{'fraction_non_alphanumeric': 0.024163568773234202, 'fraction_numerical': 0.011771995043370507, 'mean_word_length': 4.646853146853147, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
High pressure XANES and XMCD in the tender X-ray energy range

ABSTRACT We have developed an experimental setup at the ESRF beamline ID12 dedicated to X-ray absorption and magnetic circular dichroism measurements at high pressure adapted for the tender X-ray energy range and compatible with low temperatures and with high magnetic field. The focused incoming X-ray beam passes through a thin diamond disk attached to a fully perforated diamond anvil and X-ray fluorescence photons from the sample are collected in back-scattering geometry through the same diamond disk. The pressure in the cell is measured using the ruby luminescence through a full diamond anvil. The highest pressure attainable with this diamond anvil cell (DAC) depends on the thickness of the diamond disk and it is above 16 GPa for a 80-μm thick plate and exceeds 4.5 GPa in the case of 30-μm diamond disk. Excellent performances of this setup in the tender X-ray range are illustrated with X-ray absorption near-edge structure studies of the phase transitions in KCl at the potassium and chlorine K-edges (3.61 and 2.82 keV, respectively) as well as in CdS at the sulfur K-edge (2.47 keV). This DAC together with a dedicated total fluorescence yield (TFY) detector could be mounted in the main heat exchanger of a cryostat and inserted in a bore of a superconducting magnet, this makes possible to perform X-ray magnetic circular dichroism (XMCD) experiments at low temperature, high magnetic field and high pressure. Feasibility of this approach is shown with the XMCD results obtained at the U M-edges in ferromagnetic superconductor UGe.