{'fraction_non_alphanumeric': 0.02626641651031895, 'fraction_numerical': 0.0028142589118198874, 'mean_word_length': 5.753164556962025, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Simultaneous determination of optical loss, residual reflectance and transmittance of highly anti-reflective coatings with cavity ring down technique.

Cavity ring down (CRD) technique was employed to measure optical losses (absorption and scattering losses), residual reflectance and transmittance of anti-reflectively (AR) coated laser components with transmittance higher than 99.9%. By inserting the AR coated laser component with parallel optical surfaces into the ring-down cavity and measuring the ring-down time versus the angle of incidence with respect to the surface normal, the optical loss and residual reflectance of the laser component were determined respectively at normal and out-of-normal incidences with repeatability of part-per-million level. The transmittance was also determined simultaneously. Experimental results demonstrated that CRD is a simple, inexpensive and fast technique for highly accurate measurements of optical loss, residual reflectance, and transmittance of AR coated laser components widely used in high-power laser systems.