{'fraction_non_alphanumeric': 0.034782608695652174, 'fraction_numerical': 0.008695652173913044, 'mean_word_length': 4.663934426229508, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Phase Locked BWO System for Open Resonator Measurement in D-Band

The combination of a D-Band phase-locked backward-wave oscillator (BWO) with a spectrum analyzer for measurement of permittivity and low loss-tangent is presented. For measuring low loss tangent material, such as CVD diamond and high purity semi-insulating (HPSI) 4H-SiC , at millimeter wave ranges, it is necessary to precisely measure an increase of a few kHz in a line-width of 200 kHz. We describe a phase-locked loop with frequency conversion that combines a BWO source and a microwave spectrum analyzer to obtain line-width measurements with less than 2 kHz (less than 1%) standard deviation in D-Band millimeter wave.