{'fraction_non_alphanumeric': 0.03406813627254509, 'fraction_numerical': 0.018036072144288578, 'mean_word_length': 4.519337016574585, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Variables affecting electronic root canal measurement.

This study was conducted in two parts. In the first part, 20 single-rooted teeth that had been scheduled for extraction were investigated. The electronic root canal lengths were measured in vivo with a Dental Sono-Explorer type Y-III, and the actual canal lengths were measured after extraction of the teeth. The rate of agreement of the two measurements was 77.5% within a range of +/- 0.5 mm, while it was 100% at +/- 2.0 mm, which is acceptable clinically. In the second part, there were 19 simulated canals whose lengths and apical foramen sizes were known beforehand. Experiments revealed a negative correlation between the areas of the apical foramina and the difference between the electronic and the actual root canal lengths. This relationship was shown by the linear regression equation: ŷ = 0.6-1.6x. With the exception of the smallest areas of foramina, electronic root canal length measurements were less than the actual lengths.