{'fraction_non_alphanumeric': 0.032101167315175094, 'fraction_numerical': 0.0019455252918287938, 'mean_word_length': 5.43125, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A microchip microcontroller-based transducer controller for non-contact scanning probe microscopy with phase-locked loop, amplitude, and Q control.

An inexpensive yet versatile transducer controller for non-contact scanning probe microscopy (SPM) based on a PIC32 microcontroller from Microchip Technology, Inc is described. In addition to feedback control using the amplitude or phase of the signal from the non-contact transducer, the controller includes a phase-locked loop for frequency-shift feedback, as well as fixed-amplitude, quality factor (Q) control, and self-excitation modes. Apart from the input amplifiers, output buffers, and the Q-control circuit, all other functions of the controller are instantiated in software on the microchip, enabling rapid changes in operating parameters if needed. The controller communicates with a host personal computer via a simple serial connection. The controller has been tested with a quartz tuning-fork transducer but can be used with any oscillating non-contact transducer.