{'fraction_non_alphanumeric': 0.03467879476975554, 'fraction_numerical': 0.033541785105173395, 'mean_word_length': 4.4489164086687305, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Modified energy settings are mandatory to minimize oesophageal injury using the novel multipolar irrigated radiofrequency ablation catheter for pulmonary vein isolation.

AIMS
The multipolar irrigated radiofrequency (RF) ablation catheter (nMARQ™) is a novel tool for pulmonary vein isolation (PVI). We investigated the incidence of thermal oesophageal injury (EI) using the nMARQ™ for PVI.


METHODS AND RESULTS
In the initial six patients (Group 1), RF was delivered at the posterior wall with a maximum duration of 60 s and a maximum power (maxP) of 20 W for unipolar ablation, and a maxP of 10 W for the bipolar ablation. In the latter 15 patients (Group 2), RF application was limited at the posterior wall to a maximum duration of 30 s and a maxP of 15 W for unipolar ablation a max P of 10 W for bipolar ablation. Oesophageal temperature monitoring was performed in all patients and ablation was terminated at a temperature rise >41°C. Endoscopy was carried out within 2 days post-ablation. Pulmonary vein isolation was performed during sinus rhythm and was successfully achieved in 83 of 84 PVs except the septal inferior vein in one patient. Charring was seen in 3 of 21 (14.3%) patients without any evidence of embolism. Phrenic nerve palsy occurred in one patient. Endoscopy revealed severe EI in 3 of 6 (50%) patients in Group 1 and in 1 of 15 patients (6.7%) in Group 2. Procedure times between Groups 1 and 2 were similar (228.3 ± 60.2 min vs. 221.3 ± 51.8 min; P = 0.79).


CONCLUSION
An unexpectedly high incidence of thermal EI was noted following PVI using the nMARQ™ with the initial ablation protocol. However, the incidence of thermal EI can be sigificantly reduced with limited power and RF application time at the posterior left atrium.