{'fraction_non_alphanumeric': 0.02654232424677188, 'fraction_numerical': 0.021520803443328552, 'mean_word_length': 4.911016949152542, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Evaluation of the BD GeneOhm Assay Using the Rotor-Gene 6000 Platform for Rapid Detection of Methicillin-Resistant Staphylococcus aureus from Pooled Screening Swabs

ABSTRACT As health services move toward universal methicillin-resistant Staphylococcus aureus (MRSA) screening for hospital admissions, the most cost-effective approach is yet to be defined. In this study, one of the largest to date, we evaluated the performance of the BD GeneOhm MRSA assay on the Rotor-Gene 6000 thermal cycler, using samples taken directly from pooled MRSA screens. Results were compared with the same assay performed on the Smart-Cycler II platform and overnight broth culture. Samples yielding discrepant results were subjected to detailed analysis with an in-house PCR and patient note review. A total of 1,428 pooled MRSA screens were tested. Sensitivities and specificities of 85.3% and 95.8% for the Rotor-Gene and 81% and 95.7% for the Smart-Cycler were obtained, compared with broth enrichment. The sensitivity of the BD GeneOhm assay was increased to 100% when the results of in-house PCR and patient note review were taken into account. This study demonstrates that the Rotor-Gene 6000 thermal cycler is a reliable platform for use with the BD GeneOhm assay. It also proves that commercial PCR can be performed direct on pooled samples in selective broth, without the need for overnight incubation.