{'fraction_non_alphanumeric': 0.007858546168958742, 'fraction_numerical': 0.0026195153896529143, 'mean_word_length': 5.586206896551724, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Acoustic emission sensor effect and waveform evolution during fatigue crack growth in thin metallic plate

In this article, the effect of the acoustic emission sensor on the acoustic emission waveforms from fatigue crack growth in a thin aerospace specimen is presented. In situ acoustic emission fatigue experiments were performed on the test coupons made of aircraft grade aluminum plate. Commercial Mistras S9225 acoustic emission sensor and piezoelectric wafer active sensor were used to capture the acoustic emission waveforms from the fatigue crack. It has been shown that the piezoelectric wafer active sensor transducer successfully captured the fatigue crack–related acoustic emission waveforms in the thin plate. The piezoelectric wafer active sensor transducer seems to capture more frequency information of the acoustic emission waveform than the conventional acoustic emission sensor in this particular application. We have also shown the evolution of the acoustic emission waveforms as the fatigue crack grows. The signatures of the fatigue crack growth were captured by the evolution of the acoustic emission waveforms. This waveform evolution is highly related to the physical boundary conditions of the cracks as well as the fatigue crack growth mechanism. The fatigue loading and acoustic emission measurement were synchronized using the same acoustic emission instrumentation. This synchronization provided the exact load level when the acoustic emission signals had occurred during the fatigue crack growth.