{'fraction_non_alphanumeric': 0.060043668122270744, 'fraction_numerical': 0.026746724890829694, 'mean_word_length': 4.622699386503068, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
CAD/CAM post-and-core using different esthetic materials: Fracture resistance and bond strengths.

PURPOSE
To evaluate the fracture resistance (FR), and push-out bond strengths (BS) of custom-made CAD/CAM post-and-cores manufactured with different esthetic materials.


METHODS
90 single-rooted extracted teeth were selected, endodontically treated and prepared to receive the posts. The specimens were randomly divided into three equal groups according to the material: hybrid ceramic Vita Enamic (HC); nano-ceramic resin composite Lava Ultimate (RC); and experimental epoxy-resin reinforced by glass-fiber (FG). The post-and-cores were manufactured using CAD/CAM and cemented using a self-adhesive resin cement (Rely X Unicem2). A subgroup of 30 specimens (n=10) was subjected to fatigue (1,000,000 cycles at 5 Hz) and then to the FR test. Another subgroup with 60 specimens was submitted to the BS test, with and without fatigue. Data were submitted to analysis of variance (FR - one-way ANOVA; BS - two-way ANOVA) and Tukey's test (α= 0.05).


RESULTS
The FR values (mean±SD, in Newtons) were 414.5±83.9 (HC), 621.3±100.3 (RC), and 407.6±109.0 (FG), with RC showing significantly higher FR values (P< 0.05). For BS, there was no statistically significant difference among the materials, with and without fatigue (P> 0.05). The type of material used to obtain the CAD/CAM-customized post-and-cores had a significant effect on the FR, but not on the BS of the specimens. Fatigue did not influence the BS for the tested materials.


CLINICAL SIGNIFICANCE
CAD/CAM custom-made esthetic post-and-cores showed good performance relative to fracture resistance and bond strength to root canal dentin walls. The CAD/CAM materials tested could be used as an alternative to restore wide flared root canals in esthetically compromised teeth.