{'fraction_non_alphanumeric': 0.023483365949119372, 'fraction_numerical': 0.0019569471624266144, 'mean_word_length': 5.162650602409639, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Digital breast tomosynthesis using an amorphous selenium flat panel detector

A prototype breast tomosynthesis system has been developed, allowing a total angular view of ±25°. The detector used in this system is an amorphous selenium direct-conversion digital flat-panel detector suitable for digital tomosynthesis. The system is equipped with various readout sequences to allow the investigation of different tomosynthetic data acquisition modes. In this paper, we will present basic physical properties -- such as MTF, NPS, and DQE -- measured for the full resolution mode and a binned readout mode of the detector. From the measured projections, slices are reconstructed employing a special version of filtered backprojection algorithm. In a phantom study, we compare binned and full resolution acquisition modes with respect to image quality. Under the condition of same dose, we investigate the impact of the number of views on artifacts. Finally, we show tomosynthesis images reconstructed from first clinical data.