Choose a language

Calibrate Dew point

The E+E accredited dew point calibrations impress with a very low measurement uncertainty and a range from -90 °C (-130 °F) frost point to +95 °C (203 °F) dew point.

Standard calibrations are performed in nitrogen. Calibrations of dew point hygrometers in other carrier gases such as methane are possible upon request. Special calibrations of dew point meters and dew point mirrors are possible in the designated NMI laboratory.

Scope of Accreditation E+E Calibration Laboratory

CalibrationCalibration objectMeasurement conditionsMeasurement range1)Measurement uncertainty
AA 0608Lab

Dew point hygrometer
(Dew point mirror)

Dew point transmitter
(Dew point hygrometer)

Gas flow max. 5 l/min
Air pressure max. 10 bar for Td (-64.6...95) °C
950 mbar ± 150 mbar for Td <-64.6 °C
(-75...<-55) °C
(-55...<-22.5) °C
(-22.5...70) °C
(>70...95) °C
0.05-(55+Tf)*0.01 K
0.05 K
0.035 K
0.045 K
AA 0608Lab

Dew point transmitter
(Dew point hygrometer)

Pressure range 1 to 100 bar
Air or nitrogen2)
(-90...<-80) °C
(-80...<-55) °C
(-55...<-25) °C
(-25...20) °C
0.2-(80+Tf)*0.02 K
0.05-(55+Tf)*0.006 K
0.05 K
0.035 K

1) According to the BIPM Service category 3.1, "dew point" is used as denomination for the measurand. For values <0 °C the value refers to frost point. Calibration for values <0 °C can also be carried out for the calculated dew point temperature.
2) Other gases, such as methane, upon request

Calibration procedure

In a dual pressure humidity generator, air is generated with a defined mole fraction (water/gas) with a maximum gas flow of 5 l/min under normal conditions (displayed). This gas flow is conveyed to the test sample and at the sample location, generates a defined dew point temperature depending on the pressure. As an option, the test sample can be temperaturestabilised in a measurement chamber. The calibration system is the Austrian national standard for air humidity.

Order / Inquire Calibration

Personal Information
Downloads Kategorie
Brochure Calibrationservice
1437.75 kb
Accompanyslip AA0608 Calibrationservice
32.4 kb