{'fraction_non_alphanumeric': 0.020434227330779056, 'fraction_numerical': 0.0, 'mean_word_length': 5.529166666666667, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Droplet-based valveless microfluidic system for phage-display screening against spheroids.

In this study, we proposed a droplet-based valveless microfluidic system that has the necessary functions to perform the binding, washing, eluting, and collecting processes of phage-display screening against spheroids, which can be expected to present a similar repertoire and number of membrane proteins as in vivo. Although spheroids have much larger sizes than single cells, spheroids are difficult to manipulate through manual operation. The proposed microfluidic system actively controls the position and velocity of droplets using a camera, three air pumps, and three liquid pumps to perform the processes for phage-display screening. The cross section of the microchannel is large in width and height for the passage of spheroids. Valves that can close such a large cross-sectional microchannel are not readily available. Thus, we proposed valveless flow control using liquid pumps. In addition, the proposed microfluidic system involves complex flow channels with airflow subchannels to perform phage-display screening. For washing, nonspecific-binding phages remaining in the flow channels must be minimized. The proposed microfluidic system can perform selective blocking and flush washing. Selective blocking can prevent the airflow channels from becoming hydrophilic with blocking liquid, and flush washing can flush phages remaining in the flow channel. We experimentally verified the functions of the developed microfluidic device based on the proposed system.