{'fraction_non_alphanumeric': 0.009812667261373774, 'fraction_numerical': 0.0008920606601248885, 'mean_word_length': 5.064864864864865, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Response Time of a Fiber Bragg Grating Based Hydrogen Sensor for Transformer Monitoring

We developed and optimized a new fiber optic sensor using palladium foils attached to optical fiber Bragg gratings (FBG) for hydrogen measurements. Fifteen in parallel processed sensors were characterized and qualified in two custom tailored experimental set ups and their response to a 5% hydrogen/nitrogen gas mixture and the same gas bubbled trough transformer oil was measured. The hydrogen response is similar for both medium and close to the theoretical maximum sensitivity, but the response time was found to be very different, much slower in oil than in gas. A theoretical comparison of hydrogen diffusion trough palladium and hydrogen absorption on the palladium surface as well as a measurement of the hydrogen uptake and diffusion trough the oil to the sensor have been done to investigate the origin of the different response time. They indicate that the response time determining step is the absorption of hydrogen on the palladium surface and that this process is slowed down in oil compared to a pure gas environment.