{'fraction_non_alphanumeric': 0.024115755627009645, 'fraction_numerical': 0.02572347266881029, 'mean_word_length': 4.822429906542056, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Light-irradiation at 700 MPa down to 1.5 K for neutron diffraction

We present a new continuously-loaded high-pressure cell for neutron diffraction made from TiZr ‘null-matrix’ alloy that combines high mechanical resistance below 100 °C and negligible coherent neutron scattering. This cell operates at a maximum pressure of 700 MPa down to a temperature of 1.5 K. A sapphire optical window allows simultaneous illumination of the sample over the broad wavelength range ~0.4 to almost 5 μm. The pressure is applied with Fluorinert or helium gas to ensure the best possible hydrostatic conditions at cryogenic temperatures.