{'fraction_non_alphanumeric': 0.0257847533632287, 'fraction_numerical': 0.0033632286995515697, 'mean_word_length': 5.288732394366197, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Design and calibration of a parallel-moving displacement generator for nano-metrology

A displacement generator is realized which enables the calibration of a wide variety of displacement-measuring probes, such as probes of roundness testers, roughness testers and stand-alone type scanning probe microscopes (SPMs), in the range of with a standard uncertainty below 1 nm. A digital piezo translator (DPT) drives a flat mirror which serves as the calibration platform. This mirror is locked to an elastic, hysteresis-free, monolithic parallel guide. Calibration of the platform displacement is carried out by various methods including tunable and stabilized lasers, Fabry-Pérot interferometry and laser interferometry. The system is calibrated with a standard uncertainty of about 0.1 nm using three independent methods. As an example the calibration of an SPM using generated steps is shown.