{'fraction_non_alphanumeric': 0.023155627355950458, 'fraction_numerical': 0.0032310177705977385, 'mean_word_length': 5.612099644128114, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A novel electrochemical aptasensor for thrombin detection based on the hybridization chain reaction with hemin/G-quadruplex DNAzyme-signal amplification.

In this work, a novel signal amplification electrochemical aptasensor for the sensitive and selective detection of thrombin was successfully fabricated. The amplification method was based on the hybridization chain reaction (HCR) and a pseudobienzyme electrocatalytic system. HCR-based double-stranded DNA (dsDNA) polymers not only constructed an effective carrier for anchoring larger amounts of electron mediator methylene blue (MB) into the DNA duplexes to produce a strong differential pulse voltammetry (DPV) signal, but also resulted in the formation of hemin/G-quadruplex DNAzymes nanowires by intercalating hemin into two induced single-stranded DNA (ssDNA). With the addition of NADH into the electrolytic cell, the hemin/G-quadruplex acting as an NADH oxidase and HRP-mimicking DNAzyme for the pseudobienzyme amplifying system could in situ biocatalyze the formation of H₂O₂ with local concentrations and low transfer loss resulting in dramatic signal enhancements. The binding event can be detected by a decrease in the integrated charge of MB which electrostatically absorbed onto dsDNA polymers. In the presence of thrombin, the dsDNA polymers associated with MB and hemin/G-quadruplex structures were removed from the electrode surface, leading to a significant decrease of redox current. DPV signals of MB provided quantitative measures of the concentrations of thrombin, with a linear calibration range of 0.01-50 nM and a detection limit of 2 pM. Moreover, the resulting aptasensor also exhibited good specificity, acceptable reproducibility and stability, indicating that the present strategy was promising for broad potential application in clinic assay and various protein analyses.