{'fraction_non_alphanumeric': 0.0273224043715847, 'fraction_numerical': 0.009562841530054645, 'mean_word_length': 4.511278195488722, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 1, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Development of a local vacuum system for focused ion beam machining.

A local vacuum system for focused ion beam (FIB) processing, with a workpiece set in the air, has been developed. The local vacuum apparatus had a double-wall cylinder structure, used a differential exhaust, and each cylinder was connected to a vacuum exhaust pump. When the gap between the workpiece and the apparatus was 10 microm, the pressure of beam line in the machining head achieved 2.1 x 10(-3) Pa. In addition, a visualization system was developed by visualizing the current flow out from a sample by FIB irradiation. With this system, it is possible to conduct focus adjustments of the FIB and shape recognition on a workpiece in the order of microns.