{'fraction_non_alphanumeric': 0.037648612945838836, 'fraction_numerical': 0.02774108322324967, 'mean_word_length': 4.674157303370786, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Quantitative TLC-Image Analysis of Urinary Creatinine Using Iodine Staining and RGB Values.

Digital image analysis of the separation results of colorless analytes on thin-layer chromatography (TLC) plates usually involves using specially tailored software to analyze the images generated from either a UV scanner or UV lamp station with a digital camera or a densitometer. Here, a low-cost alternative setup for quantitative TLC-digital image analysis is demonstrated using a universal staining reagent (iodine vapor), an office scanner and a commonly available software (Microsoft Paint) for analysis of red, green and blue colors (RGB values). Urinary creatinine is used as a model analyte to represent a sample in complicated biological matrices. Separation was carried out on a silica gel plate using a butanol-NH4OH-H2O (40 : 10 : 50, v/v) mobile phase with a 6-cm solvent front. It is important that the TLC plate be stained evenly and with sufficient staining time. Staining the TLC plate in a 23.4 × 18.8 × 6.8 cm chamber containing about 70 g iodine crystals yielded comparable results for the staining times of 30-60 min. The Green value offered the best results in the linear working range (0.0810-0.9260 mg/mL) and precision (2.03% RSD, n = 10). The detection limit was found to be 0.24 µg per 3 µL spot. Urinary creatinine concentrations determined by TLC-digital image analysis using the green value calibration graph agree well with results obtained from high-pressure liquid chromatography (HPLC).