{'fraction_non_alphanumeric': 0.043465022455659585, 'fraction_numerical': 0.03257973662175535, 'mean_word_length': 4.717145343777197, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Different Approaches to FT-IR Microspectroscopy on X-ray Exposed Human Cells

: Fourier-Transform Infrared microspectroscopy ( μ FT-IR) has been usefully applied in the analysis of the complex biological processes occurring during X-ray radiation-cell interaction. Different experimental approaches are available for FT-IR spectra collection (transmission, attenuated total reflection (ATR), and transflection modes) from cells samples. Recently, some problems have been raised about the role of transmitted and reflected components of the infrared beam in transflection mode. For this reason, we investigated two different transflection approaches for collecting spectra from cells exposed to X-ray. In the former approach, cells were grown on MirrIR slides, and for the second approach, cell pellets were prepared. In both cases, SH-SY5Y neuroblastoma cells were used. X-ray exposure was performed at doses of 2 and 4 Gy. Spectra were obtained by using both the approaches in the 600–4000 cm − 1 spectral range from exposed and not-exposed samples. The main contributions from proteins, lipids, carbohydrates, and DNA were clearly evidenced in spectra obtained with the two different acquisition approaches. A comparison among them has been also reported.


Introduction
Fourier-Transform Infrared microspectroscopy (µFT-IR) is nowadays considered a valuable tool for investigating biochemical changes occurring in cells during the interaction with external agents [1]. In particular, µFT-IR has been usefully applied in the analysis of the complex biological processes occurring during X-ray radiation-cell interaction [2,3]. Different experimental approaches are available for FT-IR spectra collection (transmission, attenuated total reflection (ATR), and transflection modes) on cells samples. Transflection-mode FTIR spectroscopy is particularly used for this kind of samples due to the relatively low cost of required substrates compared to transmission windows, and a higher absorbance due to a double pass through the same sample approximately doubling the effective path length. Recently, some questions have been raised about the role of transmitted and reflected components of the infrared beam in transflection mode [4,5]. For this reason, we started investigating two different transflection approaches for collecting spectra from cells exposed to X-ray. In the former approach, cells were grown on MirrIR slides, a specific reflection FT-IR spectroscopy microscope slide; for the second approach cell pellets were prepared. In both cases, SH-SY5Y neuroblastoma cells were used. X-ray exposure was performed at doses of 2 and 4 Gy. FT-IR absorption spectra were obtained by using both the approaches in the 600-4000 cm −1 spectral range from exposed and not-exposed (control) samples. The main contribution from proteins, lipids, carbohydrates, and DNA has been clearly evidenced and assigned. A comparison among spectra obtained with the different experimental approaches has been also reported.

Materials
DMEM medium, fetal bovine serum, penicillin, streptomycin, L-glutamine, and formaldehyde were provided by Sigma-Aldrich Co. and used without any further treatments. SH-SY5Y (American Type Culture Collection, Manassas, VA, USA) is a human cell line subcloned from a bone marrow biopsy taken from a four-year-old female with neuroblastoma.

Sample Preparation and Treatment
SH-SY5Y cells were cultured in vitro in DMEM medium, supplemented with 15% fetal bovine serum, 1% penicillin, 1% streptomycin, and 1% L-glutamine. They were grown at 37 °C, 5% CO2 in 25 cm 2 flasks. Two types of samples were prepared: For the first (slide-cells), the cells were seeded on MirrIR slides (25 × 25 mm 2 ) (Kevley Technologies, Chesterland, Ohio), a specific reflection FT-IR spectroscopy microscope slide, nested into petri dishes (60 mm diameter); for the second approach (drop-cells) cell pellets were prepared. The number of cells was ~ 4 × 10 for the slide-cells samples, since in these conditions, cells were not confluent as to leave sufficient inter-cellular spaces for measurement of the background signal, and ~ 2 × 10 cells/pellet were used for the drop-samples.
X-ray irradiation was performed at room temperature, using a Gilardoni MGL 200/8D machine operating at 250 kVp and 6 mA (dose rate 60 cGy/min). Both types of cellular samples were exposed to various doses of X-rays (2, 4 Gy), and then investigated together with unexposed cells (0 Gy).
After X-ray exposure, the cells grown on MirrIR slides were fixed in a 3.7% formaldehyde PBS solution for 20 min at room temperature, and then briefly washed in distilled water for 3 s to remove the residue PBS from the surface of the cells. Subsequently, the samples were dried under ambient conditions and stored in a desiccator until spectral analysis. Cell pellets after the exposure were centrifuged for 8 min at 1500 rpm. The supernatant was aspirated, and the pellet resuspended in 300 µL of NaCl 0.9% until spectra acquisition.

FT-IR Micro-Spectroscopy Measurements
IR absorption spectra of the cells samples were acquired at room temperature, using a Spectrum One FTIR (PerkinElmer, Shelton, CT, USA) spectrometer equipped with a Perkin Elmer Multiscope system infrared microscope and an MCT (mercury cadmium telluride) FPA (focal-plane-array) detector. For the slide-cells samples, the measurements were performed on cells grown on 25 × 25 mm 2 MirrIR slides in transflection mode. Spectra were collected within an aperture of 100 × 100 µm 2 . The background signal was acquired in a region of the slide free of cells. Every slide was examined in different regions, and multiple spectra were acquired for each position. The signal was collected in the spectral region between 4000 and 600 cm −1 using 16 scans with a spectral resolution of 4 cm −1 and a 5 s acquisition time for each spectrum at room temperature.
For the drop-cells samples, a drop of 3 µl volume was taken from the 0.9% NaCl cells suspensions irradiated at various doses and deposited on a normal microscope slide coated with aluminum paper, which has the task of reflecting the IR light after the passage through the sample. The so-obtained slides were left to dry at ambient condition and subsequently analyzed with the spectrometer following the same procedure used for the slide-cells samples.

Preliminary Process
The whole dataset of the spectra detected from cells was preliminarily processed by subtracting the corresponding background spectrum from the measured one and, subsequently, noise corrections was performed on the whole data set by a numerical procedure based on wavelet algorithms ("MATLAB Wavelet Toolbox", MathWorks Inc., Natick, MA, USA) [6][7][8]. The signal was recalculated from detail components up to the nine level. Then, a piecewise baseline correction was performed as usually done to eliminate the possible contributions due to scattering or absorption by the supporting substrate in transmission or transflection type IR spectroscopy [9]. After this step, the spectra were processed via vector normalization procedure in order to have comparable intensities, using Standard Normal Variate method [9,10].

Average Spectra
Average spectra were obtained for each type of samples and positions of principal peaks and their absorbance values were evaluated. The related graphic elaborations were carried out using MATLAB (MathWorks Inc., Natick, MA, USA) software.

Control Spectra
FT-IR spectra of SH-SY5Y neuroblastoma cells consist of several bands arising from the vibrational modes of the biologic molecules of cells constituents (lipids, proteins, DNA, etc.). In Figure 1 average spectra of unexposed samples, for both slide-and drop-cells, obtained in the region 900-3600 cm −1 , are reported. The spectra appear to be divided into two principal zones with different visible peaks. The range from 2800 to 3600 cm −1 (Figure 1A,C) is generally indicated as high wavenumber region (HWR) and presents bands that are due to the contribution of proteins, lipids, and carbohydrates. In the so-called fingerprint region (900-1800 cm −1 ) ( Figure 1B,D), different peaks that are representative of proteins and nucleic acids are clearly visible. In Table 1, the assignments of the main peaks are reported according to literature [2,3,[11][12][13]. Table 1. FT-IR peaks observed in the spectrum of control cells, with assignments in accordance with the data reported in the literature [2,3,[11][12][13]; abbreviation: as = asymmetric, s = symmetric, ν = stretching, δ = bending, sc = scissoring, vbr = vibration, a. a. = free amino acids.

Peak
ASSIGNMENT Amide II (C-N ν, C-NH δ) ≈1450 CH3 as. δ, CH2 sc. CH3 as. δ, CH2 sc. In the HWR region, it is possible to observe lower absorbance values for the peaks in the range (3200-3500 cm −1 ) for the slide-cells samples; similarly, the peaks linked mainly to the stretching of the -CH2 and -CH3 groups in the range (2800-3000 cm −1 ) presents lower absorbance values for the slidecells samples. In the fingerprint region, some significant differences are visible: Higher absorbance values are detected for the slide-cells samples in the region (1500-1300 cm −1 ), with peaks mainly attributed to proteins and lipids. The peaks centered at ≈1450 cm −1 and ≈1250 cm −1 in the drop-cells spectrum, attributed to −CH2 and −CH3 bending of lipids and proteins and asymmetric −PO2 − stretching vibrations of the DNA phosphodiester, respectively, appears to be shifted to lower wavenumbers, to ≈1480 cm −1 and ≈1200 cm −1 , in the slide-cells spectrum. The peak at ≈1080 cm −1 , also attributed to −PO2 − stretching vibrations of the DNA, presents higher absorbance for the slide-cells spectrum in respect to the drop-cells one. The SEM values (black and blue shadows in Figure), calculated for every wavenumber, show that the greater variability between spectra of the same sample type is present in the range 2800-3000 cm −1 and in the fingerprint region, in particular for the range 1200-1400 cm −1 and for the band at ≈1080 cm −1 , for both drop-and slide-samples.

Irradiated Spectra
In Figure 2, average spectra of 2 Gy dose exposed samples, for both slide-and drop-cells, obtained in the range 2800-3600 cm −1 (Figure 2A,C) and is 900-1800 cm −1 ( Figure 2B,D), respectively, are reported. In the HWR region, it is possible to observe, as in the case of the control spectra, lower absorbance values for the peaks in the range (3200-3500 cm −1 ) and (3000-2800 cm −1 ) for the slide-cells samples. The same is observed for the spectra irradiated at 4 Gy dose (not shown). In the fingerprint region, a shift towards lower wavenumbers is detected for the peak at ≈ 1450 cm −1 in the slide-cells for both 2 Gy and 4 Gy irradiated samples, as already observed in the control sample; also, the peak centered at ≈1250 cm −1 in the drop-cells spectrum presents a shift to ≈1280 cm −1 in the slide-cells spectrum, for both doses. The average drop-cells spectrum for the 4 Gy dose presents higher absorbance values in the range (1000-1300 cm −1 ) compared to that of the slide-cells, as for the control samples. Conversely, this difference is not visible for the spectra irradiated at 2 Gy dose. In terms of inner variability, the 2 Gy spectra for the drop-cells presents higher values of SEM (black and blue shadow in figure) in the range 3000-3600 cm −1 , compared to samples of the same type at other doses and with the 2 Gy slide-cells spectrum. In the fingerprint region for both 2 and 4 Gy spectra, the greater variability is present for slide-cells spectra, in the range 1000-1400 cm −1 . Figure 2. Average spectra of 2 Gy exposed cells samples: Drop-cells sample spectrum in the range (2800-3600 cm −1 ) (A) and in the range (900-1900 cm −1 ) (B), respectively; slide-cells sample spectrum in the range (2800-3600 cm −1 ) (C) and in the range (900-1900 cm −1 ) (D), respectively; the break at (1900-2800 cm −1 ) hides a region without interesting signals from biological molecules. Data presented as Mean ± SEM.
To take into account the internal variability of the spectra obtained by using the different approaches, the mean value of the SEM (in absolute value, for each wavenumber) for both drop-and slide-samples, for the HWR and fingerprint region, at the different irradiation doses were calculated ( Table 2). This parameter allows us to have a preliminary estimate of the variations present in the spectra for the two procedures used for the preparation of the samples.

Dose
Drop-Cells Slide-Cells Control 0.05 ± 0.02 0.05 ± 0.021 2 Gy 0.05 ± 0.03 0.04 ± 0.02 4 Gy 0.03 ± 0.01 0.04 ± 0.02 The inspection of the values indicate that the two approaches are able to give similar results as far as concerns the signal variability for samples related to the same experimental conditions.

Conclusions
The results here reported indicate that the two investigated approaches enable the collection of spectra with analogous characteristics. This preliminary investigation suggests that the researcher can choose the modality which is more suitable in the execution of the particular experimental study to be carried out.