{'fraction_non_alphanumeric': 0.04294478527607362, 'fraction_numerical': 0.019427402862985686, 'mean_word_length': 5.398692810457517, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Nellix endovascular aneurysm-sealing system: a single-center experience and review of current evidence.

Aims: We report the 6-year results of our use of the Nellix® endovascular aneurysm sealing system. Materials & methods: This is a retrospective review of patients we treated from 2013 to 2019. The outcomes described include technical success rate, mortality, all procedure-related complications, reinterventions, open surgical conversion and secondary sac rupture. Results & conclusions: A total of 68 patients were treated. Mortality was 2.9%. Secondary interventions were carried out for distal embolization (3), stent occlusion (1), pseudoaneurysm (1) and endoleak (5). Open surgical conversion was required in 6.4% of cases. Medium- and long-term complications of EVAS occurred more frequently than expected. Because the durability of endovascular aneurysm sealing is questionable, strict postoperative surveillance of Nellix is crucial to identify features of failure.