{'fraction_non_alphanumeric': 0.021834061135371178, 'fraction_numerical': 0.020087336244541485, 'mean_word_length': 4.96875, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 1, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Development Of pCO2 Analytic System Using Soft-Lithography Method

This study carried out development of non-invasive optical transcutaneous partial pressure of carbon dioxide, pCO2, monitoring system by non-dispersive infrared method, NDIR. The purpose of this system is to detect CO2 from outer skin, not from the arterial blood-gathering method. There are advantages about a reduction of analysis time and real-time monitoring that this system might be available. The measurement system is composed of the IR lamp, the pyroelectric sensor including a 4.26 mum optical filter, the optical gas reaction chamber and the signal processing circuit. The optical reaction length of chamber was reduced by 1 mm using the soft-lithography method which CO2 is exhausted in human body as we considered. The fabricated pCO2 monitoring system showed linear result of 6.50times 10-6 absorbance/ppm sensitivity for CO2 concentration from 0 ~ 5,000 ppm by MFC and about 2 seconds of fast response time. The proposed system can be used in the optical biosensor field for the medical diagnosis such as pCO2 monitoring system and environment monitoring systems.