{'fraction_non_alphanumeric': 0.021080368906455864, 'fraction_numerical': 0.0, 'mean_word_length': 5.229508196721311, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Prospects for Vibrational-Mode EELS with High Spatial Resolution

Abstract Taking advantage of previous measurements by Geiger and co-workers, we discuss the possibilities and problems of measuring vibrational modes of energy loss in a transmission electron microscope fitted with a monochromator and a high-resolution energy-loss spectrometer. The tail of the zero-loss peak is seen to be a major limitation, rather than its full-width at half-maximum. Because of the low oscillator strengths and small cross-sections involved, radiation damage will limit the spatial resolution if this technique is applied to organic specimens. Delocalization of the inelastic scattering may also be a limitation, if a dipole description of the scattering process is valid.