{'fraction_non_alphanumeric': 0.057176196032672114, 'fraction_numerical': 0.050175029171528586, 'mean_word_length': 3.959537572254335, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Studies on an antibody-coated tube method of aldosterone].

We studied on the basic performance of a non-extraction coated tube RIA kit, DPC Aldosterone kit (Nippon DPC Corporation). 1. The average binding CVs of the calibrators (25 to 1,200 pg/ml) were stable in a range from 1.4 to 1.9%. 2. The minimum sensitivity was 17 pg/ml from 2SD method of 0 calibrator. 3. The intra-assay reproducibility test showed an average CV as 9.0%, and the inter-assay test showed as 8.8%. 4. The recovery ratios of a recovery test were 85.4 to 121.6%. 5. The correlation test with another maker's kit showed a correlation curve as y = 0.970 x + 19.68 and a correlation coefficiency as r = 0.970. 6. The normal range, the correlation between fasting recumbency and daytime sitting, and the correlation between walking and resting, showed similar results as former reports.