{'fraction_non_alphanumeric': 0.013829787234042552, 'fraction_numerical': 0.005319148936170213, 'mean_word_length': 4.808641975308642, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Evaluation of ITER TF Coil Joint Performance

To evaluate the ITER TF joint performance, the joint test sample, which consists of two short TF conductors and has full size joint, shall be tested using NIFS test facility under the condition of current of 68 kA and external field of 2 T. For high accuracy, the issue of voltage difference between cable and jacket had been anticipated in the evaluation of joint resistance. If a voltage difference exist between them, it is difficult to measure real joint resistance using voltage taps on the jacket. Therefore, the author first calculated the position where voltage of cable and jacket become equipotential and then decided the voltage tap position where the influence of voltage drop could be avoided. Thus, a high accuracy measurement of joint resistance could be achieved and the joint resistance was accurately evaluated as around 1 nΩ, which is well below the ITER requirement of 3 nΩ.