{'fraction_non_alphanumeric': 0.02336448598130841, 'fraction_numerical': 0.011682242990654205, 'mean_word_length': 5.443609022556391, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Compact echelle spectrometer employing a cross-grating.

The concept and the implementation of a compact and simplified echelle spectrometer are presented, and the working principle is demonstrated by first experimental measurements. The crucial element of the setup is a cross-grating, combining an echelle grating utilizing several higher diffraction orders (5th up to 11th) and a superposed perpendicular-oriented cross-dispersing grating. Two alternative manufacturing approaches for the cross-grating are presented and discussed. The first approach combines Talbot lithography for the deep echelle grating and interference lithography for the cross-dispersing structure. As a second approach, direct laser-beam writing was applied. The compact echelle spectrometer covers a spectral range from 380 to 700 nm and offers a spectral resolution of ∼2  nm.