{'fraction_non_alphanumeric': 0.01639344262295082, 'fraction_numerical': 0.0, 'mean_word_length': 5.663636363636364, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Automated point‐of‐care testing for ABO agglutination test: proof of concept and validation

ABO‐incompatible red blood cell transfusions still represent an important hazard in transfusion medicine. Therefore, some countries have introduced a systematic bedside ABO agglutination test checking that the right blood is given to the right patient. However, this strategy requires an extremely time‐consuming learning programme and relies on a subjective interpretation of ABO test cards agglutination. We developed a prototype of a fully automated device performing the bedside agglutination test that could be completed by reading of a barcoded wristband. This POCT checks the ABO compatibility between the patient and the blood bag.