{'fraction_non_alphanumeric': 0.025440313111545987, 'fraction_numerical': 0.011741682974559686, 'mean_word_length': 5.7368421052631575, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Compressive Optical Spectrometry Based on Sequency-Ordered Hadamard Transform

We propose a compressive Hadamard transform spectrometer (CHTS) based on the energy compaction trait of the sequency-ordered Hadamard matrix (SOHM), in which the spectrum is reconstructed from compressed measurements using simple matrix multiplication. The evaluation of a 128-channel CHTS on the measurements of a fluorescence spectrum demonstrates compression ratios up to 8:1, with a relative root mean square error of around 3%.