{'fraction_non_alphanumeric': 0.03104575163398693, 'fraction_numerical': 0.0, 'mean_word_length': 5.191919191919192, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Analysis of P/N leak subtraction of pClamp acquisition software].

Using blind whole cell recording technique on rat hippocampal slices, the function and mechanisms of P/N leak subtraction of Clampex program in pClamp acquisition software were studied. The ways of picking up sub-pulse current and properly selecting parameters, including time course, polarization, sub-holding potential, numbers and location of P/N sub-pulse were analyzed in detail. Our results indicate that voltage-gated ion channel currents recorded by Clampex will be more accurate and reproducible by properly using P/N leak subtraction.