{'fraction_non_alphanumeric': 0.01762114537444934, 'fraction_numerical': 0.014684287812041116, 'mean_word_length': 5.314814814814815, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
New Low- and High-Temperature Powder Diffraction Facilities at Daresbury Laboratory

A closed-cycle cryogenic system has been assembled and an induction furnace has been constructed to enhance the non-ambient powder diffraction facilities at the SRS, Daresbury Laboratory. The cryostat and the furnace are designed to operate at temperature ranges of lo290 K and 290-1500 K, respectively. In this paper we describe the commissioning of the lowand high-temperature devices for high resolution powder diffractometry. Results from standard tungsten and silver samples are presented to demonstrate the operation of the apparatus and some of the early experimental results are included.