{'fraction_non_alphanumeric': 0.02275960170697013, 'fraction_numerical': 0.01849217638691323, 'mean_word_length': 5.068965517241379, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Static microperimetry with the laser scanning ophthalmoscope].

We have used the scanning laser opthalmoscope (SLO) with a personal computer to develop static microperimetry techniques. They allow to see in real-time on a television monitor the precise retinal localization of the stimulus and fixation. The testing is performed under strict conditions. The size of stimuli can vary between 6 and 30 minarc on a side. 255 different intensity levels are possible with the instrument. We have selected 12 of them, representing a logarithmic scale. Stimulus duration can vary between 50 and 500 ms. Examples of macular pathology including subretinal neovascularization, drusen and macular edema are given.