{'fraction_non_alphanumeric': 0.03481234293247641, 'fraction_numerical': 0.028797429242368063, 'mean_word_length': 5.200255427841634, 'pattern_counts': {'":': 0, '<': 4, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Gene expression profiling in uveal melanoma: technical reliability and correlation of molecular class with pathologic characteristics

Background A 15-gene expression profile test has been clinically validated and is widely utilized in newly diagnosed uveal melanoma (UM) patients to assess metastatic potential of the tumor. As most patients are treated with eye-sparing radiotherapy, there is limited tumor tissue available for testing, and technical reliability and success of prognostic testing are critical. This study assessed the analytical performance of the 15-gene expression test for UM and the correlation of molecular class with pathologic characteristics. Methods Inter-assay, intra-assay, inter-instrument/operator, and inter-site experiments were conducted, and concordance of the 15-gene expression profile test results and associated discriminant scores for matched tumor samples were evaluated. Technical success was determined from de-identified clinical reports from January 2010 - May 2016. Pathologic characteristics of enucleated tumors were correlated with molecular class results. Results Inter-assay concordance on 16 samples run on 3 consecutive days was 100%, and matched discriminant scores were strongly correlated (R2 = 0.9944). Inter-assay concordance of 46 samples assayed within a one year period was 100%, with an R2 value of 0.9747 for the discriminant scores. Intra-assay concordance of 12 samples run concurrently in duplicates was 100%; discriminant score correlation yielded an R2 of 0.9934. Concordance between two sites assessing the same tumors was 100% with an R2 of 0.9818 between discriminant scores. Inter-operator/instrument concordance was 96% for Class 1/2 calls and 90% for Class 1A/1B calls, and the discriminant scores had a correlation R2 of 0.9636. Technical success was 96.3% on 5516 samples tested since 2010. Increased largest basal diameter and thickness were significantly associated with Class 1B and Class 2 vs. Class 1A signatures. Conclusions These results show that the 15-gene expression profile test for UM has robust, reproducible performance characteristics. The technical success rate during clinical testing remains as high as first reported during validation. As molecular testing becomes more prevalent for supporting precision medicine efforts, high technical success and reliability are key characteristics when testing such limited and precious samples. The performance of the 15-gene expression profile test in this study should provide confidence to physicians who use the test’s molecular classification to inform patient management decisions. Electronic supplementary material The online version of this article (doi:10.1186/s13000-017-0650-3) contains supplementary material, which is available to authorized users.


Background
Uveal melanoma (UM) is a rare, intraocular cancer that affects approximately 1600 patients per year in the United States [1]. Most patients are treated with eyesparing radiotherapy, primarily through plaque brachytherapy or proton beam radiation, while only a small proportion (~10%) will undergo enucleation. Despite the high rate of primary tumor control,~50% of patients will develop metastatic disease, primarily to the liver, after which prognosis is poor [2]. While the clinicopathologic features assessed in AJCC staging, including tumor size, ciliary body involvement, and extraocular extension, are important factors when assessing metastatic risk, even Stage I-II patients have a 12-30% UM-related mortality rate by 10 years [3,4]. Because of this, frequent surveillance was generally recommended to monitor all UM patients for disease spread [5,6].
Gene expression profiles that are associated with lowrisk (Class 1A), intermediate-risk (Class 1B), or highrisk (Class 2) outcomes have been shown to provide information useful for risk-tailored surveillance plans [7][8][9]. Because treatment of the primary tumor is highly effective with plaque radiotherapy, enucleation is not common, and the amount of tumor tissue available for molecular prognostic testing is limited. During the development of the gene expression profiling test for UM prognostication (also known as DecisionDx®-UM), focus was placed on providing patients with a test that was robust and could be run successfully and reproducibly on a very small amount of tissue obtained through a fine needle aspirate biopsy (FNAB), as well as formalin-fixed paraffin embedded (FFPE) tissue from enucleations [8][9][10].
Recommendations for the development of clinically useful molecular biomarkers have suggested that a biomarker test needs to demonstrate clinical validity, utility, and analytic validity, and that supporting data for each must be transparent and readily available for both physicians and patients [11]. Several widely-used genomic tests for different types of cancer have achieved high levels of evidence for each of these criteria [12][13][14][15][16][17]. The clinical validity and clinical utility of the 15-gene expression profile test for UM was reported by the Cooperative Ocular Oncology Group (COOG) [9], and the prognostic accuracy of the test has been confirmed in multiple single-and multicenter studies [18][19][20][21][22][23]. The test's impact on clinical decision-making for UM patients has also been demonstrated [23,24]. The focus of this study was to evaluate performance metrics of the test in a CLIA-certified laboratory setting, describe the rate of technical success on both FNAB and FFPE tissue from enucleations, and report correlations with pathological variables.

Tissue acquisition and processing
All samples were acquired during routine clinical testing for risk prognostication in UM patients. For FNAB samples, UM tumor aspirates were frozen in RNase-free RNA stabilization buffer by the treating physician immediately after biopsy and shipped to the Castle Biosciences' laboratory on dry ice (Fig. 1). All samples were immediately processed with the PicoPure RNA Isolation Kit (Molecular Devices, Sunnyvale, CA). For tumors removed through enucleation, five-micron sections from FFPE tissue were used; the first of six sequential sections were stained with hematoxylin and eosin. Tumor tissue containing at least 80% tumor nuclei density was marked and manually dissected from unstained slides using a sterile scalpel and processed for RNA isolation according to the Ambion RecoverAll Total Nucleic Acid Isolation Kit (Life Technologies Corporation, Grand Island, NY). Reverse transcription of RNA into cDNA was performed using the Applied Biosystems High Capacity cDNA Reverse Transcription Kit (Life Technologies Corporation).

DecisionDx-UM gene expression profiling and molecular classification
Gene expression profiling was performed as previously described [9,25]. From 2010 through 2014, quantitative PCR was performed on a 7900HT Fast Real-Time PCR System (Thermo Fisher Scientific), after which it was performed on the current system, a QuantStudio 12K Flex Real-Time PCR System (Applied Biosystems). Complementary cDNA was pre-amplified with 14 cycles and then diluted 20-fold with Tris-EDTA (TE). Fifty microliters of diluted, pre-amplified cDNA was mixed with 50 uL of 2X Taqman Gene Expression Master Mix and loaded in triplicate onto customized microfluidics PCR cards containing the gene-specific primers and probes for the 12 discriminating genes and endogenous control genes (Additional file 1: Table S1).
After PCR amplification, the average Ct values for each triplicate are calculated. The average Ct value for each of the 12 discriminating genes is subtracted from the geometric mean of the control genes to determine the ΔCt value for each gene. A support vector machine (SVM)-learning algorithm, trained on a training set of UM cases with known gene expression profiles and long-term outcomes, which has been locked since inception, is used to determine classification of test cases. The SVM algorithm places the UM samples within the training set into a hyperplane with n-dimensional space and maximizes the hyperplane between the Class 1 and Class 2 specimens. A discriminant score for each sample is generated, which reflects the inverse distance of the patient sample to the hyperplane separating Class 1 and 2 training set samples. The output of the algorithm is either a negative discriminant score (Class 1) or positive discriminant score (Class 2). The discriminant score can theoretically range from >0 to positive or negative infinity, however, only the absolute (positive) value of the discriminant score is reported clinically alongside the respective Class result. A discriminant score is considered reduced confidence if it is <0.1. For Class 1 cases, summation of the RAB31 and CDH1 ΔCt values determined whether the sample was Class 1A or Class 1B based on a mathematical threshold. Class 1B UMs have differential RAB31/CDH1 expression relative to Class 1A tumors.

Technical reliability studies
De-identified DecisionDx-UM test results from January 2010 through May 2016 were analyzed according to tissue type, successful molecular classification, and resultant class assignment. Institutional review board submission of the technical reliability study was not required as it constitutes technical data on file only, and contains no patient-specific protected health information. As such, this analysis is exempt from the regulatory review requirements as set forth in section 46.101 (b) of 45 CFR 46 [26]. Analytic validation experiments were performed on cDNA generated from the RNA extracted from UM tumor specimens. For these experiments, deidentified residual samples were used. Given that the majority of specimens are obtained via FNAB, most tumor samples are exhausted by performing the 15-gene expression profile test and residual samples are limited.
As such, samples used in reproducibility experiments were not consecutive samples, but in each experiment, leftover samples from approximately the same time period were used. Samples were between 1 and 10 weeks old (from date of tissue receipt) and remained frozen after initial processing until reliability experiments.

Inter-assay and intra-assay repeatability
To evaluate the repeatability of gene expression profiling and class assignment agreement on the same sample between separate PCR runs, DecisionDx-UM testing was performed on 16 samples on three consecutive days, resulting in 48 molecular classifications ( Table 1). The class assignment concordance for each sample across all three days was 100% (eight Class 1A, three Class 1B, and five Class 2 tumors). Multiple regression analysis of the discriminant scores from the three runs showed an average R 2 value of 0.9944 (Fig. 2a). The majority of variation in discriminant scores between runs fell within the 95% confidence interval (−0.1809 to 0.1219), with an estimated bias of −0.0295 (Fig. 2b).
In a second reproducibility study, matched duplicate runs of samples used for proficiency testing or instrument verification at six time points during a one-year period were analyzed for inter-assay concordance (Table 1). Overall, reproducibility was evaluated on 46 samples run in duplicate during this one-year interval, and the discriminant scores and molecular classifications were compared for the matched samples. The molecular classifications Fig. 1 Overview of DecisionDx-UM testing and analytic validation. Fine-needle aspirate biopsies are collected prior to plaque or proton beam clip placement, snap-frozen in RNA stabilization buffer, and shipped frozen. Alternatively, six 5-μm section slides containing tumor tissue from enucleations can be used. RNA is extracted and reverse transcribed into cDNA. The cDNA is pre-amplified followed by qPCR for 12 discriminating genes and 3 control genes. A support vector machine (SVM) algorithm assigns a Class 1 or 2 call comparing the patient sample to a locked-down training set. The summation of CDH1 and RAB31 is used to determine Class 1A or 1B subclassification were 100% concordant for all samples (7 Class 1A, 15 Class 1B, and 24 Class 2 tumors). Regression analysis showed an R 2 of 0.9747 (Fig. 2c). The estimated bias in discriminant scores was 0.0049 and most variation in discriminant scores fell within the 95% confidence interval (−0.2816 to 0.2914) (Fig. 2d).
Intra-assay repeatability was tested by running two replicates of four samples on the same PCR card, and repeating this experiment on a second card. This was performed with three sets of four samples, resulting in 24 paired molecular classifications from 12 tumor specimens. Class assignment agreement was 100% for the four samples within each card and between the two cards (Table 1; five Class 1A, two Class 1B and five Class 2 tumors). The R 2 value from correlation analysis was 0.9934 (Fig. 2e). As shown on a Bland-Altman plot (Fig.  2f ), all (24/24) of the paired intra-assay discriminant scores had differences that fell within the 95% confidence interval (−0.1182 to 0.1472) and the estimated bias was 0.0145.
Transfer of the test to a new laboratory location afforded the opportunity to test the inter-assay reliability of the test when performed in different laboratories. Twenty-nine samples were run at our CLIA-certified contract laboratory (St. Joseph's Hospital and Medical Center, Phoenix AZ) and a new Castle Biosciences, Inc. facility ( Table 1). The DecisionDx-UM Class calls were 100% concordant between the two laboratories. The R 2 value of the correlation between discriminant scores was 0.9818 (p < 0.001) (Fig. 3a). The estimated bias was 0.0233 and 26/29 (90%) of the discriminant scores had an inter-lab difference that fell within the 95% confidence interval (−0.1939 to 0.2405; Fig. 3b).
Twenty-eight samples were run on two instruments by separate operators to analyze inter-instrument and -operator reliability (Table 1). Two Class 1 calls were discordant for their 1A versus 1B subclassification. One discordant sample had a Class 1B result because the RAB31/CDH1 expression fell just under the subclassification threshold, and its variability fell within the 95% confidence interval for RAB31/CDH1 in reliability experiments. The remaining discordant case had a difference in RAB31/CDH1 between runs that fell outside of this confidence interval and was thus an outlier. Class 1 vs. Class 2 calls were 96% concordant (27/28). The case that was discordant for Class 1 vs. Class 2 did not have any unique characteristic that could potentially explain the discrepancy and thus seems to represent an outlier. As shown in Fig. 3c, correlation analysis of the discriminant scores between the two instruments generated an R 2 value of 0.9636 (p < 0.001). The estimated bias was −0.0540 (95% CI = −0.3266 to 0.2186) and only 1 sample fell outside of this confidence interval (Fig. 3d).
Failure to generate a successful gene expression profile report can be due to i) the failure of multiple control and/or discriminating genes to amplify (multi-gene failure [MGF]), or ii) insufficient tumor cell density (<80%) within a manually dissectable area of FFPE tissue as determined by a pathologist, which results in rejection of the tumor sample prior to testing. Overall, 180 (3.3%) of the samples tested resulted in MGFs and 21 (0.4%) samples had insufficient tumor density (Table 2, Fig. 4c). Eighteen of the MGFs (10%) were related to out-ofspecification FNAB specimens, meaning the FNABs

Discriminant scores and confidence intervals
The discriminant score associated with the DecisionDx-UM Class reflects the inverse distance of the patient a b c d e f The average discriminant score for Class 2 was 0.816144 (95% CI = 0.0528 to 1.567). A score between −0.1 to 0.1 is considered of reduced confidence and is reported clinically as such. However, a discriminant score with reduced confidence has not been associated with a different outcome than a normal confidence score [25], and the vast majority (~97%) of the discriminant scores reported clinically have been reported within the normal confidence range.

Pathologic characteristics of DecisionDx-UM-tested tumors
Associated de-identified pathological data was available for 527 FFPE tumors from enucleations tested between January 2010 and May 2016. The pathologic characteristics of these tumors are described in Table 3, with statistical analysis of correlation with molecular class presented in Table 4

Discussion
Given the advances in molecular cancer diagnostics in the last decade, recommendations have been established to guide careful, thorough assessments of these tests [11,27]. One of the major criteria for a clinically A B C D Fig. 3 Inter-lab and inter-instrument/operator reliability of DecisionDx-UM. a Correlation of 29 paired discriminant scores from samples run at two CLIA-certified laboratories. b Bland-Altman plot of the estimated bias (average difference between discriminant scores, red line) between labs and the 95% confidence interval (dashed lines). c Correlation analysis of 28 paired inter-instrument/operator discriminant values. d Bland-Altman plot paired inter-instrument/operator discriminant values useful, well-validated assay is that the test must demonstrate analytic validity, which includes reliability and reproducibility of the test to measure the intended molecular analytes [11,27]. In this study, we report on the analytic validity of the 15-gene expression profile test (DecisionDx-UM) as performed in a CLIA-certified clinical laboratory. Reproducibility and reliability were demonstrated through i) inter-assay concordance, ii) inter-assay concordance of samples used for proficiency testing and instrument verification throughout one year, iii) intra-assay concordance, iv) inter-lab concordance between two CLIA-certified laboratories, and v) interinstrument/operator concordance. Molecular classifications by DecisionDx-UM were 100% concordant for four of these assessments, with only 3 discordant specimens out of the total 143 samples in the analytic experiments. The reported discriminant scores for the same tumor sample were highly correlated, as reflected by R 2 values  above 0.96 for the inter-assay, intra-assay, interinstrument, and inter-lab experiments. Furthermore, the run-to-run, lab-to-lab, instrument-instrument, and intrarun variabilities between discriminant scores were all within acceptable limits that would not impact patient care. These results demonstrate that the DecisionDx-UM class assignment for the same tumor specimen is consistent within the same PCR card and when run on separate days in different PCR cards, even with several weeks in between runs. To our knowledge, this is the only report of  analytic validity for any prognostic test for UM. Comparable analytic validation has not yet been reported for fluorescence in situ hybridization (FISH), microsatellite analysis (MSA), single nucleotide polymorphism (SNP) arrays, or multiplex-ligation probe amplification (MLPA), all of which can be used to detect monosomy 3 and other chromosomal copy number changes that have been associated with UM metastasis, or next-generation sequencing to detect mutations and chromosomal aberrations to estimate UM prognosis. Gene expression profiling is one of the clinically significant variables for disease prognostication recommended by the 2017 AJCC staging guidelines [28], and as such, the DecisionDx-UM test is routinely used across the United States and has been clinically available since 2010. The prognostic test was developed after two distinct molecular subtypes of UM were identified and shown to have correlation with outcomes [7,10,29,30]. A 15-gene RT-PCR test was developed to identify these UM molecular subtypes and prospectively validated in multiple studies [8,9,20,21,23]. As eye-sparing treatments are frequently utilized in the contemporary management of UM and enucleations are less common, the standard practice for physicians utilizing DecisionDx-UM is to collect a biopsy prior to or at the time of radioactive plaque or proton beam clip placement. Due to the lack of residual tissue aside from that obtained during biopsy, successful gene expression profiling on a single biopsy is critical. While the technical success of the 15gene expression profile assay has been previously reported as ranging from 95 to 99% [8,9,19], many of the samples in those studies were tested in the research laboratory that developed the assay, which was then licensed by Castle Biosciences, Inc. in 2009. In this report, we demonstrate that 96.4% of 5516 samples that have been clinically tested at Castle Biosciences' laboratory generated successful molecular classification reports, establishing the consistency of technical success from the test. Of the unsuccessful tests, 39 out of 201 (19%) were due to samples outside of quality control specifications (for FNABs, incorrect volume and/or not frozen, n = 18; for FFPE tissues, insufficient tumor volume, n = 21). Thawed samples and those with insufficient/excessive volume have been previously reported to be associated with technical failures of gene expression profiling [8]. The technical success rate of DecisionDx-UM testing is substantially higher than what has been previously reported for other molecular methods used in UM prognostication, including FISH, MSA, and array CGH, which range from 50 to 87% in much smaller sample sets [31][32][33][34].
Of the 5315 successful tests reported, 43.4% were Class 1A, 22.4% were Class 1B, and 34.2% were Class 2. Overall, these proportions are similar to class those reported in the Cooperative Ocular Oncology Group (COOG) study, which were 47% Class 1A, 13%, Class 1B and 40% Class 2 [9]. Likewise, other single-and multicenter studies that analyzed subsets of the clinically tested patients reported here have shown similar proportions of Class 1A, 1B, and 2 results [21,23,35].
Of the enucleation specimens tested for which there were associated pathology data, the proportion of Class 1 patients shifted to be predominately Class 1B vs. Class 1A, and the proportion of Class 2 tumors was also increased compared to the total clinical population. Given the significantly increased LBDs and thicknesses of enucleation specimens (i.e. those that necessitated removal of the globe), an increase in Class 2 and Class 1B tumor classification is not unexpected. Importantly, there were no significant differences between Class 1A and 1B tumors in terms of cell morphology, ciliary body involvement, or extraocular extension, underscoring the utility of molecular testing to delineate risk in these tumors that otherwise share similar pathologic features. Overall, the pathology and molecular class data in our clinically tested cohort reflect published reports that greater LBD and tumor thickness tend to be clinicopathologic features associated with more aggressive tumors [9,20,22,23], and these riskier phenotypes are most frequently seen in Class 1B or Class 2 tumors. An advantage of gene expression profiling is that it reflects objective, intrinsic tumor biology, whereas measurements of LBD in particular can be subjective due to variation between observers and in techniques used for size measurement [36]. Additionally, cytopathologic analysis can be impaired by a high rate of insufficient cellularity from FNABs [19]. Several studies have shown that GEP is the most significant independent prognostic factor for metastatic risk when compared to clinicopathologic features, including LBD [9, 18-20, 22, 23, 37].

Conclusions
In summary, the results of these analytic performance data demonstrate the reproducibility and reliability of the DecisionDx-UM test. This is confirmed by the high correlation of discriminant scores and concordance of molecular classifications on samples subjected to repeat testing. The robust high technical success rate of the test on even small amounts of tissue obtained by FNAB has been maintained from the test's original development through clinical implementation and testing of more than 5000 patients, and represents an important aspect of testing, given that patients who receive eye-sparing radiotherapy, usually do not have the opportunity to be biopsied again after treatment.

Additional file
Additional file 1: Table S1.