{'fraction_non_alphanumeric': 0.027058823529411764, 'fraction_numerical': 0.0058823529411764705, 'mean_word_length': 4.86896551724138, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
CMR-B-Scalar Sensor Application for High Magnetic Field Measurement in Nondestructive Pulsed Magnets

In this paper, we present the investigation of the axial and radial magnetic-field distribution inside and outside of the bore of a nondestructive pulsed-field coil presently installed at the Dresden High Magnetic Field Laboratory (HLD). The array used for these magnetic-field measurements was made up of three CMR-B-scalar sensors based on nanostructured La-Sr-Mn-O films. The investigations were performed at a temperature of 270 K and at a peak field of 46 T. The experimental results fit well with calculations obtained using ANSYS code based on the finite element method (FEM). It is concluded that the CMR-B-scalar sensors can be successfully used for the investigation of the magnetic-field distribution in pulsed high magnetic field coils.