{'fraction_non_alphanumeric': 0.0192822710230316, 'fraction_numerical': 0.012854847348687734, 'mean_word_length': 5.35374149659864, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Comparison of high definition oscillometric and Doppler ultrasound devices with invasive blood pressure in anaesthetized dogs.

OBJECTIVE
To use the American College of Veterinary Internal Medicine (ACVIM) validation criteria to evaluate the performance of high definition oscillometric (HDO) and Doppler blood pressure measurement techniques against invasive blood pressure measurements in anaesthetized dogs.


STUDY DESIGN
Prospective clinical study.


ANIMALS
Twenty client-owned dogs.


MATERIALS AND METHODS
Invasive blood pressure was measured using a catheter inserted into a pedal artery and an electronic transducer. The sites of cuff placement for the HDO measurements were the mid antebrachium or the proximal tail and, for the Doppler technique, the distal tibia. Agreement between invasive and non-invasive blood pressure measurements was estimated by the Bland-Altman method.


RESULTS
Only 10% and 34% of Doppler measurements were within 10 and 20 mmHg of invasive blood pressure values, respectively. The Doppler device failed to meet the ACVIM validation criteria for blood pressure measurement devices. The best agreement between HDO and invasive blood pressure measurement technique was observed for mean arterial blood pressure (MAP); 67% and 95% of readings were within 10 and 20 mmHg of invasive blood pressure values respectively. In addition, 52% and 87% of diastolic arterial blood pressure (DAP) measurements were within 10 and 20 mmHg of invasive readings. High definition oscillometric readings did not meet ACVIM recommended limits for SAP.


CONCLUSION AND CLINICAL RELEVANCE
The Doppler technique overestimated and the HDO device showed limited agreement with invasive blood pressure measurement in anaesthetized dogs. High definition oscillometry met most of the ACVIM requirements for MAP and DAP while the Doppler technique did not.