{'fraction_non_alphanumeric': 0.06384505021520803, 'fraction_numerical': 0.05380200860832138, 'mean_word_length': 5.065217391304348, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
The F, Cl, Br and I Contents of Reference Glasses BHVO‐2G, BIR‐1G, BCR‐2G, GSD‐1G, GSE‐1G, NIST SRM 610 and NIST SRM 612

Halogen contents for the widely distributed reference glasses BHVO‐2G, BIR‐1G, BCR‐2G, GSD‐1G, GSE‐1G, NIST SRM 610 and NIST SRM 612 were investigated by pyrohydrolysis combined with ion chromatography, total reflection X‐ray fluorescence analysis, instrumental neutron activation analysis, the noble gas method, electron probe microanalysis and laser ablation‐inductively coupled plasma‐mass spectrometry. Glasses BHVO‐2G, GSD‐1G and GSE‐1G have halogen contents that can be reproduced at the 15% level by all bulk techniques and cover a significant range in halogen mass fractions for F (~ 20–300 μg g−1), Cl (~ 70–1220 μg g−1) and Br (~ 0.2–285 μg g−1) and I (~ 9–3560 ng g−1). The BIR‐1G glass has low F (< 15 μg g−1), Cl (~ 20 μg g−1), Br (15 ng g−1) and I (3 ng g−1). The halogen contents for the silica‐rich NIST SRM 610 and 612 glasses were poorly reproduced by the different techniques. The relatively high Cl, Br and I abundances in glasses GSD‐1G and GSE‐1G mean that these glasses are well suited for calibrating spatially resolved micro‐analytical studies on silicate glasses, melt and fluid inclusions. Combined EPMA and laser ablation‐inductively coupled plasma‐mass spectrometry data for glass GSE‐1G demonstrate homogeneity at the 10% level for Cl and Br.