{'fraction_non_alphanumeric': 0.049926578560939794, 'fraction_numerical': 0.02643171806167401, 'mean_word_length': 5.089285714285714, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Determination of ambroxol in an automated multi-pumping pulsed flow system.

A new flow methodology exploiting the multi-pumping approach was developed for the spectrophotometric determination of ambroxol hydrochloride in pharmaceutical preparations. The flow manifold was implemented by using, exclusively, multiple solenoid-actuated micro-pumps, which acted simultaneously as sample insertion, solutions propelling and reagents commutation units. Linear calibration plots were obtained over an ambroxol concentration ranging from 10 to 200 mg l(-1) (r.s.d. < 0.5%, n = 15) and a sampling rate of about 60 samples per hour (flow rate = 1.92 ml min(-1), sample volume = 80 microl).