{'fraction_non_alphanumeric': 0.021289178001182733, 'fraction_numerical': 0.0029568302779420462, 'mean_word_length': 5.409090909090909, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Clinical and Radiographic Evaluation of Procedural Errors during Preparation of Curved Root Canals with Hand and Rotary Instruments: A Randomized Clinical Study

Background: The process of cleaning and shaping the canal is not an easy goal to obtain, as canal curvature played a significant role during the instrumentation of the curved canals. Aim: The present in vivo study was conducted to evaluate procedural errors during the preparation of curved root canals using hand Nitiflex and rotary K3XF instruments. Materials and Methods: Procedural errors such as ledge formation, instrument separation, and perforation (apical, furcal, strip) were determined in sixty patients, divided into two groups. In Group I, thirty teeth in thirty patients were prepared using hand Nitiflex system, and in Group II, thirty teeth in thirty patients were prepared using K3XF rotary system. The evaluation was done clinically as well as radiographically. The results recorded from both groups were compiled and put to statistical analysis. Statistical Analysis: Chi-square test was used to compare the procedural errors (instrument separation, ledge formation, and perforation). Results: In the present study, both hand Nitiflex and rotary K3XF showed ledge formation and instrument separation. Although ledge formation and instrument separation by rotary K3XF file system was less as compared to hand Nitiflex. No perforation was seen in both the instrument groups. Conclusion: Canal curvature played a significant role during the instrumentation of the curved canals. Procedural errors such as ledge formation and instrument separation by rotary K3XF file system were less as compared to hand Nitiflex.