{'fraction_non_alphanumeric': 0.04209621993127148, 'fraction_numerical': 0.023625429553264604, 'mean_word_length': 4.666666666666667, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
SU-F-T-230: A Simple Method to Assess Accuracy of Dynamic Wave Arc Irradiation Using An Electronic Portal Imaging Device and Log Files.

PURPOSE
The Dynamic Wave Arc (DWA) technique, where the multi-leaf collimator (MLC) and gantry/ring move simultaneously in a predefined non-coplanar trajectory, has been developed on the Vero4DRT. The aim of this study is to develop a simple method for quality assurance of DWA delivery using an electronic portal imaging device (EPID) measurements and log files analysis.


METHODS
The Vero4DRT has an EPID on the beam axis, the resolution of which is 0.18 mm/pixel at the isocenter plane. EPID images were acquired automatically. To verify the detection accuracy of the MLC position by EPID images, the MLC position with intentional errors was assessed. Tests were designed considering three factors: (1) accuracy of the MLC position (2) dose output consistency with variable dose rate (160-400 MU/min), gantry speed (2.4-6°/s), ring speed (0.5-2.5°/s), and (3) MLC speed (1.6-4.2 cm/s). All the patterns were delivered to the EPID and compared with those obtained with a stationary radiation beam with a 0° gantry angle. The irradiation log, including the MLC position and gantry/ring angle, were recorded simultaneously. To perform independent checks of the machine accuracy, the MLC position and gantry/ring angle position were assessed using log files.


RESULTS
0.1 mm intentional error can be detected by the EPID, which is smaller than the EPID pixel size. The dose outputs with different conditions of the dose rate and gantry/ring speed and MLC speed showed good agreement, with a root mean square (RMS) error of 0.76%. The RMS error between the detected and recorded data were 0.1 mm for the MLC position, 0.12° for the gantry angle, and 0.07° for the ring angle.


CONCLUSION
The MLC position and dose outputs in variable conditions during DWA irradiation can be easily detected using EPID measurements and log file analysis. The proposed method is useful for routine verification. This research is (partially) supported by the Practical Research for Innovative Cancer Control (15Ack0106151h0001) from Japan Agency for Medical Research and development, AMED. Authors Takashi Mizowaki and Masahiro Hiraoka have consultancy agreement with Mitsubishi Heavy Industries, Ltd., Japan.