{'fraction_non_alphanumeric': 0.03614457831325301, 'fraction_numerical': 0.022375215146299483, 'mean_word_length': 4.542857142857143, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A Porphyrin Based Potentiometric Sensor for Zn2+ Determination

M) with a Nernstian slope (30.0 mV/decade of activity). The useful pH range of the sensor is 3.0 –7.4, beyond which a drift in potential was observed. The response time of the sensor is 10s and the lifetime was about 2 months during which it could be used without any measurable divergence. It had good stability and reproducibility. The membrane worked satisfactorily in non-aqueous medium up to 40% (v/v) non-aqueous content. The selectivity coefficient values indicate that the electrode is highly selective for Zn