{'fraction_non_alphanumeric': 0.025537634408602152, 'fraction_numerical': 0.012096774193548387, 'mean_word_length': 5.127572016460905, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Sense and Nonsense of Elevated Column Temperature in Proteomic Bottom-up LC-MS Analyses.

Elevated column temperature represents a simple means for improving chromatographic separation of peptides. Here, we demonstrated the advantages of the column temperature in peptide separation using state-of-the-art columns. More importantly, we also determined how temperature can impair proteomic bottom-up analyses. We found that an elevated temperature in combination with the acidic pH of the mobile phase induced in-column peptide hydrolysis with high specificity to Asp and accelerated five modification reactions of amino acids. The positive effects of temperature dominated in the 30 min long gradients since the column operated at 90 °C provided the largest number of identified peptides and proteins. However, the adverse effects of temperature on peptide integrity in longer liquid chromatography-mass spectrometry (LC-MS) analyses required its reduction to obtain optimum results. The largest number of peptides was identified using the column maintained at 75 °C in 60 min long gradients, at 60 °C in 120 min long gradients, and at 45 °C in 240 min long gradients. Our results indicate that no universal column temperature exists for bottom-up LC-MS analyses. Quite the contrary, the temperature setting must be selected rationally to exploit the full capabilities of the state-of-the-art mass spectrometers in proteomic LC-MS analyses, with the gradient time being a critical factor.