{'fraction_non_alphanumeric': 0.012345679012345678, 'fraction_numerical': 0.002244668911335578, 'mean_word_length': 4.406060606060606, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Revision of a BFX total hip replacement stem using a Kyon stem and a head adaptor in two dogs.

OBJECTIVE
To report revision of BFX cementless press-fit stem loosening with a Kyon cementless stem and a head adaptor in two dogs.


METHODS
Total hip arthroplasty stem revision was performed in two dogs with loosening of a previously implanted Biomedtrix press-fit BFX stem. Both dogs had a well-integrated BFX cup and single stage revision was performed using a standard Kyon stem and a head adaptor in order to couple with a 17 mm head and maintain the BFX cup.


RESULTS
Revisions resulted in a stable functional prosthesis with successful bone integration at the one-year postoperative re-evaluation.


CONCLUSIONS
Use of a Kyon stem with a head adaptor may be a successful option to provide immediate stem stability for revision of a loosened BFX stem with a properly osseointegrated cup.