{'fraction_non_alphanumeric': 0.06927551560021153, 'fraction_numerical': 0.017979904812268643, 'mean_word_length': 5.183006535947713, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Molecular dynamic-secondary ion mass spectrometry (D-SIMS) ionized by co-sputtering with C60+ and Ar+.

Dynamic secondary ion mass spectrometry (D-SIMS) analysis of poly(ethylene terephthalate) (PET) and poly(methyl methacrylate) (PMMA) was conducted using a quadrupole mass analyzer with various combinations of continuous C(60)(+) and Ar(+) ion sputtering. Individually, the Ar(+) beam failed to generate fragments above m/z 200, and the C(60)(+) beam generated molecular fragments of m/z ~1000. By combining the two beams, the auxiliary Ar(+) beam, which is proposed to suppress carbon deposition due to C(60)(+) bombardment and/or remove graphitized polymer, the sputtering range of the C(60)(+) beam is extended. Another advantage of this technique is that the high sputtering rate and associated high molecular ion intensity of the C(60)(+) beam generate adequate high-mass fragments that mask the damage from the Ar(+) beam. As a result, fragments at m/z ~900 can be clearly observed. As a depth-profiling tool, the single C(60)(+) beam cannot reach a steady state for either PET or PMMA at high ion fluence, and the intensity of the molecular fragments produced by the beam decreases with increasing C(60)(+) fluence. As a result, the single C(60)(+) beam is suitable for profiling surface layers with limited thickness. With C(60)(+)-Ar(+) co-sputtering, although the initial drop in intensity is more significant than with single C(60)(+) ionization because of the damage introduced by the auxiliary Ar(+), the intensity levels indicate that a more steady-state process can be achieved. In addition, the secondary ion intensity at high fluence is higher with co-sputtering. As a result, the sputtered depth is enhanced with co-sputtering and the technique is suitable for profiling thick layers. Furthermore, co-sputtering yields a smoother surface than single C(60)(+) sputtering.