{'fraction_non_alphanumeric': 0.023474178403755867, 'fraction_numerical': 0.01056338028169014, 'mean_word_length': 5.413533834586466, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Simultaneous Online Monitoring of Multiple Reactions Using a Miniature Mass Spectrometer.

Advances in chemical sampling using miniature mass spectrometer technology are used to monitor slow reactions at a frequency of ca. 180 h-1 (on the Mini 12) with no sample carryover and with inline derivatization in the case of poorly ionizing compounds. Moreover, we demonstrate high reproducibility with a relative error of less than 10% for major components. Monitoring is enabled using a continuous-flow nanoelectrospray (CF-nESI) probe contained in a custom-built 3D-printed rotary holder. The holder position is automatically set using a stepper motor controlled by a microcontroller. Reaction progress of up to six reactions, including hydrazone formation and Katritzky transamination, can be monitored simultaneously without carryover for several hours.