{'fraction_non_alphanumeric': 0.05054566341183228, 'fraction_numerical': 0.04135554279149914, 'mean_word_length': 4.730263157894737, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Clinical utility of a type 4 portable device for in-home screening of sleep disordered breathing.

BACKGROUND
Portable monitoring devices have been developed for in-home screening and to aid in the diagnosis of sleep disordered breathing (SDB) while increasing accessibility and reducing costs. Although there are many different devices available in the market, most have not undergone rigorous validation. Therefore, although such devices are promising, more research on their clinical utility is necessary. The purpose of this study was to assess the clinical utility of a type 4 home sleep apnea test (HSAT) as an in-home screening for SDB.


METHODS
We investigated consecutive subjects who underwent in-laboratory overnight polysomnography following in-home screening using HSAT. We evaluated the correlation between apnea-hypopnea index (AHI) by in-laboratory overnight polysomnography and by HSAT and evaluated the sensitivity and specificity for AHI ≥5 and AHI ≥30 by the receiver operating characteristic (ROC) analysis.


RESULTS
Finally, data of 387 participants (86.8% men, mean age 55.3±13.3 years and body mass index 25.1±4.1 kg/m2) were assessed. In all patients, AHI by HSAT correlated significantly with AHI by polysomnography (r=0.670, P<0.001). The area under curves of ROC for AHI ≥5 and AHI ≥30 were 0.854±0.029 and 0.841±0.022, respectively. The best cut-off of AHI by HSAT for detecting AHI by polysomnography ≥5 was 10.3 events/h (sensitivity, 82.8%; and specificity, 76.0%), and AHI by HSAT for detecting AHI by polysomnography ≥30 was 24.5 events/h (sensitivity, 75.8%; and specificity, 80.4%).


CONCLUSIONS
This type 4 HSAT may have potential as a screening tool for SDB and thus have sufficient clinical utility.