{'fraction_non_alphanumeric': 0.025728987993138937, 'fraction_numerical': 0.0137221269296741, 'mean_word_length': 4.3577981651376145, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A new instrument for measuring surface pressure.

A new instrument for surface pressure measurements using a floating-foil technique has been constructed. The instrument makes use of a probe in the form of a thin foil suspended at the gas/liquid or liquid/liquid interface tension. The present instrument has a dynamic range of 0.1-70 mN/m and a sensitivity better than 0.1 mN/m. Its speed of response is 50 ms but much faster measurements are possible. Measurements on behenic acid and cholesterol films on water and on a cerebroside film at a heptane/water interface are presented.