{'fraction_non_alphanumeric': 0.0368610747051114, 'fraction_numerical': 0.02416448230668414, 'mean_word_length': 4.547024079963653, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
New fluoroscopic imaging technique for investigation of 6DOF knee kinematics during treadmill gait

Introduction This report presents a new imaging technique for non-invasive study of six degrees of freedom (DOF) knee kinematics during treadmill gait. Materials and methods A treadmill was integrated into a dual fluoroscopic imaging system (DFIS) to formulate a gait analysis system. To demonstrate the application of the system, a healthy subject walked on the treadmill at four different speeds (1.5, 2.0, 2.5 and 3.0 MPH) while the DFIS captured the knee motion during three strides under each speed. Characters of knee joint motion were analyzed in 6DOF during the treadmill walking. Results The speed of the knee motion was lower than that of the treadmill. Flexion amplitudes increased with increasing walking speed. Motion patterns in other DOF were not affected by increase in walking speed. The motion character was repeatable under each treadmill speed. Conclusion The presented technique can be used to accurately measure the 6DOF knee kinematics at normal walking speeds.


Introduction
Accurate data of six degrees-of-freedom (6DOF) knee kinematics is instrumental for investigation of biomechanical mechanisms of knee pathology such as osteoarthritis, ligamentous injuries and total knee arthroplasty. Traditional gait analysis used multiple video cameras to track the three-dimensional (3D) motions of reflective markers fixed to the skin [1], which was limited to reveal relative motion of the femoral and tibial bones. Invasive methods, such as using reflective markers directly fixed to bone using a thin rod [2] or opaque markers embedded within the bones, [3][4][5][6][7] were applied to detect bony motion in order to eliminate the effect of skin motion and enhance the accuracy of motion data. In another way, a point-cluster technique, which is noninvasive, has also been proposed to improve the traditional gait analysis method in order to reduce the effect of relative motion of the skin and bones [8].
Recently, fluoroscopic imaging technique, due to its relative accessibility, easiness to operate, and low radiation dosage compared to traditional X-rays, has been used for the analysis of knee joint motion during gait [9][10][11]. However, the use of just a single image might not detect knee joint motion in the out-of-plane degrees-of-freedom in the same accuracy as compared to the accuracy in in-plane motion [12,13]. In our laboratory, we validated the method using the cine function of two fluoroscopes to simultaneously capture dynamic knee joint motion [14]. This study presents how to use this technique to determine 6DOF knee joint motion during treadmill gait with different speeds.

DFIS Setup
The dual fluoroscopic imaging system (DFIS) setup that was validated previously is used for the treadmill gait analysis (Fig. 1) [14,15]. The DFIS consists of two pulse fluoroscopes (BV Pulsera, Philips) that are set to generate 8 ms width X-ray pulses with an effective dose of 13 mrem per scanning. In this study, the fluoroscopes took 30 evenly distributed snapshot images per second during dynamic knee joint motion.
The diameter of the image intensifier of the fluoroscopes is ~310 mm. In general, given the size of the image intensifier of the fluoroscopes might be difficult to capture the entire range of knee motion during the treadmill gait. Therefore, the two fluoroscopes are positioned so that their intensifiers form an angle between 120 and 130°( Fig. 1). In this setup, the dual fluoroscopic system has a common field of view with a length of ~450 mm. Therefore, the entire knee motion could be captured by both fluoroscopes during the gait cycle.

Treadmill gait
To demonstrate the methodology of treadmill gait analysis, one healthy subject (male, 45 years old) performed gait on the treadmill at different speeds: 1.5, 2.0, 2.5 and 3.0 mile/hour MPH (or 0.67, 0.89, 1.12 and 1.34 m/s, respectively). Two laser-positioning devices were attached to the two fluoroscopes to help the subject align the target knee (left) within the field of view of the fluoroscopes during gait with the assistance of a technician. The knee was then imaged from heel strike to toe-off during three consecutive strides after about 30 seconds of practice. The subject took 5 minute rest after testing for each speed.

Reproduction of in-vivo knee kinematics
The anatomic model of the target knee, including the bony geometry of the tibia and femur, was reconstructed by tracing the bony contours on sagittal plane magnetic resonance (MR) images of the knee in solid modeling software (Rhinoceros ® , Robert McNeal & Associates, Seattle, WA). The MR images were obtained using a 3.0 Tesla MR scanner (MAGNETOM ® Trio, Siemens, Erlangen, Germany) while the subject was lying supine with the knee in a relaxed, extended position. The MR scanner employed a 3D double echo water excitation sequence and the following parameters: field-of-view = 160 × 160 × 120 mm, voxel resolution = 0.31 × 0.31 × 1.00 mm, time of repetition (TR) = 24 ms, time of echo (TE) = 6.5 ms, and flip angle = 25°. A joint coordinate system described previously ( Fig. 1B) was adopted to determine the 6DOF knee joint kinematics [16].
The model and the dual fluoroscopic images were placed into a virtual DFIS environment where the in-vivo positions of knee were reproduced by matching projections of the models to their outlines on the fluoroscopic images [12]. The knee positions during three strides at each treadmill speed were reproduced. For each stride, the knee position was analyzed at each 10% of the stance phase from heel strike to toe-off.
The average speed of the knee during stance phase was calculated by dividing the maximal traveling distance by the corresponding traveling time. The data on 6DOF knee kinematics, including knee flexion, internal-external tibial rotation, as well as medial-lateral translation and varus-(A) Setup of the DFIS system and a treadmill

Results
The duration of the stance phase decreased with the treadmill speed (Fig. 2). At 1.5 MPH, the stance phase time was 0.99 second, while at treadmill speed of 3.0 MPH, the stance phase time decreased to 0.49 second. The average speed of the knee during stance phase was lower than the treadmill speed (Fig. 2). At 1.5 MPH treadmill speed, the knee speed was 0.28 ± 0.02 m/second, only 41% of the treadmill speed. At 2.5 MPH treadmill speed, the knee speed was 0.39 ± 0.05 m/second that was 35% of the treadmill speed. At 3.0 MPH treadmill speed, the knee speed was 0.81 ± 0.02 m/second that was 60% of the treadmill speed.
Knee kinematics under different treadmill speeds showed similar patterns in both, the rotations and the translations (Fig. 3). After heel strike, the tibia showed an increase in flexion angle to 6.71 ± 0.86° and 13.81 ± 2.73° and an increase in internal tibial rotation to 4.56 ± 0.29° and 5.45 ± 0.76° for the treadmill speeds of 1.5 MPH and 3.0 MPH, respectively (Fig. 2B). During mid-stance, the knee showed maximal hyperextension of about -2.5° and external tibial rotation of about -1.5° at all speeds. At toeoff, the knee had flexion angles of 43.05 ± 2.18° and 52.35 ± 5.09° and internal tibial rotation of 4.73 ± 0.35°a nd 7.56 ± 1.50° for treadmill speeds of 1.5 and 3.0 MPH, respectively. The knee also showed an increase in valgus rotation after heel strike, a decrease in valgus rotation during stance phase and sharper increase in valgus rotation at toe-off ( Fig. 2B and Fig. 3C).
Femoral translations during gait at different speeds showed similar patterns (Figs. 3D and 3E). The femur translated anteriorly during loading response and early midstance and moved posteriorly thereafter until terminal stance when it shifted anteriorly again. In medial-lateral direction, the femur moved laterally during early stance and medially towards toe-off.

Discussion
This paper introduced the technique of using the DFIS for measurement of 6DOF in-vivo knee kinematics during treadmill gait. The data showed that this technique is feasible to analyze the dynamic knee motion during a wide Top: Average speed of the knee and duration of the stance phase of gait on treadmill at four different walking speeds  Few studies have utilized fluoroscopes to investigate human knee motion during gait [17]. For example, Zihlmann et al. [17] moved a fluoroscope to follow the knee motion to overcome the limited field of view of the image intensifier. They estimated an accuracy of 0.2 mm for in-plane translation and of 3.25 mm for out-plane translation and an accuracy of 1.57° for rotation in a knee after total joint arthroplasty during level walking. To overcome the limitation of the image intensifier size in the DFIS set up, the two fluoroscopes are positioned so that their common image zone covers the knee motion during the complete gait cycle on a treadmill [14].
The DFIS has been recently validated to measure dynamic knee motion [14]. Using standard geometry, the sphere positions could be determined with a SD below 0.2 mm when sphere moved at a speed up to 0.5 m/second. The dynamic validation using cadaveric knees, demonstrated that the DFIS on average has an accuracy of less than 0.15 mm and 0.1 mm/s in determining translation and velocity, respectively. Varadarajan et al. [15] demonstrated that the DFIS can measure translation in knee after total joint arthroplasty with an accuracy of less than 0.4 mm at a speed of 0.5 m/s. The accuracy of the DFIS depends on the speed of the moving joint.
The data of this paper revealed that the knee traveling speed is lower than the treadmill speed (Fig. 2). At the treadmill speed of 2.5 MPH, the knee speed during stance phase is less than 0.4 m/second, while at treadmill speed of 3.0 MPH, the knee speed is about 0.8 m/second. Our data also showed that with increasing speed, the amplitude of knee flexion during stance phase increases. This finding is in agreement with other studies in the literature [18][19][20][21].
Treadmill gait was also shown to be repeatable across the multiple strides as indicated by the standard deviation calculated from three strides at each treadmill speed.
The pulse imaging character of the fluoroscopes is an important factor for analyzing treadmill gait. In a pulsed fluoroscopic system such as the one used in our set up, the pulse width and frame rate are decoupled parameters. The inverse of frame rate corresponds to time difference between two consecutive images, whereas pulse width corresponds to excitation time for each image. If the rate at which pulses are generated is matched to the rate of acquisition then each image corresponds to a pulse. In this case, pulse width limits the image quality for a given frame rate. Theoretically, the maximal pulse rate (and matched frame rate) is limited by the pulse width. Therefore, a pulse width of 8 ms has a maximum frame rate of 125 frames/second, which is higher than the recommended minimal frame rate of 60 frames/second for gait analysis. However, a reduced rate of image capture (e.g. 15 or 30 pulses/second and matched frame rates) can be employed to limit unnecessary radiation exposure and data processing without adversely affecting image quality. This is because the image quality is actually related to pulse width even though fewer images are taken. Therefore, we could chose to use 15 or 30 pulses/second in our application, depending on the moving speed of the joint.
In-vivo knee kinematics during the stance phase of gait on the treadmill In summary, this paper introduced the DFIS technique for measurement of 6DOF in-vivo knee kinematics during treadmill gait. The technique showed feasibility to analyze the dynamic knee motion during wide range of walking speeds (up to 3 MPH). The fluoroscopic system has a low radiation dosage, is non-invasive, and can be constructed using any pair of readily available fluoroscopes. Since this technique reproduced the knee positions during gait using 3D anatomic models of the knee, 6DOF tibiofemoral joint kinematics can be accurately obtained. This technique can be used as an alternative option for treadmill gait analysis in healthy, injured, and surgically treated knees.