{'fraction_non_alphanumeric': 0.03125, 'fraction_numerical': 0.030208333333333334, 'mean_word_length': 4.652941176470589, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[A new evaporation procedure for monitoring of iodine-125 in liquid waste].

A simplified monitoring method of 125I in liquid waste was devised. The waste water of 200 cm3 was taken on a Saran (polyvinylidene chloride) film covering a stainless steel vat. A stable iodine (20 mg) and sodium hydroxide (1 mmol) was added. The water was evaporated using an infra-red lamp. After heating to dryness, the Saran film was folded and transferred into a polyethylene tube. The radioactivity of 125I was counted with a well type NaI(Tl) scintillation counter. When a multi-channel analyzer was available for counting, an absolute decay rate of 125I was calculated with single and sum photo-peak counts. The radioactivity of 125I counted by a single-channel counter must be corrected with the counting efficiency of about 55%, with a special emphasis of a self absorption of photons. The recovery of 125I for concentrations below the permissible level was more than 98%.