{'fraction_non_alphanumeric': 0.021370670596904937, 'fraction_numerical': 0.01400147383935151, 'mean_word_length': 4.542857142857143, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
The Progress of a Microbeam Facility in the Institute of Plasma Physics

The progress of a microbeam facility in the Institute of Plasma Physics was discussed in this paper. This kind of equipment can supply single-particle beam which may be implanted into cells in micrometer-radius and measured by a new outstanding detector among global microbeam systems. Measurements by some plain targets showed that the highest current after the accelerator tube can be larger than 20 μA, the H+2 current before the second bending magnet is near 0.9 μA, the current after the second bending magnet is near 0.8 μA, and the current of the beam line (after a 2-mm diameter aperture) is near 0.25 nA which is enough for the single-particle microbeam experiment. It took scientists 3 months to do their microbeam experiment after setting up the accelerator beam line and get the microbeam from this equipment. Two pre-collimators were installed between the 2-mm diameter aperture and the collimator to survey the beam. Tracks on the CR39 film etched in the solution of NaOH showed that the beam can go through the collimator including a 10 μm diameter aperture and the 3.5 μm thick vacuum sealing film (Mylar). A new method, which is called optimization of the beam quality, was put forward in this paper, in order to get smaller diameter of beam-spot in microbeam system.