{'fraction_non_alphanumeric': 0.024026910139356077, 'fraction_numerical': 0.015377222489187891, 'mean_word_length': 4.627027027027027, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Physical aspects of dynamic stereotactic radiosurgery with very small photon beams (1.5 and 3 mm in diameter).

Stereotactic radiosurgery is often used for treating functional disorders. For some of these disorders, the size of the target can be on the order of a millimeter and the radiation dose required for treatment on the order of 80 Gy. The very small radiation field and high prescribed dose present a difficult challenge in beam calibration, dose distribution calculation, and dose delivery. In this work the dose distribution for dynamic stereotactic radiosurgery, carried out with 1.5 and 3 mm circular fields, was studied. A 10 MV beam from a Clinac-18 linac (Varian, Palo Alto, CA) was used as the radiation source. The BEAM/EGS4 Monte Carlo code was used to model the treatment head of the machine along with the small-field collimators. The models were validated with the EGSnrc code, first through a calculation of percent depth doses (PDD) and dose profiles in a water phantom for the two small stationary circular beams and then through a comparison of the calculated with measured PDD and profile data. The three-dimensional (3-D) dose distributions for the dynamic rotation with the two small radiosurgical fields were calculated in a spherical water phantom using a modified version of the fast XVMC Monte Carlo code and the validated models of the machine. The dose distributions in a horizontal plane at the isocenter of the linac were measured with low-speed radiographic film. The maximum sizes of the Monte Carlo-calculated 50% isodose surfaces in this horizontal plane were 2.3 mm for the 1.5 mm diameter beam and 3.8 mm for the 3 mm diameter beam. The maximum discrepancies between the 50% isodose surface on the film and the 50% Monte Carlo-calculated isodose surfaces were 0.3 mm for both the 1.5 and 3 mm beams. In addition, the displacement of the delivered dose distributions with respect to the laser-defined isocenter of the machine was studied. The results showed that dynamic radiosurgery with very small beams has a potential for clinical use.