{'fraction_non_alphanumeric': 0.016798418972332016, 'fraction_numerical': 0.008893280632411068, 'mean_word_length': 4.82183908045977, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Gamma dose measurement in a water phantom irradiated with the BNCT facility at THOR.

It has been proposed that a LiF thermoluminescence dosemeter (TLD) is used as a gamma dosemeter in a water phantom irradiated with the BNCT facility at THOR. Based on the TLD neutron sensitivity and neutron fluxes in the water phantom, which were simulated by the MCNP code, TLD-700 was chosen as a gamma dosemeter in this report. For the correction of the neutron influence on TLD-700, the thermal neutron sensitivity to TLD-700 was investigated with MCNP simulation and the thermal neutron flux was measured with gold foils using the cadmium difference technique. The correction to the neutron influence on the TLD was established on the TLD thermal neutron sensitivity. the thermal neutron flux, and the conversion factor from energy deposition in the TLD to the TLD response. By comparing the experimental data with the thermal neutron influence correction, these data are in very good agreement with the MCNP predictions.