{'fraction_non_alphanumeric': 0.013333333333333334, 'fraction_numerical': 0.0022222222222222222, 'mean_word_length': 5.723880597014926, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A high-content screening platform utilizing polarization anisotropy and FLIM microscopy

An automated high-content screening microscope has been developed which uses fluorescence anisotropy imaging and fluorescence lifetime microscopy to identify Förster resonant energy transfer between eGFP and mRPF1 in drug screening assays. A wide-field polarization resolved imager is used to simultaneously capture the parallel and perpendicular components of both eGFP and mRFP1 fluorescence emission to provide a high-speed measurement of acceptor depolarization. Donor excited state lifetime measurements performed using laser scanning microscopy is then used to determine the FRET efficiency in a particular assay. A proof-of-principle assay is performed using mutant Jurkat human T-cells to illustrate the process by which FRET is first identified and then quantified by our high-content screening system.