{'fraction_non_alphanumeric': 0.023109243697478993, 'fraction_numerical': 0.0021008403361344537, 'mean_word_length': 5.6465116279069765, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
E2Rebot: A robotic platform for upper limb rehabilitation in patients with neuromotor disability

The use of robotic platforms for neuro-rehabilitation may boost the neural plasticity process and improve motor recovery in patients with upper limb mobility impairment as a consequence of an acquired brain injury. A robotic platform for this aim must provide ergonomic and friendly design, human safety, intensive task-oriented therapy, and assistive forces. Its implementation is a complex process that involves new developments in the mechanical, electronics, and control fields. This article presents the end-effector rehabilitation robot, a 2-degree-of-freedom planar robotic platform for upper limb rehabilitation in patients with neuromotor disability after a stroke. We describe the ergonomic mechanical design, the system control architecture, and the rehabilitation therapies that can be performed. The impedance-based haptic controller implemented in end-effector rehabilitation robot uses the information provided by a JR3 force sensor to achieve an efficient and friendly patient–robot interaction. Two task-oriented therapy modes have been implemented based on the “assist as needed” paradigm. As a result, the amount of support provided by the robot adapts to the patient’s requirements, maintaining the therapy as intensive as possible without compromising the patient’s health and safety and promoting engagement.