{'fraction_non_alphanumeric': 0.02129817444219067, 'fraction_numerical': 0.0067613252197430695, 'mean_word_length': 5.151767151767152, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Description and performance of the OGSE for VNIR absolute spectroradiometric calibration of MTG-I satellites

The Meteosat Third Generation (MTG) Programme is being realised through the well-established and successful Cooperation between EUMETSAT and ESA. It will ensure the future continuity of MSG with the capabilities to enhance nowcasting, global and regional numerical weather prediction, climate and atmospheric chemistry monitoring data from Geostationary Orbit. This will be achieved through a series of 6 satellites named MTG-I and MTG-S to bring to the meteorological community continuous high spatial, spectral and temporal resolution observations and geophysical parameters of the Earth based on sensors from the geo-stationary orbit. In particular, the imagery mission MTG-I will bring an improved continuation of the MSG satellites series with the Flexible Combined Imager (FCI) a broad spectral range (from UV to LWIR) with better spatial and spectral resolutions. The FCI will be able to take high spatial resolution pictures of the Earth within 8 VNIR and 8 IR channels. As one of the mission of this instrument is to provide a quantitative analysis of atmosphere compounds, the absolute observed radiance needs to be known with a specified accuracy for VNIR as low as to 5% at k=3 over its full dynamic. While the FCI is regularly recalibrated every 6 month at equinoxes, it is however requiring initial ground calibration for the beginning of its mission. The Multi Optical Test Assembly (MOTA) is one of the Optical Ground Support Equipment (OGSE) dedicated to various missions necessary for the integration of the FCI . This equipment, provided by Bertin Technologies, will be delivered to TAS-F by the end of 2016. One of its mission, is the on-ground absolute calibration of VNIR channels. In order to handle this, the MOTA will be placed in front of the FCI under representative vacuum conditions and will be able to project a perfectly known, calibrated radiance level within the full dynamic of FCI instrument. The main difficulty is the very demanding calibration level with respect to primary standards down to 3% (k=3) coupled with constraining environment (vacuum), large dynamic (up to factor 100), high spectral resolution of 3 nm. Another main difficulty is to adapt the specific MOTA etendue (300 mm pupil, 9 mrad field) to available primary standards. Each of these constraints were addressed by specific tool design and production, a fine optimization of the calibration procedure with a large involvement of metrology laboratories. This paper introduces the missions of MTG satellites and particularly of the FCI instrument. The requirements regarding the absolute calibration over the different spectrometric channels and the global strategy to fulfill them are described. The MOTA architecture and calibration strategy are then discussed and final expected results are presented, showing state of the art performances.