{'fraction_non_alphanumeric': 0.02459016393442623, 'fraction_numerical': 0.0, 'mean_word_length': 4.894886363636363, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 1, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Development of a Position Decoding ASIC for SPECT using Silicon Photomultiplier

Single Photon Emission Computed Tomography(SPECT) is a widely used diagnosis modality for detecting metabolic diseases. In general, SPECT system is consisted of a sensor, a pre-amplifier, position decoding circuits(PDC) and a data acquisition(DAQ) system. Due to such complexity, it is quite costly to assemble SPECT system by putting discrete components together. Moreover, using discrete components would make the system rather bulky. In this work, we designed a channel module ASIC for SPECT system. This system was composed of a transimpedance amplifier(TIA), comparators and digital logics. In this particular module, a TIA was selected as a preamplifier because the decay time and the rise time are shorter than that of other preamplifier topologies. In the proposed module, the amplified pulse from the TIA was split into two separate signals and each signal was then fed into two comparators with different reference levels, e.g., a low and high level. Then an XOR gate combined the comparator outputs and the output of XOR gate was sent to the suceeding digital logic. Furthermore, the output of each component in the module is composed of a signal packet. The packet includes the information on the energy, the time and the position of the incident photon. The energy and position information of a detected radiation can be derived from the output of the D-flipflop(DFF) in the module via time-over-threshold(TOT). The timing information was measured using a delayed rising edge from the low-level referenced comparator. There are several advantages in developing the channel module ASIC. First of all, the ASIC has only digital outputs and thus a correction circuit for analog signal distortion can be neglected. In addition, it is possible to cut down the system production cost because the volume of the system can be reduced due to the compactness of ASIC. The benefits of channel module is not only limited to SPECT but also beneficial to many other radiation detecting systems.