{'fraction_non_alphanumeric': 0.02288329519450801, 'fraction_numerical': 0.011441647597254004, 'mean_word_length': 4.7631578947368425, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Response times on urethral pressure profiles: carbon dioxide instrument.

This study shows that the response time between an actual event and a recorded event on a urethral pressure profile is a function of gas flow rate and gas circuit volume. Response times varied from 1.8 to 10.8 sec. Considerable variation was noted between flow setting and actual gas flow; therefore, calibration of the instrument's gas flow meter is recommended.