{'fraction_non_alphanumeric': 0.03841630733045864, 'fraction_numerical': 0.023128185025480204, 'mean_word_length': 4.406779661016949, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Investigation of the use of external aluminium targets for portal imaging in a medical accelerator using Geant4 Monte Carlo simulation.

OBJECTIVE
To install a low-Z target on the wedge tray mount of a medical linear accelerator to create a new image beam and to confirm image contrast enhancement.


METHODS
Experimental low-energy photon beams were produced with the linac running in the 6 MeV electron mode and with a low-Z target installed on the wedge tray mount [denoted 6 MeV (low-Z target)]. Geant4 Monte Carlo simulation was performed to analyse the energy spectrum and image contrast of a 6 MeV (low-Z target) beam. This study modelled the 6 MeV (low-Z target) beam and the 6 MV (megavoltage) radiotherapy photon beam and verified model validity by measurement. In addition, a contrast phantom was modelled to quantitatively compare the image contrasts of the 6 MeV (low-Z target) beam and the 6 MV radiotherapy photon beam. A low-Z target was fabricated to generate low-energy photons (25-150 keV) from incident electrons, and a portal image of the Alderson RANDO phantom was acquired using a clinical linear accelerator for qualitative analysis.


RESULTS
The measured and calculated percentage depth dose of the 6 MV photon and 6 MeV (Al) beams were consistent within 1.5 and 1.6%, respectively, and calculated lateral profiles of the 6 MV photon beam and the 6 MeV (Al) beam were consistent with the measured results within 1.5 and 1.9%, respectively. Although low-energy photons (25-150 keV) of the 6 MV photon beam were only 0.3%, the Be, C, and Al low-Z targets, but not the Ti target, generated 34.4 to 38.5% low-energy photons. In 5 to 20 cm water phantoms, contrast of the 6 MeV (Al) beam was approximately 1.16 times greater than that of the 6 MV beam. The contrasts of 6 MeV (Al) and 6 MV photon beams in the 20 cm water phantom were ~34% lower than those in the 5 cm water phantom. 6 MeV (Al)/CR (computed radiography) images of the human body phantom were more vivid and detailed than 6 MV/EPID (electronic portal imaging device) and 6 MeV (Al)/EPID images.


CONCLUSION
The experimental beam with a low-Z target, which was simply installed on the wedge tray mount of the radiotherapy linear accelerator, generated significantly more low-energy photons than the 6 MV radiotherapy photon beam, and provided better quality portal images. Advances in knowledge: This study shows that, unlike the existing low-Z beam studies, a low-Z target can be installed outside the head of a linear accelerator to improve portal image quality.