{'fraction_non_alphanumeric': 0.03667875856509472, 'fraction_numerical': 0.035872632003224506, 'mean_word_length': 4.615384615384615, 'pattern_counts': {'":': 0, '<': 6, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Validation of the Omron M6 (HEM-7001-E) upper-arm blood pressure measuring device according to the International Protocol in adults and obese adults

BackgroundElectronic blood pressure (BP) measurement devices are the preferred choice of patients owing to their user-friendly nature; however, there is a requirement to investigate the accuracy and reliability of these devices. The objective of this study is to evaluate the accuracy of the Omron M6 upper-arm BP device against the mercury sphygmomanometer in adults and obese adults according to the International Protocol criteria. MethodsOne hundred and twenty-one patients, older than 30 years of age, were studied and classified on the basis of the range of the International Protocol. BP measurements at the upper arm with the Omron M6 were compared with the results obtained by two trained observers using a mercury sphygmomanometer. Nine sequential BP measurements were taken. A total of 33 participants were selected for each validation study. During the validation study, 99 measurements were performed on 33 participants for comparison. The first phase was performed on 15 participants, and if the device passed this phase, 18 more participants were selected. Having a two-fold purpose, this study was conducted on both adult and obese adult patients. ResultsMean discrepancies and standard deviations of the monitor-mercury sphygmomanometer were 1.1±4.0 mmHg for systolic BP (SBP) and −0.5±3.5 mmHg for diastolic BP (DBP) in the adult group. The device passed phase 1 in 15 participants. In phase 2.1, out of a total of 99 comparisons, 88, 96, and 97 for SBP, and 88, 98, and 99 for DBP were <5, <10, and <15 mmHg, respectively. Mean discrepancies and standard deviations of the monitor-mercury sphygmomanometer were 1.7±4.8 mmHg for SBP and −0.8±4.3 mmHg for DBP in the obese adult group. The device passed phase 1 in 15 participants. In phase 2.1, out of a total of 99 comparisons, 82, 90, and 97 for SBP, and 80, 97, and 99 for DBP were <5, <10, and <15 mmHg, respectively. It was found that the Omron M6 automatic monitor, which measures BP at the upper arm, produced results in accordance with the criteria of phases 2.1 and 2.2 in both SBP and DBP, when applied to adults and to obese adults. ConclusionIt was concluded that the Omron M6 device, which measures BP at the upper arm, was deemed to be in accordance with the International Protocol criteria and can be recommended for use by adults and obese adults.