{'fraction_non_alphanumeric': 0.014023732470334413, 'fraction_numerical': 0.002157497303128371, 'mean_word_length': 5.065359477124183, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Memory and consciousness intertwingled.

and is a commonly reported problem with other supraglottic intubating devices. The tip of the TotaltrackTM blade is protected by a silicone cover to minimize epiglottic or oropharyngeal injury. The TotaltrackTM does require a minimum inter-incisor distance of at least two cm and the device may be more difficult to insert in the presence of enlarged breasts. In all our study patients, a full view of the glottis could be obtained and adequate ventilation established. Securing the airway, once one is expert, takes approximately 40 s. These findings suggest that the TotaltrackTM is a promising device for airway management and has the potential to become a preferred device in the emergency situation of a difficult direct laryngoscopy, when there is urgency to secure the airwaywith a cuffed tracheal tube. Further evaluation of its efficacy and safety is required. Acknowledgements