{'fraction_non_alphanumeric': 0.043010752688172046, 'fraction_numerical': 0.016666666666666666, 'mean_word_length': 4.63939393939394, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 3, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Evaluation of an improved pretreatment method for the measurement of (1-->3)-beta-D-glucan in blood samples].

Measurement of blood (1-->3)-beta-D-glucan is useful for early diagnosis and follow-up of the therapeutic process of deep seated mycoses. The Fungitec G test MK (Seikagaku Corp., Tokyo) kit using alkaline-pretreatment followed by chromogenic kinetic assay has been widely used in Japan because of its high sensitivity and easy handling of a large number of samples. Discrepancy in the levels of (1-->3)-beta-D-glucan and/or in the quantitative judgement, however, has been pointed out between this kit and other commercial kits. One of the reasons for this discrepancy has been reported to be non-specific reactions caused by substances other than beta-glucan. In this study, we have improved the alkaline pretreatment reagent by changing the concentration of KOH and salts, resulting in a marked reduction of the non-specific reaction. Recovery of standard beta-glucan added to plasma or serum after the improved pretreatment was 80 to 120%, and no amidolytic activity was detected either in plasma or in serum. By the improved pretreatment, the incidence of non-specific reactions, i.e., those that exceed the quantitation limit (3.9 pg/mL), were markedly decreased from 139 to 16 out of 200 plasma samples and from 106 to 22 out of 170 serum samples. The incidence of strong non-specific reactions, i.e., those that exceed the cut-off level (20 pg/mL), were also decreased from seven to one with plasma and seven to zero with serum samples. Correlation between corrected beta-glucan measurements by the current pretreatment and non-corrected ones by the improved pretreatment was quite good. The improved method is thus expected to decrease the frequency of non-specific false-positive reactions, with the high sensitivity of Fungitec G test MK.