{'fraction_non_alphanumeric': 0.029559118236472944, 'fraction_numerical': 0.009018036072144289, 'mean_word_length': 5.441935483870968, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Insulin Pump-Associated Adverse Events in a Brazilian Reference Center for the Treatment of Diabetes Mellitus: Proposal for a Taxonomy of Device Failures in Adults, Adolescents, and Children.

BACKGROUND
Since the introduction of continuous subcutaneous insulin infusion (CSII), the benefits have been numerous. However, adverse events (AEs) are experienced by up to 40% of users per year, exposing them to potentially fatal risks. The available evidence on the variables that trigger AEs associated with CSII remains limited, indicating the importance of studies on the subject.


AIM
To propose a taxonomy based on the prevalent AEs experienced by patients from a reference diabetes mellitus (DM) center in Brazil using different CSII devices.


METHODS
118 patients participated in an online interview and answered the questions of the data collection instrument. Identifying categories and subcategories of analysis contributed to constructing the AEs taxonomy.


RESULTS
The five analysis categories identified were: CSII User Interface (n = 45), CSII Alert System (n = 13), CSII Software and Connection (n = 11), CSII Durability (n = 30), and Electrical and Mechanical System of CSII (n = 60) A total of 159 AEs were identified, including conflicting alert messages and error/warning notification failures, errors resulting from engine malfunctions, data loss, patient interface deficiencies, button problems, and battery failure.


CONCLUSIONS
The study describes in a taxonomic format the AEs directly associated with the use of modern CSIIs that may contribute with additional information to the Food and Drug Administration (FDA) Medical Device Report (MDR) adverse event codes. In addition to guiding educational actions in the treatment of DM and providing information for health professionals and medical device developers, prospective studies examining the frequency of such problems, including the potential psychosocial impact of this technologically advanced therapy, are needed.