{'fraction_non_alphanumeric': 0.02702702702702703, 'fraction_numerical': 0.006974716652136007, 'mean_word_length': 5.13903743315508, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Residual field suppression for magnetocardiography measurement inside a thin magnetically shielded room using bi-planar coil

For a magnetocardiography (MCG) system inside a magnetically shielded room (MSR), the residual field should be further suppressed to obtain high-quality MCG signals. In this paper, a compensation system has been developed based on the bi-planar coil and the proportional-integral-derivative (PID) controller. The bi-planar coil, derived from the target-field theory and the Tikhonov regularization method, is utilized to generate magnetic field with high uniformity in the pre-defined target region. And the PID controller is introduced to provide dynamic compensation current for the coil, according to the residual field change monitored by a reference SQUID magnetometer. The measurement results show that the noise suppression ratio (NSR) can reach above 20 dB in the low-frequency range from 0.1 Hz to 50 Hz. The DC component and the fluctuation of the residual field in time-domain can be respectively suppressed to 0 pT and 4 pT, indicating that this proposed compensation method is useful for the MCG measurement.