{'fraction_non_alphanumeric': 0.026905829596412557, 'fraction_numerical': 0.010890454836643177, 'mean_word_length': 4.916666666666667, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Shifted detector super short scan reconstruction for the rotate-plus-shift trajectories and its application to C-arm CT systems

Mobile and compact C-arm systems are routinely used in interventional procedures for fluoroscopic CT imaging. The mechanical requirements guarantee for a maximum of flexibility and mobility but restrict the mechanical rotation range (e.g. 165°) and the lateral size of the field of measurement (FOM), typically about 160 mm. Recently, the rotate-plus-shift trajectory for the acquisition of complete datasets from 180° minus fan-angle has been published.1, 2 Here, we combine the rotate-plus-shift trajectory with a shifted detector approach for a fully motorized C-arm system. As the isocenter in non-centric C-arms can be freely chosen, the shifted detector can be equally well absorbed with an offset of the C parallel to the transaxial detector direction. The typical rotation range of 360° used in shifted detector trajectories is replaced by a double rotate-plus-shift scan requiring a rotation range of at least 180° minus fan-angle. The trajectory increasing the diameter of the FOM by up to a factor of two is presented and the practical application of variations with an asymmetric FOM is shown. For image reconstruction we use our modified FDK algorithm that is equipped with a generalized redundancy weight. The presented trajectory can increase the applicability and flexibility of C-arm systems and has the potential to perform intra-operative large volume control or overview scans and thus reduce the patient’s risk.