft
|Method for positioning the focus of a gradient field and treatment facility
|23 Aug 2011
|30 Jun 2015
|Omni Medsci, Inc.
|Laser-based method and system for selectively processing target tissue material in a patient and optical catheter assembly for use therein
|21 Feb 2014
|7 Jul 2015
|Omni Medsci, Inc.
|Broadband or mid-infrared fiber light sources
|13 Jul 2011
|11 Aug 2015
|Volcano Corporation
|Apparatus and method for use of RFID catheter intelligence
|23 Oct 2003
|23 Dec 2004
|Medtronic, Inc.
|Medical lead and method
|16 Sep 2003
|17 Mar 2005
|Robert Sutherland
|Localization of a sensor device in a body
|2 Aug 2004
|30 Jun 2005
|Failla Gregory A.
|Deterministic computation of radiation doses delivered to tissues and organs of a living organism
|21 Jul 2004
|4 Aug 2005
|Gabor Fichtinger
|Registration of ultrasound to fluoroscopy for real time optimization of radiation implant procedures
|24 Jun 2005
|27 Oct 2005
|Aptus Endosystems, Inc.
|Endovascular aneurysm repair system
|17 Mar 2003
|15 Dec 2005
|Robert Sutherland
|Angiogram display overlay technique for tracking vascular intervention sites
|11 Jan 2006
|26 Oct 2006
|Volcano Corporation
|Vascular image co-registration
|14 Nov 2005
|16 Nov 2006
|Failla Gregory A
|Deterministic computation of radiation transport for radiotherapy dose calculations and scatter correction for image reconstruction
|3 Nov 2011
|Volcano Corporation
|Apparatus and Method for Use of RFID Catheter Intelligence
|17 Nov 2011
|Klaus Klingenbeck
|Method for positioning the focus of a gradient field and treatment facility
|24 Nov 2011
|Nucletron B.V.
|Real time radiation treatment planning system
|7 May 2012
|8 Nov 2012
|Edward H Cully
|Echogenic Sleeve
|10 Jan 2013
|Tenney John A
|Calibration and transformation of a camera system's coordinate system
|18 Apr 2013
|2 Jun 2015
|Volcano Corporation
|Vascular image co-registration
|24 Nov 2005
|31 May 2007
|Siemens Ag
|X-ray brachytherapy instrument comprises probe with x-ray source irradiating zone limited by shield and illustrated with x-ray markers in known spatial relationship
|24 Nov 2005
|8 Apr 2010
|Siemens Ag
|Einrichtung für die Röntgen-Brachytherapie mit einer in das Innere eines Körpers zur Röntgen-Brachytherapie einführbaren Sonde
|4 Jun 2003
|18 Dec 2003
|Varian Medical Systems, Inc.
|An angiogram display overlay technique for tracking vascular intervention sites
|22 Sep 2006
|26 Apr 2007
|Aptus Endosystems Inc
|Endovascular aneurysm devices, systems, and methods
|U.S. Classification
|600/466, 600/467
|International Classification
|Cooperative Classification
|European Classification
|14 Feb 2005
|Fee payment
Year of fee payment: 4
|23 Feb 2009
|Maintenance fee reminder mailed
|14 Aug 2009
|Lapse for failure to pay maintenance fees
|6 Oct 2009
|Expired due to failure to pay maintenance fee
Effective date: 20090814


text:
- Taco Day
Some Guy Tried to Eat a Burger with 1,050 Strips of Bacon
Today on The Daily Meal
Recipe of the day
So, Japanese Burger Kings decided to take "Have it your way" literally, and started offering 15 slices of bacon for $1, which is a pretty good deal.
But then, Rocket News 24 had to go get this guy to buy a hamburger with 1,050 rashers of bacon, paying some $70 plus tax/tip for the monstrous order. It's Epic Meal Time-worthy, except with one poor stupid guy trying to eat his way through. Where is Kobayashi when you need him? Watch below.
Be a Part of the Conversation
Join the Daily Meal's Community and Share your Thoughts
Have lots to say?
Contribute your own article.
Know a great recipe worth sharing?
We want it. Post it now!
Enjoyed (or not enjoyed) a restaurant experience?
Write a review.
Tasted a bottle of wine you\'d like to tell us about?
Write a review.

subset: commoncrawl


text:
News . Feature Stories
News Highlight
September 17, 2015
By Alyssa Brown
From a young age, Shriya Garg has had a passion and talent for art. She fondly remembers her mother bringing home huge boxes of crayons and colored pencils, which Garg used to draw all over the walls. Instead of attempting to rein in those examples of her artistic exuberance, Garg's parents preserved it, and her earliest work still decorates the walls of her former family home.
As she grew older, Garg took art classes outside of her normal academic schedule, and participated in summer art school at Parsons School of Design. While her family lives in India, Garg is...continue reading
March 04, 2014
Ten CIA grads talk about automotive design careers in three new videos
Alumni working at Chrysler, Ford and GM interviewed more
February 28, 2014
Progressive acquires artwork by CIA instructor Dan Tranberg
Thirteen of Tranberg's recent works now in renowned collection more
February 26, 2014
CIA students explore sound art and contemporary music as they relate to the visual arts
Adjunct faculty member will deconstruct sound in March 7 Lunch On Fridays talk more
February 24, 2014
Teens in museum program learn about art careers at CIA
Students build games, 21st century skills in Industrial Design workshop more
February 19, 2014
Photography major captures images on hand-blown glass
Senior Emma Howell featured on Wired.com for her innovative process more
February 14, 2014
Glass major wins award in Niche magazine competition
Amanda Wilcox's "Cling" was chosen from among hundreds of entries more
For more information about this or other CIA news, contact us here.


text:
We use cookies to improve your experience, show you products you may like and save your cart. Continue if you're ok with this, or read more in our privacy policy.
If you choose to pay by Debit/Credit Card, you can complete your order immediately.
If you choose PayPal, you will be redirected to the PayPal sign in page.
Please check the data you entered carefully. Make sure, your Billing Address is correct, otherwise your order will be delayed. Click the 'I agree and continue' button to go to the order verification section.
If you still need help, please visit the Customer Care Center
In the event that you have purchased the wrong product, please do the following:
Note: The Money Back Guarantee is valid for up to 14 days from the date of purchase for goods bought via the Global eStore.
Share this page


text:
- Find a rental
- Manage your booking
- Monte San Savino rentals
- How to book
- Help Center
- Write a review
Villa Fabian
From $112 / week
Description from the manager
Description from the manager
Bedrooms: 2/
Bathrooms: 2/
Sleeps: 6/
Type: House
Two Bedroom Split Level Villa in Arezzo area.
Villa Fabian is situated in a lovely position close to Monte San Savino a pretty hill town in the verdant countryside between Arezzo and the Chianti area. There are beautiful views of the valley and town from the swimming pool ideally placed in the grounds below the villa, where there is also a Jacuzzi tub. Simply furnished in homely country style, the villa is surrounded by a large garden with gazebo perfect for dining al fresco whilst enjoying the lovely panorama.
Villa Fabian has two floors, the ground floor consists of a kitchen with dining table and chairs, a spacious room with fireplace, settee, ...Read more
About the manager
About the manager
La Bella Toscana
La Bella Toscana
- Response rate: 100%
- Response time: A few hours
- Calendar updated: today
Overall rating:
Additional Location Information
Show rates details
Also consider
- FlipKey has rentals in over 11,000 destinations around the world:
- Myrtle Beach vacation rentals - 1,126 rentals from $28/night
- Gatlinburg vacation rentals - 1,254 rentals from $103/night
- Port Aransas vacation rentals - 559 rentals from $149/night
- Mission Beach vacation rentals - 53 rentals from $178/night
- Fort Lauderdale vacation rentals - 440 rentals from $111/night


text:
NYCC: The Dark Knight 30th Anniversary with Frank Miller and More
Comic Books
Showing results 17393-17412 of 22677
Sponsored by Image Comics, ComicBookResources.Com and Things From Another World, the Comic Book Legal Defense Fund kicks off Comic-Con with a welcome party on Thursday night.
Marvel has provided CBR with an advance look at solicitation for its Ultimate Comics line of titles shipping in October including new issues of Ultimate Comics "Avengers," "Armor Wars" and "Spider-Man."
Courtesy of Image, CBR brings you advance looks at new comics on sale next week including "Gemini" #4, "Invincible" #64, "Tasty Bullet" and "Spawn Origins" vol. 2.
Courtesy of Dynamite Entertainment, CBR brings you exclusive first looks at variant covers for "Project Superpowers: Chapter 2" #3 by Lucio Parillo and Stephen Segovia.
Courtesy of Marvel, CBR brings you advance looks at new comics on sale next Wednesday including "Amazing Spider-Man" #600, "Dark Wolverine" #76, "Deadpool: Suicide Kings" #4, "Immortal Weapons" #1 and more.
Wizard Conventions has announced their latest convention offering to be held at the Anaheim Convention Center in Southern California April 16th-18th, 2010.
Courtesy of Top Cow, CBR brings you exclusive first looks at Adriana Melo and Sunny Gho's covers for "Witchblade" #130 by Ron Marz and Stjepan Sejic, concluding the "War of the Witchblades" storyline.


text:
Intermarkets' Privacy Policy
Donate to Ace of Spades HQ!
Recent Entries
The Real Pay Gap Progressives Won't Talk About: Federal Workers Make 78 Extra Cents on the Dollar Compared to Their Private Sector Analogues
"Homeland" Opener Offers Some Eye-Opening Analysis of the War on Terror
Spencer Stone Stabbed While Defending Friend in Barfight
Bombshell: McCarthy Drops Out Of Race For Speaker Of The House
Open Thread
Thursday Morning News Dump
Top Headlines Comments
Overnight Open Thread (10-7-2015) – Lonely Women and Beardless Men Edition
Hillary's Liberal Media Defenders Got the Word From Hillary HQ, I Guess
Hillary Comes Out Against TPP, Pleasing Her Left Flank
« Overnight Open Thread | Main | Tuesday Financial Briefing »
June 29, 2010
Al Gore: Release My Second Chakra
It's just crazy enough to possibly be true.
Gore said he was tired from travel and described in detail the massage he wanted. It included work on the adductor muscles, which are on the inside of the thighs. "I mentally noted that a request for adductor work is a bit unusual," the masseuse told police, because it can be "a precursor to inappropriate behavior by a male client."
| Access Comments
Recent Comments
Mike Hammer's Cat: "176 Homeland was interesting tonight, in the conte ..."
Sherry McEvil, Stiletto Corsettes: "Everbody hates Skyler. Her sister is ok. Posted b ..."
Bruce With a Wang!: " While Walter White was seen sympathetically by t ..."
Alberta Oil Peon: "It's a guided tour though and it takes about 30 mi ..."
Bertram Cabot Jr.: "[i]I just got a survey from the GOP. Oh, they ..."
All Hail Eris, Wrapped in Bacon: "Posted by: Bruce With a Wang! at October 08, 2015 ..."
@votermom: "303 Say my name! Posted by: Walter White at Octob ..."
Lauren: "One of the questions: "Should the Republican Pa ..."
Grump928(c): "[i]Everbody hates Skyler. Her sister is ok.[/i] ..."
Lauren: "I just got a survey from the GOP. Oh, they bet ..."
Bertram Cabot Jr.: "[i]"oh my God nothing will get done!" "Govt gr ..."
Pappy O'Daniel: "Obama wanted the Muslim Brotherhood running Egypt ..."
Recent Entries
The Real Pay Gap Progressives Won't Talk About: Federal Workers Make 78 Extra Cents on the Dollar Compared to Their Private Sector Analogues
"Homeland" Opener Offers Some Eye-Opening Analysis of the War on Terror
Spencer Stone Stabbed While Defending Friend in Barfight
Bombshell: McCarthy Drops Out Of Race For Speaker Of The House
Open Thread
Thursday Morning News Dump
Top Headlines Comments
Overnight Open Thread (10-7-2015) – Lonely Women and Beardless Men Edition
Hillary's Liberal Media Defenders Got the Word From Hillary HQ, I Guess
Hillary Comes Out Against TPP, Pleasing Her Left Flank
Polls! Polls! Polls!
Election Projection
Federal Review
Real Clear Politics
Rasmussen Reports
FoxNews Polls
Sabato/Crystal Ball
Frequently Asked Questions
The (Almost) Complete Paul Anka Integrity Kick
Primary Document: The Audio
Paul Anka Haiku Contest Announcement
Integrity SAT's: Entrance Exam for Paul Anka's Band
AllahPundit's Paul Anka 45's Collection
AnkaPundit: Paul Anka Takes Over the Site for a Weekend (Continues through to Monday's postings)
George Bush Slices Don Rumsfeld Like an F*ckin' Hammer
Top Top Tens
Democratic Forays into Erotica
New Shows On Gore's DNC/MTV Network
Nicknames for Potatoes, By People Who Really Hate Potatoes
Star Wars Euphemisms for Self-Abuse
Signs You're at an Iraqi "Wedding Party"
Signs Your Clown Has Gone Bad
Signs That You, Geroge Michael, Should Probably Just Give It Up
Signs of Hip-Hop Influence on John Kerry
NYT Headlines Spinning Bush's Jobs Boom
Things People Are More Likely to Say Than "Did You Hear What Al Franken Said Yesterday?"
Signs that Paul Krugman Has Lost His Frickin' Mind
All-Time Best NBA Players, According to Senator Robert Byrd
Other Bad Things About the Jews, According to the Koran
Signs That David Letterman Just Doesn't Care Anymore
Examples of Bob Kerrey's Insufferable Racial Jackassery
Signs Andy Rooney Is Going Senile
Other Judgments Dick Clarke Made About Condi Rice Based on Her Appearance
Collective Names for Groups of People
John Kerry's Other Vietnam Super-Pets
Cool Things About the XM8 Assault Rifle
Media-Approved Facts About the Democrat Spy
Changes to Make Christianity More "Inclusive"
Secret John Kerry Senatorial Accomplishments
John Edwards Campaign Excuses
John Kerry Pick-Up Lines
Changes Liberal Senator George Michell Will Make at Disney
Torments in Dog-Hell
Greatest Hitjobs
The Ace of Spades HQ Sex-for-Money Skankathon
A D&D Guide to the Democratic Candidates
Margaret Cho: Just Not Funny
More Margaret Cho Abuse
Margaret Cho: Still Not Funny
Iraqi Prisoner Claims He Was Raped... By Woman
Wonkette Announces "Morning Zoo" Format
John Kerry's "Plan" Causes Surrender of Moqtada al-Sadr's Militia
World Muslim Leaders Apologize for Nick Berg's Beheading
Michael Moore Goes on Lunchtime Manhattan Death-Spree
Milestone: Oliver Willis Posts 400th "Fake News Article" Referencing Britney Spears
Liberal Economists Rue a "New Decade of Greed"
Artificial Insouciance: Maureen Dowd's Word Processor Revolts Against Her Numbing Imbecility
Intelligence Officials Eye Blogs for Tips
They Done Found Us Out, Cletus: Intrepid Internet Detective Figures Out Our Master Plan
Shock: Josh Marshall Almost Mentions Sarin Discovery in Iraq
Leather-Clad Biker Freaks Terrorize Australian Town
When Clinton Was President, Torture Was Cool
What Wonkette Means When She Explains What Tina Brown Means
Wonkette's Stand-Up Act
Wankette HQ Gay-Rumors Du Jour
Here's What's Bugging Me: Goose and Slider
My Own Micah Wright Style Confession of Dishonesty
Outraged "Conservatives" React to the FMA
An On-Line Impression of Dennis Miller Having Sex with a Kodiak Bear
The Story the Rightwing Media Refuses to Report!
Our Lunch with David "Glengarry Glen Ross" Mamet
The House of Love: Paul Krugman
A Michael Moore Mystery (TM)
The Dowd-O-Matic!
Liberal Consistency and Other Myths
Kepler's Laws of Liberal Media Bias
John Kerry-- The Splunge! Candidate
"Divisive" Politics & "Attacks on Patriotism" (very long)
The Donkey ("The Raven" parody)
Powered by
Movable Type 2.64


text:
Check Dates
Hollyhock Country House is surrounded by vineyards located on a quiet country road two miles from historic Sonoma Plaza. Confortable living in a 1920's farmhouse reflects the simple life from years past. Enjoy a charming bathroom with a clawfoot tub by a picture window overlooking the garden. The guest room is furnished with interesting antiques including an armoire from the early 1900,s, and original paintings created by the Innkeeper. A Continental breakfast is served in the original sunporch, and on cool mornings a fire burns in a freestanding fireplace. A time to enjoy a warm and cozy moment.
BedandBreakfast.com gift cards are the perfect gift for anyone who needs a relaxing getaway.
Score Breakdown based on22 Reviews
Write A Review
on October 28, 2014
We loved staying with Lucille in her beautiful country house!
So very different and very homely filled with wonderful things.
Sonoma very near is a nice old town.
Innkeeper ResponseThank you for being a satisfied guest at Hollyhock Country House.
on September 06, 2014
We checked in a couple of days after the Napa Earthquake. Even though the house was in Sonoma there were a couple of cracks in the plaster. The house is in need of some TLC and the home of an accomplished artist.
The bedroom is small and the bad a normal double -- adequate for one but crowded for two. Fortunately we had a cool spell in the evening as there is no air conditioning. Price is right and one of the few B&B's that did not want to sell you the whole 3 day holiday weekend.
There is no shower only a bathtub with a funky plastic hose to use to wash your hair.
There is a cat on premise -- so if you have allergies it may be an issue. Cat is actually very tame and always wanted to be petted.
The hostess is charming and the breakfast good. We would not do it again however.
on February 22, 2014
This was such a nice place to stay if you are looking for something different than a big hotel. Hollyhock is located in the middle of vineyards and if you arrive at night it may seem a little sketch, but we were inside we didn't want leave! It is beautifully decorated and the innkeeper was and very welcoming. Our breakfast was out on the sun deck and it was beautifully decorated with art and artifacts from her travels. We loved it!
Innkeeper ResponseThank you for taking the time to review Hollyhock Country House.
on December 20, 2013
Our wine country weekend get away was perfectly completed by our stay at the Hollyhock. Very peaceful, very quaint. Highly recommend! DYLLIC!
on November 22, 2013
Lucille was a great host and we found her home to be very warm and inviting. We only stayed one night, but it was the perfect little hideaway for our one day wine tasting in Sonoma. The breakfast was tasty and Lucille shared her recent travel stories with us. It was very nice.
Innkeeper ResponseThank you for your reservation at Hollyhock Country House. I enjoyed getting to know you and hope to see you again. I like to think there are no strangers at Hollyhock Country House, only friends who have not yet met!
on July 10, 2013
Our room at Hollyhock Country House was clean and comfortable, and well appointed. Since we were the only couple there, we did not have to share the bathroom--a definite plus. The owner/innkeeper is very friendly and nice, and bakes delicious homemade pastries from scratch for her guests every day. It's very low key, and we were free to come in late, which we appreciated. The only drawback to Hollyhock Country House is that it's on well water, which has a very distinct smell. The owner does use bottled water for her cooking, coffee, etc., but neither my husband or I wanted to bathe there because we didn't like the smell of the water. Otherwise, Hollyhock Country House is a good value compared to the other expensive options in the area. No two-night minimum was also a plus. .
on October 07, 2012
This was my first time staying at a B&B. Lucille greeted us warmly into her country home and made us feel very welcomed. We found the artistic decor unique and entertaining. Loved the baked goods she offered each morning. Would stay again.
on October 03, 2012
Great location, Cape Cod farm house, comfortable, quiet; charming innkeeper with expert baking skills.
on March 16, 2012
What a find! Only place in Sonoma County that books for a single night -- great find since we live in bay area & were attending a wine country wedding.
on August 28, 2011
Great antique looking house with interesting furniture and painting by the owner. Lucille made a tasty breakfast of freshly baked pastries.
View More Reviews
Small town,
Wine country
San Francisco, CA (SFO) 60 miles
From Hwy 101 take 37 to Hwy 121. Bear right on 121. Turn left on 8th St. Cross over Napa Road and turn right on Denmark Street.
Amusement Park,
Antique Shopping,
Art Galleries,
Concert venue,
Gourmet/Fine Dining,
Historical interests,
Live Music,
Movie Theater,
National/State parks,
Nature area,
15 mn to Napa Valley,
Outlet Stores,
Performing Theater,
Picnic area,
Religious Services,
Fireplace, Piano, TV, Telephone, VCR
Bicycle rental, Bicycle riding trails, Bird watching, Boating, Canoeing, Golfing, Hiking trails, Horseback riding, Kayaking, Mountain Bike trails, Sailing, Shopping, Sight-seeing, Swimming, Swimming pool, Tennis courts
Check Dates
Lucille Rella-Clohessy
English, German
Artist, photographer, lived in Europe for 10 years, travelled to Asia many times. Enjoy skiing, wind sailing, backpacking and meeting new people.
$90 per night
Enter your travel dates
Innkeepers: Lucille Rella-Clohessy
Guests: 2
Simple! BedandBreakfast.com guarantees you're getting the best price on your bed and breakfast stay when you book your room on BedandBreakfast.com. Within 24 hours of your booking, if you find a better price for the same room on the same date(s) at the same B&B, BedandBreakfast.com will refund the difference to you!
The Best Price Guarantee covers any Bed and Breakfast booked online through BedandBreafkast.com. The Bed and Breakfast must be booked using the Book It button on BedandBreakfast.com in order to qualify.
Just call 1-512-322-2710 within 24 hours of booking and give our customer support representative the details on the lower price you found.
The BedandBreakfast.com Best Price Guarantee ensures that when you book and pay for your reservation online through the BedandBreafkast.com website, you are booking your room at the lowest available price. If you find a lower price for the same room on the same dates at the same B&B, and contact us within 24 hour of making your booking, we will refund the difference in price to you. This guarantee does not apply to bookings made over holidays and/or for special events. All requests are subject to verification by BedandBreakfast.com. BedandBreafkast.com will not accept screenshots or other purported evidence of a lower price that cannot be independently confirmed by BedandBreakfast.com personnel. Nor will BedandBreakfast.com verify any requests that it believes, in its sole discretion, is the result of a printing or other error or is made fraudulently or in bad faith.


text:
Club News
Newsletter 6th Oct
U10's in Slieverue tournament U9's at Kilmacow's tournamentIntermediate ChampionshipThe Intermediates had 2 championship games over the last few weeksThe 1st round was against The ...
Posted 6 Oct 2015, 14:41 by Tullogher Rosbercon
Newsletter 17th September
All-Ireland Senior FinalCongratulations to Walter Walsh on winning his 3rd All-Ireland Senior Medal last week in Croke Park. Playing a huge part in Kilkenny's victory with ...
Posted 17 Sep 2015, 14:09 by Tullogher Rosbercon
Newsletter - 31st August
Senior FinalBest of Luck to Walter Walsh in the All-Ireland Final next Sunday. Its Walters 3rd Senior final so hopefully it'll be another success for Kilkenny and ...
Posted 31 Aug 2015, 14:47 by Tullogher Rosbercon
Newsletter 12th Aug
U10's after winning the Martin Codd Cup in RathnureAll-Ireland Semi FinalsWell done to Walter Walsh and Conor Hennessey who both had huge games for the Senior ...
Posted 12 Aug 2015, 15:06 by Tullogher Rosbercon
Newsletter 28th July
U10's being presented cup by TJ Reid after win in blitz in ballyhaleJunior A's Tullogher defeated The Rower-Inistioge in a hard fought encounter on a windy ...
Posted 28 Jul 2015, 14:36 by Tullogher Rosbercon
Development Draw
September 2015
Posted 7 Oct 2015, 05:35 by Tullogher Rosbercon
Winners of September 2015 Draw (held in the Clubhouse on 6th October 2015)
1st Prize of €500 Jimmy Heffernan
2nd Prize of €200 Martin Roche
3rd – 6th Prize of €100
3 Rob Martin
4 Seamus Phelan
5 Ollie and Mary Ryan
6 John Grennan
7th – 10th Prize of €50
7 John Healy
8 Helen Ryan
9 Kieran Barron
10 Pierse Handrick
Winners of Annual Draw 2015 Draw
1st Prize of €240 - Mary Barron
2nd Prize of €100 - Brian Dunlop
3rd Prize of €50 - James Gaule
August 2015
Posted 26 Aug 2015, 00:45 by Tullogher Rosbercon
Winners of August 2015 Draw (held in the Clubhouse on 25th August 2015)
1st Prize of €500 Tommy and Anna Grennan
2nd Prize of €200 Matt O'Brien
3rd – 6th Prize of €100
3 John and Georgina Murphy, Brownsford
4 Michael and Canice Grennan
5 Jason Murphy
6 Paddy Breen
7th – 10th Prize of €50
7 Derek Furness
8 Gerry and Bernie Arthurs, Glenpipe
9 Martin Ryan, Listerlin
10 Patrick Grennan, Ballycurran


text:
Being a patient partner means collaborating for better care
By Anna Deadrick
This column is the third in our series "The Patient's Voice," developed with the National Partnership for Women & Families.
I am a patient partner with the primary care division of Essentia Health, a large, multifaceted integrated health system in Minnesota. On matters small and large, from the "talent" chosen to record voice mail greetings played when the practice is closed to the systems used to manage the intensive care waiting room, my opinion matters.
Until recently, I lived in Ely, Minn., a town of approximately 4,500 people. My clinicians there were in a practice that embodied patient- and family-centered care. The doctors and the staff took time to know me. They were attentive when my husband had gastrointestinal problems and my children came in for checkups. They understood our preferences and gave us space to make the choices that were best for us.
Most of all, they supported me when I needed them most, after my breast cancer diagnosis in 2006. My doctors were true partners in my decision to have a mastectomy, reconstructive surgery, and follow-up treatment. I always felt like I was their most important patient, and I always had the information I needed to make the best choices for myself and my family. I never felt as if I were alone in navigating the health care system.
If I hadn't already realized how lucky I was, it would have become clear when I met patients from other practices who were having very different experiences. They didn't have medical teams who connected with them, listened to them, or made them feel special. They suffered as a result. I often wished they could rely on the kind of practice that was helping me.
Several years later, the clinic administrator of my primary care facility told me that Essentia Health was looking for patient partners to help more of their practices implement patient- and family-centered principles. I was asked to join Essentia's Patient and Family Primary Care Advisory Council, and because of my earlier experiences and my interest in this area, it was an easy decision to say yes.
Essentia's patient partners offer advice on all aspects of the health care system. We've shared our views about how Essentia's website is structured and how to make the information there clearer and more accessible. We've told the doctors that we want to be roomed differently; we want to talk to them in our clothing, like regular people, when we arrive and before we change into our gowns for an exam.
We've told them that little things, like giving patients warm blankets when they are sitting in wheelchairs waiting for tests, can make a big difference. We've helped doctors find better ways to manage the patient questions they find stacked up at the end of each day by explaining that, in most cases, we are happy with e-mailed responses and fully understand they don't have time to call every patient.
Perhaps most significantly, I was on an adjunct patient- and family-centered care committee that helped Essentia improve management of visitors to its intensive care units. We told the staff that already-stressed family members were feeling very unwelcome, almost like prisoners, because they were penned off in the waiting room, forced to press a buzzer and then wait to be admitted to visit a loved one.
We said, "You can trust us to be quiet and appropriate when we get into the ICU." In fact, that has proven to be the case. Essentia took our advice, giving families 24-hour access to ICU patients. Staff even put up signage that was more welcoming and less stern. Essentia also took the advice of the pediatric patient advisory council and allowed family members to stay with their children as the children were anesthetized, which calms both children and families, in turn improving outcomes.
These are commonsense measures to people like me, but Essentia didn't think to implement them systemwide until patients were given the chance to weigh in. It's fair to say that we've all learned as we've gone through this process. Essentia has learned to respect patient partners' time, to involve us in all levels of decision making, and to be less afraid that we will make unreasonable demands. The patient partners have learned how to be constructive and contribute effectively.
Being a patient partner lets me make a contribution to improving care. The process is respectful, and my input is considered. On the subject of health care, I have always had a lot to say. I am very glad that now someone is listening.
Anna Deadrick lives in Grand Rapids, Minn., and is a patient partner with Essentia Health.
View the printer-friendly version.
This is a printer-friendly version of this page
Internist Archives Quick Links
Fenway Guide to Lesbian, Gay, Bisexual, and Transgender Health, 2nd Edition
This new edition reflects recent clinical and social changes and continues to present the important issues facing practitioners and their LGBT patients. Read more about the Guide. Also see ACPs recent policy position paper on LGBT health disparities.
Join Us in Washington, DC for the Most Comprehensive Meeting in Internal Medicine
Register now and enjoy:
Discounted rates, the best national faculty, a wealth of clinical and practice management topics and hands-on sessions! Learn more about the meeting.