{'fraction_non_alphanumeric': 0.027105517909002903, 'fraction_numerical': 0.00484027105517909, 'mean_word_length': 4.712707182320442, 'pattern_counts': {'":': 1, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Newly developed hardness testing system, "Cariotester": measurement principles and development of a program for measuring Knoop hardness of carious dentin.

We previously discovered that when a cone-shaped indenter coated with paint was pressed into an object, the paint disappeared in accordance with the depth of the indentation. Based on this fact, we developed the Cariotester, a portable system for measuring the Knoop hardness (KHN) of carious dentin. The Cariotester is composed of a handpiece with an indenter, a microscope, and a computer. In this system, the painted indenter is forced into the material with a 150-gf load, and the indentation depth (CT depth) is obtained from the paint disappearance. The CT depth by the Cariotester and the KHN by a microhardness tester were determined at 14 dentin regions. From the data, a program was created to convert the CT depth of the carious dentin into the KHN. As a result, if the CT depth is measured with this system, the KHN of carious dentin can be displayed in real time.