{'fraction_non_alphanumeric': 0.03776873910517141, 'fraction_numerical': 0.041836141778036025, 'mean_word_length': 5.0, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Preparation technique of S2OF10 gas standard sample and determination method of the trace S2OF10 in SF6].

In this paper a series of methods and techniques for the S2OF10 standard sample preparation and quantitative determination are presented. They are, the preparation of S2OF10 by preparative chromatography with adsorption/thermal desorption, the standard sample of S2OF10 gas prepared by exponential dilution and the gas chromatography/flame photometric detector(GC/FPD) determination of trace S2OF10 from an SF6 sample with quantitative calibration factor. Especially, the S2OF10 gas from a used SF6 sample was directly separated and concentrated through a U-adsorbent-tube packed with 300 mg of Porasil A in a cold trap (-63 degrees C) with liquid-nitrogen and chloroform. Then it was purified by preparative-GC and to be injected into a preparative system of standard gas sample. In the meantime, the S2OF10 gas obtained was confirmed by the methods of GC/FPD, infrared spectrophotometer(IR) and gas chromatography/mass spectrometer(GC/MS) separately. The sub-ppm(by volume) level of the S2OF10 and SF6 mixture samples were prepared by use of the exponential dilution system. The GC/FPD experimental results showed that the detection linear range of S2OF10 gas concentration was 0.80 x 10(-6)-2.60 x 10(-4) (volume fraction) and the quantitative calibration factor of the S2OF10 was 0.197 based on SF6. The determination errors of quantitative calibration factor were 1.8%-20% and S2OF10 recovery of the adsorption/thermal desorption was 98.2% (n = 9) and its relative standard deviation was 6.2%. In addition, the results also showed that it is a simple and rapid method with good linearity and reproducibility.