{'fraction_non_alphanumeric': 0.04597701149425287, 'fraction_numerical': 0.028397565922920892, 'mean_word_length': 5.040816326530612, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Validation of a sensitive LC/MS/MS method for the determination of telaprevir and its R-isomer in human plasma.

The purpose of this study was to validate a reversed-phase high-performance liquid chromatographic (HPLC), tandem mass spectrometry (MS/MS) assay for the determination of telaprevir and its R-diastereomer (VRT-127394) in acidified and nonacidified human plasma. The chromatographic baseline separation of telaprevir and telaprevir-R was performed on a Waters XBridge(TM) BEH Shield C18 , 2.1 × 75 mm column with a 2.5 µm particle size, under isocratic conditions consisting of a mobile phase of 50:45:5 water-acetonitrile-isopropanol with 1% ammonia at 0.2 mL/min. This method utilized a stable isotope internal standard with 11 deuterium atoms on the structure of the telaprevir molecule (telaprevir-d11). An internal standard for the telaprevir-R (telaprevir-R-d11) was also prepared by incubating telaprevir-d11 in basic solution, which facilitated isomer inter-conversion. The detection and quantitation of telaprevir, telaprevir-R, telaprevir-IS and telaprevir-R-IS was achieved by positive ion electrospray (ESI+) MS/MS detection. The assay quantifiable limit was 5.0 ng/mL when 0.100 mL of acidified human plasma was extracted. Accuracy and precision were validated over the calibration range of 5.0-5000 ng/mL. It was demonstrated using patient samples that, contrary to previous recommendations, quantitation of telaprevir does not require acidified plasma.