{'fraction_non_alphanumeric': 0.02645198389879241, 'fraction_numerical': 0.017826336975273145, 'mean_word_length': 4.958904109589041, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Hemodynamic investigation and in vitro evaluation of a novel mixed flow blood pump.

INTRODUCTION
Ventricular assist devices (VADs) are considered as an effective treatment for patients with advanced heart failure, while complications associated with blood damage remain a burden. Structure design innovation has potential to reduce hemolysis and improve hemocompatibility.


METHODS
In this research, a novel mixed flow blood pump that integrates structural features of the axial and centrifugal VADs was proposed. The pump consists of an inducer, a mixed impeller supported by two ceramic pivot bearings and a volute. The flow field and laminar viscous shear stress were analyzed by the in-silico simulation. The hydraulic and hemolytic performance were evaluated in vitro by using a 3D printed pump.


RESULTS
The flow field distribution showed that streamlines in the connection area were smoothly transitioned through structural integration and no irregular flow occurred in the entire flow channel. The axial blades work as a fluid accelerator (generating 18.56% of the energy), and the centrifugal blades provide the main pressure head. The proportion of fluid inside the pump exposed to low laminar viscous shear stress (<50 Pa) and high laminar viscous shear stress (>150 Pa) was 99.02% and 0.03%, respectively. The in vitro hemolysis test results showed that the NIH (Normalized Index of Hemolysis) value of the mixed pump is 0.0079 ± 0.0039 g/100 L (n=6).


CONCLUSION
It can be concluded that the mixed flow structure is effective to improve hydraulic performance, eliminate flow disturbance and minimize shear stresses. This novel pump design is expected to provide a new direction for the development of next-generation VADs.