{'fraction_non_alphanumeric': 0.015137180700094607, 'fraction_numerical': 0.008514664143803218, 'mean_word_length': 4.845303867403315, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A Multiwire Secondary Emission Profile Monitor for Small Emittance Beams

A secondary emission monitor using two multiwire grids separated by a positively biased collector has been constructed and tested with a 1 GeV electron beam at the Orsay Linac. The monitor installed just before the electron-positron converter has 8 gold-plated-tungsten wires of 0.1 mm diameter equally spaced 0.2 mm apart in each plane. Each wire is connected with an integrator using a low-bias current operational amplifier. The wire planes and the collector are moved into the beam by a stepping motor : that allows beam-position verification. We measured narrow profiles for 1 Amp peak current pulses of 30 nanoseconds width. Profiles are displayed on a scope and allow emittance determination by the three gradient method. Such a monitor is very useful to control the electron beam position and dimensions on the converter, because the positron source dimensions are rather bigger than those of the incident beam and the geometrical acceptance of the positron Linac is limited.