{'fraction_non_alphanumeric': 0.05658263305322129, 'fraction_numerical': 0.03137254901960784, 'mean_word_length': 4.66984126984127, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Performance evaluation of automated cell counts compared with reference methods for body fluid analysis

Abstract Objectives Cellular analysis of body fluids (BFs) can assist clinicians for the diagnosis of many medical conditions. The aim of this work is the evaluation of the analytical performance of the UF-5000 body fluid mode (UF-BF) analyzer compared to the gold standard method (optical microscopy, OM) and to XN-1000 (XN-BF), another analyzer produced by the same manufacturer (Sysmex) and with a similar technology for BF analysis. Methods One hundred BF samples collected in K3EDTA tubes were analyzed by UF-BF, XN-BF and OM. The agreement was evaluated using Passing and Bablok regression and Bland–Altman plot analysis. The receiver operating characteristic (ROC) curves were selected for evaluating the diagnostic agreement between OM classification and UF-BF parameters. Results Comparison between UF-BF and OM, in all BF types, showed Passing and Bablok’s slope comprised between 0.99 (polymorphonuclear cells count, PMN-BF) and 1.39 (mononuclear cells count, MN-BF), the intercepts ranged between 26.47 (PMN-BF parameter) and 226.80 (white blood cell count). Bland–Altman bias was comprised between 7.3% (total cell count, TC-BF) and 52.9% (MN-BF). Comparison between UF-BF and XN-BF in all BF showed slopes ranged between 1.07 (TC-BF and PMN-BF) and 1.16 (MN-BF), intercepts ranged between 8.30 (PMN) and 64.78 (WBC-BF). Bland–Altman bias ranged between 5.8 (TC-BF) and 21.1% (MN-BF). The ROC curve analysis showed an area under the curve ranged between 0.9664 and 1.000. Conclusions UF-BF shows very good performance for the differential counts of cells in ascitic, pleural and synovial fluids and therefore it is useful to screen and count cells in this type of BF.