{'fraction_non_alphanumeric': 0.027216174183514776, 'fraction_numerical': 0.0023328149300155523, 'mean_word_length': 6.071428571428571, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Raman Distributed Temperature Sensing at CERN

A field trial has been performed at the CERN high-energy accelerator-mixed (CHARM) field facility, newly developed for testing devices within accelerator representative radiation environments, to validate the use of Raman-based optical fiber sensors for distributed temperature measurements in highly radiative environments. Experimental results demonstrate that Raman distributed temperature sensors, operating in loop configuration on radiation-tolerant optical fibers, are able to compensate for wavelength-dependent losses and are, therefore, robust to harsh environments, in which a mixed-field radiation, including protons, neutrons, photons, and other particles, is potentially altering the fiber material properties. The temperature profile measured on commercial radiation-tolerant optical fibers shows that a temperature resolution <;1 °C, 0.5-m spatial resolution, is highly reliable and sets the first step toward a distributed measurement system able to monitor the temperature at the CERN's large hadron collider for safety purposes. Such a system will also be helpful in correcting the radiation-induced attenuation temperature dependence in distributed radiation sensing systems based on radiation-sensitive optical fibers.