{'fraction_non_alphanumeric': 0.031995661605206074, 'fraction_numerical': 0.01898047722342733, 'mean_word_length': 4.951612903225806, 'pattern_counts': {'":': 0, '<': 1, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Accuracy assessment of consecutive test strip lots for whole blood INR point-of-care instruments: clarifying the role of frozen plasma pools

Abstract Background In the Netherlands, each new lot of test strips for the CoaguChek XS is validated by a group of collaborating centers. The purpose of this study was to assess the accuracy of the international normalized ratio (INR) measured with consecutive test strip lots and the suitability of frozen plasma pools for accuracy evaluation. Methods Each year, a particular lot of CoaguChek XS test strips is used as reference lot. The reference lots have been validated with the International Standard for thromboplastin rTF/09, yielding a mathematical relationship (R1) between reference lot INR and International Standard INR. New lots are compared to the reference lot using patients’ capillary blood samples, yielding a relationship (R2) between the new lot INR and the reference lot INR. INRs of the blood samples were within the 1.5–4.5 interval. In parallel, three frozen plasmas pools are analyzed with the test strips. The distance of each plasma point to the line of relationship R2 was assessed. Results Fifty-four test strip lots have been evaluated during 3 years (2014–2016). Mean INR differences between test strip lot and International Standard rTF/09 varied between −0.14 and +0.20 (−4% and +8%, respectively). A positive trend with strip lot sequence number was observed (p<0.001). In several cases, the distance of the frozen plasmas to the whole blood relationship (R2) was greater than the critical value for commutability. Conclusions Using whole blood, all evaluated test strip lots met the analytical bias criterion of ±10%. Frozen plasma pools behave differently compared to whole blood and are not suitable for assessing absolute accuracy of new CoaguChek XS test strips.