{'fraction_non_alphanumeric': 0.014094432699083862, 'fraction_numerical': 0.028893587033121917, 'mean_word_length': 5.173913043478261, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Deformation Monitoring of the Submillimetric UPV Calibration Baseline

Abstract A 330 m calibration baseline was established at the Universitat Politècnica de València (UPV) in 2007. Absolute scale was subsequently transferred in 2012 from the Nummela Standard Baseline in Finland and distances between pillars were determined with uncertainties ranging from 0.1 mm to 0.3 mm. In order to assess the long-term stability of the baseline three field campaigns were carried out from 2013 to 2015 in a co-operative effort with the Universidad Complutense de Madrid (UCM), which provided the only Mekometer ME5000 distance meter available in Spain. Since the application of the ISO17123-4 full procedure did not suffice to come to a definite conclusion about possible displacements of the pillars, we opted for the traditional geodetic network approach. This approach had to be adapted to the case at hand in order to deal with problems such as the geometric weakness inherent to calibration baselines and scale uncertainty derived from both the use of different instruments and the high correlation between the meteorological correction and scale determination. Additionally, the so-called the maximum number of stable points method was also tested. In this contribution it is described the process followed to assess the stability of the UPV submillimetric calibration baseline during the period of time from 2012 to 2015.