{'fraction_non_alphanumeric': 0.05015873015873016, 'fraction_numerical': 0.020317460317460317, 'mean_word_length': 5.2047244094488185, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
High-temperature reversed-phase liquid chromatography coupled to isotope ratio mass spectrometry.

Compound-specific isotope analysis (CSIA) by liquid chromatography coupled to isotope ratio mass spectrometry (LC/IRMS) has until now been based on ion-exchange separation. In this work, high-temperature reversed-phase liquid chromatography was coupled to, and for the first time carefully evaluated for, isotope ratio mass spectrometry (HT-LC/IRMS) with four different stationary phases. Under isothermal and temperature gradient conditions, the column bleed of XBridge C(18) (up to 180 °C), Acquity C(18) (up to 200 °C), Triart C(18) (up to 150 °C), and Zirchrom PBD (up to 150 °C) had no influence on the precision and accuracy of δ(13) C measurements, demonstrating the suitability of these columns for HT-LC/IRMS analysis. Increasing the temperature during the LC/IRMS analysis of caffeine on two C(18) columns was observed to result in shortened analysis time. The detection limit of HT-RPLC/IRMS obtained for caffeine was 30 mg L(-1) (corresponding to 12.4 nmol carbon on-column). Temperature-programmed LC/IRMS (i) accomplished complete separation of a mixture of caffeine derivatives and a mixture of phenols and (ii) did not affect the precision and accuracy of δ(13)C measurements compared with flow injection analysis without a column. With temperature-programmed LC/IRMS, some compounds that coelute at room temperature could be baseline resolved and analyzed for their individual δ(13)C values, leading to an important extension of the application range of CSIA.