{'fraction_non_alphanumeric': 0.01935483870967742, 'fraction_numerical': 0.01827956989247312, 'mean_word_length': 5.248322147651007, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Digital subtraction angiography with an Isocon camera system: clinical applications.

A new imaging system for digital subtraction angiography (DSA) was evaluated in 30 clinical studies. The image receptor is a 25 X 25 cm, 12 par gadolinium oxysulfate rare-earth screen whose light output is focused to a low-light-level Isocon camera. The video signal is digitized and processed by an image-array processor containing 31 512 X 512 memories 8 bits deep. In most patients, intraarterial DSA studies were done in conjunction with conventional arteriography. In these arterial studies, images adequate to make a specific diagnosis were obtained using half the radiation dose and half the amount of contrast material needed for conventional angiography. In eight intravenous studies performed either to identify renal artery stenosis or for evaluation of congenital heart anomalies, the images were diagnostic but objectionably noisy.