{'fraction_non_alphanumeric': 0.041926345609065156, 'fraction_numerical': 0.0339943342776204, 'mean_word_length': 4.624203821656051, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A 3D-printed metal column for micro gas chromatography.

In this work, a 3D-printed metal column was developed for micro gas chromatography (GC) applications and its properties and gas separation performances were characterized. By using a Ti6Al4V grade 23 powder, a square spiral one meter-long column (3D-column) was 3D-printed on a planar substrate of 3.4 × 3.3 × 0.2 cm and then perhydropolysilazane (PHPS) was deposited as a pre-treatment agent, followed by a coating of stationary phase (OV-1) onto the inner wall of the micro-channel. The 500 μm-diameter circular channel and two 800 μm-wide ports of the 3D-column were confirmed to be uniform by 3D X-ray microscopy without any distortion. The physical and thermal properties of the 3D-column were found to be very similar to that of the standard Ti6Al4V grade 23 alloy with near zero porosity (∼0.07%). The 3D-column with pre-treatment and stationary coating demonstrated efficient separation performance of gas mixtures containing alkanes, aromatics, alcohols, and ketones compared to a bare or only pretreated 3D-column in terms of the peak shape, broadening, and resolution (R > 1) within 2-3 min. The well-matched thermal responses to the target temperatures were demonstrated at the ramping rates of 10-20 °C min-1 upto 200 °C with uniform heat distribution over the 3D-column. In addition, the column bleed profiles showed that the 3D-column with PHPS had a 71% lower baseline intensity at 350 °C than that without PHPS. The 3D-column was then employed to separate a gas mixture of twelve alkanes (C9-C18, C22, C24) without any significant column bleeding and peak tailing. Therefore, the thermal responses and stability of the 3D-column promise its applicability in high temperature GC applications.