{'fraction_non_alphanumeric': 0.02621231979030144, 'fraction_numerical': 0.012450851900393184, 'mean_word_length': 5.2326530612244895, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
384-channel parallel microfluidic cytometer for rare-cell screening.

We have constructed a 384-channel parallel microfluidic cytometer (PMC). The multichannel architecture allows 384 unique samples for a cell-based screen to be read out in approximately 6-10 min, about 30-times the speed of a conventional fluorescence-activated cytometer system (FACS). This architecture also allows the signal integration time to be varied over a larger range than is practical in single-channel FACS and is suitable for detection of rare-cells in a high background of negatives. The signal-to-noise advantages have been confirmed by using the system to count rare clonal osteocytes in the most difficult early stages of an expression-cloning screen for the carboxy-terminal parathyroid hormone receptor (CPTHR). This problem requires finding several dozen positive cells in a background of one million negatives. The system is automated around a scanning laser confocal detector and a 96-tip robotic pipettor and can maintain in vitro cultures on-system in 384-well plates. It is therefore directly practical for biology applications using existing high-throughput culture facilities. The PMC system lends itself to high-sample-number cytometry with an unusual capability for time synchronization and rare-cell sensitivity. A limited ability to handle large sample numbers has restricted applications of single-channel FACS in combinatorial cell assays; therefore the PMC could have a significant application in high-throughput screening.