{'fraction_non_alphanumeric': 0.024173027989821884, 'fraction_numerical': 0.013994910941475827, 'mean_word_length': 5.10077519379845, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Frequency Measurement Capability of a Fiber-Based Frequency Comb at 633 nm

A fiber-based frequency comb has been developed to measure the frequency of a 633-nm iodine-stabilized laser at the National Metrology Institute of Japan (NMIJ). The measured frequency was consistent with the previous results measured with a Ti:sapphire-based comb. The NMIJ comb was shipped to Australia for the validation of the measurement capability of the National Measurement Institute, Australia (NMIA) and NMIJ combs using a common microwave reference and a common optical frequency. Consequently, the frequency consistency of the two combs was approximately 8 times10-17. Furthermore, we demonstrate that an absolute mode number of the comb can easily and clearly be determined by using the two combs.