{'fraction_non_alphanumeric': 0.040840140023337225, 'fraction_numerical': 0.014002333722287048, 'mean_word_length': 4.72, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Radiographic screen-film noise power spectrum: variation with microdensitometer slit length.

When the noise power spectrum (NPS) of a radiographic screen-film system is measured by microdensito-metrically scanning the film with a long narrow slit, sufficient slit length allows estimation of a section of the 2-D NPS from the 1-D film scans; insufficient length causes underestimation of the NPS, particularly at low frequencies ( greater, similar1 cycle/mm). Spectra of Hi-Plus, Par Speed, and Detail screens used with XRP films measured as a function of microdensitometer slit length tended to plateau at long slit lengths. The slit length was considered sufficient when NPS components at 0.4 cycle/mm were within 5% of the plateau. This occurred for slit lengths of at least 4.2, 2.6, and 2.5 mm for Hi-Plus, Par Speed, and Detail systems, respectively.