{'fraction_non_alphanumeric': 0.0350609756097561, 'fraction_numerical': 0.009908536585365854, 'mean_word_length': 4.516806722689076, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Detection of sulfur-free odorants in natural gas using ion mobility spectrometry.

Beside the primary motivation of the public gas suppliers for odorizing natural gas with a sulfur-free odorant, which relates to the image of the environment-friendly fuel, natural gas, competing with low-sulfur heating fuel and diesel, a question of crucial importance of how to detect such sulfur-free odorants comes up. Concerning the replacement of sulfur-containing by sulfur-free odorization, the availability of a fast and sensitive detection method that can, further, be used on-site plays a key role. The minimum concentration of the new sulfur-free odorant Gasodor S-Free (S-Free) in natural gas should be added at a level of at least 8.8 mg m(-3) to assure a significant warning smell. Therefore, a dynamic range between 0 and approx. 25 mg m(-3) must be realised in the rather complex matrix of natural gas. By means of a handheld ion mobility spectrometer, the odorant content in natural gas is determined within less than 80 s total analysis time directly at the gas pipe. The concentration of S-Free is monitored between 4 and 23 mg m(-3) respecting the quality of the natural gas (high- and low-caloric gas). Results of the validation using a gas chromatograph as a reference standard will be discussed in detail.