{'fraction_non_alphanumeric': 0.01971447994561523, 'fraction_numerical': 0.006118286879673691, 'mean_word_length': 5.082644628099174, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Perfusion analysis using a wide coverage flat-panel volume CT: feasibility study

We developed a Flat-panel detector based Volume CT (VCT) prototype scanner with large z-coverage. In that prototype scanner a Varian 4030CB a-Si flat-panel detector was mounted in a multi slice CT-gantry (Siemens Medical Solutions) which provides a 25 cm field of view with 18 cm z-coverage at isocenter. The large volume covered in one rotation can be used for visualization of complete organs of small animals, e.g. rabbits. By implementing a mode with continuous scanning, we are able to reconstruct the complete volume at any point in time during the propagation of a contrast bolus. Multiple volumetric reconstructions over time elucidate the first pass dynamics of a bolus of contrast resulting in 4-D angiography and potentially allowing whole organ perfusion analysis. We studied to which extent pixel based permeability and blood volume calculation with a modified Patlak approach was possible. Experimental validation was performed by imaging evolution of contrast bolus in New Zealand rabbits. Despite the short circulation time of a rabbit, the temporal resolution was sufficient to visually resolve various phases of the first pass of the contrast bolus. Perfusion imaging required substantial spatial smoothing but allowed a qualitative discrimination of different types of parenchyma in brain and liver. If a true quantitative analysis is possible, requires further studies.