{'fraction_non_alphanumeric': 0.051319648093841645, 'fraction_numerical': 0.016617790811339198, 'mean_word_length': 5.376947040498442, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Characterization of batch-microfabricated scanning electrochemical-atomic force microscopy probes.

A procedure for the batch microfabrication of scanning electrochemical-atomic force microscopy (SECM-AFM) probes is described. The process yields sharp AFM tips, incorporating a triangular-shaped electrode (base width 1 microm, height 0.65 microm) at the apex. Microfabrication was typically carried out on (1)/(4) 3-in. wafers, yielding 60 probes in each run. The measured spring constant of the probes was in the range 1-1.5 N m(-1). To date, processing has been carried out twice successfully, with an estimated success rate for the fabrication process in excess of 80%, based on field emission-scanning electron microscopy imaging of all probes and current-voltage measurements on a random selection of approximately 30 probes. Steady-state voltammetric measurements for the reduction of Ru(NH(3))(6)(3+) in aqueous solution indicate that the electrode response is well-defined, reproducible, and quantitative, based on a comparison of the experimental diffusion-limited current with finite element simulations of the corresponding mass transport (diffusion) problem. Topographical imaging of a sputtered Au film with the SECM-AFM probes demonstrates lateral resolution comparable to that of conventional Si(3)N(4) AFM probes. Combined electrochemical-topographical imaging studies have been carried out on two model substrates: a 10-microm-diameter disk ultramicroelectrode (UME) and an array of 1-microm-diameter UMEs, spaced 12.5 microm apart (center to center). In both cases, an SECM-AFM probe was first employed to image the topography of the substrates. The tip was then moved back a defined distance from the surface and use to detect Ru(NH(3))(6)(2+) produced at the substrate, biased at a potential to reduce Ru(NH(3))(6)(3+), present in bulk solution, at a diffusion-controlled rate (substrate generation-tip collection mode). These studies establish the success of the batch process for the mass microfabrication of SECM-AFM tips.