{'fraction_non_alphanumeric': 0.03686366296079579, 'fraction_numerical': 0.018139262726740785, 'mean_word_length': 5.042402826855124, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Evaluation of Mutagenicity of Restorative Dental Materials Using the Ames Salmonella/Microsome Test

Compounds of five commercially available dental material kits were examined for mutagenic potential by use of the Ames Salmonella/microsome test. Dimethyl sulfoxide (DMSO) was used as the solvent for all materials, except MONO-LOK2 Primer and Right-On Activator, which were dissolved in 95% ethanol. The Tenure, Fuji, and Gluma materials were initially screened at 2 mg (solids) or 2 μL (liquids) per plate, without S9 (Aroclor 1254-induced rat liver microsomes), and dose-response studies were conducted for those materials showing potential mutagenicity. The components of three dental kits, MONO-LOK2, Right-On, and Gluma, were further examined, both with and without S9 activation, at doses of 0.2, 4, 20, 100, 500, and 2500 μg (nL for liquid components) per plate. The data showed no mutagenic potential for the components of Fuji Glass Ionomer (Type III), Tenure Dentin Bonding System, and MONO-LOK2 kits, or Right-On Paste. However, Gluma 3, a component of the Gluma/Lumifor Bonding System, exhibited mutagenic activity in a dose-response manner. The mutagenicity of Gluma 3 was demonstrated in repeated experiments. Gluma 4 and the resin of the Gluma dental kit, as well as Right-On Activator, caused a slight increase in the number of revertants, and judgment regarding their mutagenicity could not be made conclusively. The results of this study indicate that some commercial dental materials may not have been adequately tested for mutagenicity, and that a reliable test program for evaluation of mutagenicity of dental materials is needed so that their safe use in dental practice can be ensured.