{'fraction_non_alphanumeric': 0.022604422604422605, 'fraction_numerical': 0.0063882063882063885, 'mean_word_length': 5.114114114114114, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Fabrication of Fine Electrodes on the Tip of Hypodermic Needle Using Photoresist Spray Coating and Flexible Photomask for Biomedical Applications.

We have introduced a fabrication method for electrical impedance spectroscopy (EIS)-on-a-needle (EoN: EIS-on-a-needle) to locate target tissues in the body by measuring and analyzing differences in the electrical impedance between dissimilar biotissues. This paper describes the fabrication method of fine interdigitated electrodes (IDEs) at the tip of a hypodermic needle using a photoresist spray coating and flexible film photomask in the photolithography process. A polyethylene terephthalate (PET) heat shrink tube (HST) with a wall thickness of 25 µm is employed as the insulation and passivation layer. The PET HST shows a higher mechanical durability compared with poly(p-xylylene) polymers, which have been widely used as a dielectric coating material. Furthermore, the HST shows good chemical resistance to most acids and bases, which is advantageous for limiting chemical damage to the EoN. The use of the EoN is especially preferred for the characterization of chemicals/biomaterials or fabrication using acidic/basic chemicals. The fabricated gap and width of the IDEs are as small as 20 µm, and the overall width and length of the IDEs are 400 µm and 860 µm, respectively. The fabrication margin from the tip (distance between the tip of hypodermic needle and starting point of the IDEs) of the hypodermic needle is as small as 680 µm, which indicates that unnecessarily excessive invasion into biotissues can be avoided during the electrical impedance measurement. The EoN has a high potential for clinical use, such as for thyroid biopsies and anesthesia drug delivery in a spinal space. Further, even in surgery that involves the partial resection of tumors, the EoN can be employed to preserve as much normal tissue as possible by detecting the surgical margin (normal tissue that is removed with the surgical excision of a tumor) between the normal and lesion tissues.