{'fraction_non_alphanumeric': 0.020652173913043477, 'fraction_numerical': 0.009782608695652175, 'mean_word_length': 4.685185185185185, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
3-dimensional X-ray inspection of very large objects 600 kV Digital Laminography offers a solution

Testing large objects especially when 3-dimensional data is needed is still a challenge. There are three main obstacles in the process of acquiring 3-dimensional data of these objects. Due to the general planar geometry, it is not possible to penetrate the object in an axial Computed Tomography setup in all directions. Also due to the size of the object and the need of a sufficient resolution only a Region-of-Interest (ROI) can be scanned at a time. Furthermore, the material itself or the sheer amount of material denies penetration with conventional X-ray systems. Therefore a 600 kV Digital Laminography system was developed offering a solution for the above mentioned challenges in many cases. Contrary to axial CT, it is neither necessary to rotate the object nor to rotate the imaging system around the object.