{'fraction_non_alphanumeric': 0.026792750197005517, 'fraction_numerical': 0.013396375098502758, 'mean_word_length': 4.38135593220339, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Evaluation of an ultra-high-precision x-ray optics

We are developing an ultra high precision soft X-ray telescope. The design of the telescope is a normal incident one for 13.5nm band using Mo/Si multilayers. Two ideas are introduced. One is the optical measurement system in order to monitor the precision of the optics system. The other is the adaptive optics system with a deformable mirror. Using an X-ray-optical separation filter, we can always monitor the deformation of the optics by optical light. With this information, we can control the deformable mirror to compensate the system distortion as a closed loop system. The telescope system is now integrating and checking by optical light. The shape of the primary mirror is an off-axis paraboloid with a focal length of 2m and an effective diameter of 80mm. This primary mirror was coated by Mo/Si multilayers. The reflectivity of the primary mirror at 13.5nm was ranging from 30 to 50 %. The secondary mirror is a basically flat mirror but actually an deformable mirror with 31 piezo-actuators. The detector is now a wave front sensor (shack-hartmann type). The closed loop control has been performed and factor of 2.4 improvement of the wave front shape has been performed comparing to the un-control case.