{'fraction_non_alphanumeric': 0.020179372197309416, 'fraction_numerical': 0.0, 'mean_word_length': 6.0473684210526315, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Flow-through microvial facilitating interface of capillary isoelectric focusing and electrospray ionization mass spectrometry.

A capillary electrophoresis mass spectrometry (CE-MS) interface utilizing a flow-through microvial is used to ensure the electric continuity and supply the catholyte and mobilizer solutions during the capillary isoelectric focusing (cIEF) and mobilization process. The flow-through microvial provides a stable chemical environment and helps to improve the ionization efficiency without significantly diluting the analyte. The CE-MS interface facilitates the transfer of the mobilized cIEF effluent to the site of electrospray ionization, and the gaseous ions can be detected directly by a mass spectrometer. It also allows for complete focusing and mobilization processes to be performed automatically in programmed sequences with commercial CE systems. Two different strategies, using either a part of the capillary or the flow-through microvial of the CE-MS interface as the catholyte reservoir for bare fused silica capillaries or neutral coated capillaries, respectively, were developed for automated cIEF-electrospray ionization (ESI)-MS. Reasonable separation efficiency was achieved using proper concentration of carrier ampholytes and suitable strategies of electroosmotic/electrophoretic mobilization.