{'fraction_non_alphanumeric': 0.03655107778819119, 'fraction_numerical': 0.02999062792877226, 'mean_word_length': 4.650793650793651, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Temperature influence on coulometric trace humidity measurement

Planar coulometric sensors were tested in humidified synthetic air at various gas temperatures. Generated frost point temperature in the gas ranged from −30 °C to −60 °C and were measured by coulometric sensors and in addition by a calibrated dew point hygrometer. The gas temperatures, which were measured by a calibrated Pt100 sensor, were set to −20 °C, 0 °C, 23 °C, 40 °C, 50 °C, and 60 °C during the experiments. Empiric nonlinear functions were calculated between the generated humidity and the sensor signals. In comparison to the measured signal at 23 °C, the sensor signals were lower at the other gas temperatures. The measurements at 60 °C showed indistinct results due to a great signal noise. The response behavior of the sensors was similar at 23 °C, 40 °C and 50 °C. In contrast to that, the sensors reacted slowly at a gas temperature of −20 °C and 0 °C. In summary, with coulometric sensors it was possible to measure continuously trace humidity with an expanded uncertainty below 2 K.