{'fraction_non_alphanumeric': 0.025241276911655532, 'fraction_numerical': 0.021529324424647365, 'mean_word_length': 5.127272727272727, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Second COS FUV Lifetime Position : Verification of FUV Bright Object Aperture ( BOA ) Operations ( FCAL 4 )

As part of the calibration of the second lifetime position on the Cosmic Origins Spectrograph (COS) far-ultraviolet (FUV) detectors, observations of the external target, G191-B2B, were obtained with the G130M, G160M, and G140L gratings in combination with the Bright Object Aperture. The observations were designed to verify the performance of these spectroscopic modes by reproducing similar observations taken during the SM4 Servicing Mission Observatory Verification (SMOV) of COS. These observations allowed for a detailed determination of the spatial location and profile of the spectra from the three gratings, as well as a determination of the spectral resolution of the G130M grating prior to and after the lifetime move. In general, the negligible differences which exist between the two lifetime positions can be attributed to slight differences in the optical path. In particular, the spectral resolution appears to be slightly improved. The stability of the absolute and relative flux calibration was investigated for G130M as well using STIS echelle data of G191-B2B. We determine that the COS absolute flux calibration with the BOA is accurate to ∼10%, and flux calibrated data are reproducible at the 1-2% level since SMOV.