{'fraction_non_alphanumeric': 0.04636459430979979, 'fraction_numerical': 0.045310853530031614, 'mean_word_length': 4.952978056426332, 'pattern_counts': {'":': 0, '<': 5, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Intraoperative Versus Postoperative Vault Measurement After Implantable Collamer Lens Implantation in a Large Cohort of Patients.

PURPOSE
To compare intraoperative and postoperative central vault measurement after implantable Collamer lens (ICL; STAAR Surgical) implantation using optical coherence tomography (OCT).


METHODS
A total of 574 eyes of 338 consecutive patients undergoing myopic ICL implantation were evaluated (mean age of 30.8 ± 5.7 years, 117 men and 221 women). Central ICL vault was measured both intraoperatively using the microscope-integrated iOCT (Optomedical Technologies GmbH) mounted on a standard surgical microscope (HS Hi-R NEO 900A; Haag-Streit Surgical GmbH), and postoperatively using the CASIA2 swept-source OCT (Tomey GmbH) at 4 and 24 hours.


RESULTS
Mean differences between intraoperative and postoperative vault values were 11.5 ± 29.0% of the mean value 4 hours postoperatively (P < .001) and 2.7 ± 33.5% of the mean value 24 hours postoperatively (P < .001). Correlation analysis shows significant agreement between vault values obtained intraoperatively and at the two postoperative times, 4 hours (Spearman Rho = 0.850, P < .001) and 24 hours (Spearman Rho = 0.745, P < .001). In 73% of cases, postoperative vault values 4 hours after the surgery could be predicted from intraoperative vault values, and so were 56% of vault values 1 day after surgery. Significant differences in vault were found at the different times, grouping cases by ICL size (Kruskal-Wallis, P = .028 for intraoperative vault, and P < .001 for both postoperative vault times), with smaller vault values for the smaller ICL size.


CONCLUSIONS
Intraoperative determination of ICL vault using iOCT is an effective method for predicting postoperative ICL vault and minimizing postoperative vault surprises that could require surgical retouching. [J Refract Surg. 2021;37(7):477-483.].