{'fraction_non_alphanumeric': 0.03470540758676352, 'fraction_numerical': 0.0008071025020177562, 'mean_word_length': 5.391752577319588, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Fabrication of Micro-Patterned Chip with Controlled Thickness for High-Throughput Cryogenic Electron Microscopy.

A major limitation for the efficient and high-throughput structure analysis of biomolecules using cryogenic electron microscopy (cryo-EM) is the difficulty of preparing cryo-EM samples with controlled ice thickness at the nanoscale. The silicon (Si)-based chip, which has a regular array of micro-holes with graphene oxide (GO) window patterned on a thickness-controlled silicon nitride (SixNy) film, has been developed by applying microelectromechanical system (MEMS) techniques. UV photolithography, chemical vapor deposition, wet and dry etching of the thin film, and drop-casting of 2D nanosheet materials were used for mass-production of the micro-patterned chips with GO windows. The depth of the micro-holes is regulated to control the ice thickness on-demand, depending on the size of the specimen for cryo-EM analysis. The favorable affinity of GO toward biomolecules concentrates the biomolecules of interest within the micro-hole during cryo-EM sample preparation. The micro-patterned chip with GO windows enables high-throughput cryo-EM imaging of various biological molecules, as well as inorganic nanomaterials.