{'fraction_non_alphanumeric': 0.022478070175438597, 'fraction_numerical': 0.014802631578947368, 'mean_word_length': 4.757097791798107, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Enhancing benefits of bolus use through minimising the effect of air-gaps on dose distribution in photon beam radiotherapy

Abstract Introduction: Bolus material is frequently used on patient’s skin during radiation therapy to reduce or remove build-up effect for high-energy beams. However, the air-gaps formed between the bolus and the skin’s irregular surface reduce the accuracy of treatment planning. To achieve a good treatment outcome using bolus, experimental investigations are required to choose its thickness and to quantify the air-gap effect. Material and methods: Measurements for a 6 MV photon beam with a fixed source surface distance were carried out using the 31021 Semiflex 3D chamber into the water phantom. Firstly, the depth of maximum dose (R100) and the dose value at surface (Ds) were evaluated as a function of bolus thickness for some square fields. Secondly, to test the effect of the air-gaps ranged from 5 to 30 mm with a step of 5 mm between the bolus and the phantom surface, a water-equivalent RW3 (Goettingen White Water) slab form of 10 mm thickness was considered as a bolus. Results: We observed that the linear behaviour of R100 in terms of the bolus thickness makes the choice of this parameter more convenient depending on field size. In addition, increasing the air-gaps widens the penumbra and created electrons that have a greater probability to quit the radiation field borders before reaching the surface. The dose spread of the off-field area could have a significant influence on the patient treatment. Conclusion: Based on dose distribution comparisons between the measurements with and without air-gaps for the field size of 100 mm × 100 mm, it has been demonstrated that a maximum air-gap value lower than 5 mm would be desirable for an efficient use of the bolus technique.