{'fraction_non_alphanumeric': 0.025559105431309903, 'fraction_numerical': 0.0, 'mean_word_length': 5.064516129032258, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Novel dual-probes atomic force microscope for line width measurements

Dual-probe Atomic Force Microscope (AFM) can effectively eliminate the influence of the probe size on measurement of the line width, and realize true three-dimensional measurement. Novel dual-probe AFM consists of probe system, scanning system, alignment system and displacement measurement system. As displacement measurement system, the interferometers are added to the novel dual-probes AFM. In order to simplify the dual-probe AFM structure, self-sensing tuning fork probe is used. Measurement method has two steps: the first step is to align two probes and obtain the reference point; the second step is to scan two sides of measured line by two probes separately, and calculate the line width value according to the reference point. In the alignment of two probes, the alignment method is improved by using the edge alignment and the feedback scanning alignment.