{'fraction_non_alphanumeric': 0.04204035874439462, 'fraction_numerical': 0.03699551569506727, 'mean_word_length': 4.6847133757961785, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Impact of the Insoluble Gas Concentration on Measured Stroke Volume at Rest and Submaximal Exercise using the Innocor Device.

INTRODUCTION
The Innocor® device utilizes an insoluble gas (SF6) to estimate lung volume and the rate of disappearance of a soluble gas (N2O) to measure pulmonary blood flow (PBF), which approximates cardiac output assuming no shunt. We sought to identify error in the measurement of the insoluble gas in an effort to reduce variation in Innocor® measurement.


METHODS
We enrolled 28 participants from the Dallas Heart Study (mean age 63 years; 57% male; 43% white). Stroke volume was measured at rest and at submaximal (20 and 40 W) exercise using both echocardiography (Philips iE33) and the Innocor® device. We defined a priori peak and equilibrium SF6 measurement errors as greater or less than 20% of the mean observed value. Three Innocor measurements were obtained at rest (N = 27) for a total of 81 measurements. Of these, 22 % had SF6 measurements that fell outside of the priori range.


RESULTS
Resting Innocor® stroke volume measures with peak SF6 measured above a priori range (>0.12%) was associated with larger stroke volumes compared to stroke volume measures without peak SF6 error [101.4 (26.8) vs. 64.9 (8.7) mL, p = 0.006] and overestimated stroke volume when compared to stroke volume by echo [101.4 (26.8) vs. 59.9 (16.3) mL, p = 0.017]. A similar pattern was observed at submaximal exercise. In contrast, there was no consistent association between variation in equilibrium SF6 concentrations and measured stroke volume.


CONCLUSIONS
Variability in peak SF6 concentration is common while using the Innocor® device and results in overestimated stroke volume. These findings have implications for research protocols using this device.