{'fraction_non_alphanumeric': 0.026330224904004388, 'fraction_numerical': 0.007131102578167855, 'mean_word_length': 4.941368078175896, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Force system generated by elastic archwires with vertical V bends: a three-dimensional analysis

Background
Our previous understanding of V-bend mechanics is primarily from two-dimensional (2D) analysis of archwire bracket interactions in the second order. These analyses do not take into consideration the three-dimensional (3D) nature of orthodontic appliances involving the third order.


Objective
To quantify the force system generated in a 3D two bracket set up involving the molar and incisors with vertical V-bends.


Materials and methods
Maxillary molar and incisor brackets were arranged in a dental arch form and attached to load cells capable of measuring forces and moments in all three planes (x, y, and z) of space. Symmetrical V-bends (right and left sides) were placed at 11 different locations along rectangular beta-titanium archwires of various sizes at an angle of 150degrees. Each wire was evaluated for the 11 bend positions. Specifically, the vertical forces (Fz) and anterio-posterior moments (Mx) were analysed. Descriptive statistics were used to interpret the results.


Results
With increasing archwire size, Fz and Mx increased at the two brackets (P < 0.05). The vertical forces were linear and symmetric in nature, increasing in magnitude as the bends moved closer to either bracket. The Mx curves were asymmetric and non-linear displaying higher magnitudes for molar bracket. As the bends were moved closer to either bracket a distinct flattening of the incisor Mx curve was noted, implying no change in its magnitude.


Conclusions
This article provides critical information on V-bend mechanics involving second order and third order archwire-bracket interactions. A model for determining this force system is described that might allow for easier translation to actual clinical practice.