{'fraction_non_alphanumeric': 0.02223926380368098, 'fraction_numerical': 0.003834355828220859, 'mean_word_length': 5.334951456310679, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A new vestibulo-ocular reflex recording system designed for routine vestibular clinical use.

A new vestibulo-ocular reflex (VOR) recording system was developed, which consists of an infrared eye camera, a small velocity sensor and a frequency modulator. Using this system, the head velocity signal was frequency modulated and simultaneously recorded as a sound signal on the audio track of a Hi8 video recorder with eye images. This device enabled recording of the VOR response in routine vestibular clinical practice. The reliability and effectiveness of this system were estimated by recording and analysing the VOR response against manually controlled rotation in normal subjects (n = 22) and in patients with unilateral severe vestibular hypofunction (n = 11). VOR gain on clockwise rotation viewed from the top was defined as R gain, and counterclockwise rotation as L gain. Directional preponderance (DP%) was also calculated. VOR gain towards the diseased side was significantly lower than that towards the intact side, and also significantly lower than that of normal subjects. DP% of unilateral vestibular hypofunction cases was significantly larger than that of normal subjects. These findings indicate that this VOR recording system reliably detects severe unilateral vestibular hypofunction.