{'fraction_non_alphanumeric': 0.01805869074492099, 'fraction_numerical': 0.011286681715575621, 'mean_word_length': 5.434782608695652, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Downtime experience for clinical linear accelerators.

The downtime for four linear accelerators in clinical radiation therapy has been studied for about four years. The time lost during patient treatment hours is on the average 3% with an in‐house technical service group, which invests approximately 1000 man‐hours per machine and year in preventive maintenance and repair. The time lost decreases significantly with time after installation.