{'fraction_non_alphanumeric': 0.01452081316553727, 'fraction_numerical': 0.001936108422071636, 'mean_word_length': 5.670967741935484, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Use of an acoustic helium analyzer and microprocessor for rapid measurement of absolute lung volume during mechanical ventilation

A technique developed to measure functional residual capacity during mechanical ventilation uses an acoustic helium analyzer and microprocessor to perform open-circuit helium washout. The acoustic helium analyzer has acceptable linearity up to a helium concentration of 50% and is stable. The microprocessor combines output from the acoustic analyzer and a ventilator with an expiratory flow signal, to compute lung volume rapidly during mechanical ventilation. In a sample of ten patients with various lung problems, open-circuit helium washout gave results similar to those of a closed-circuit helium-dilution technique during assist control or intermittent mandatory ventilation modes, with or without positive end-expiratory pressure. This open-circuit system has an advantage over other techniques because it is small and portable and can measure lung volume quickly without complicated maneuvers.