{'fraction_non_alphanumeric': 0.020460358056265986, 'fraction_numerical': 0.023017902813299233, 'mean_word_length': 5.069767441860465, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Alpha pulse height distributions with ZnS(Ag) scintillator.

A flask coated with ZnS(Ag) scintillator is one of the most accurate detectors available for measuring 222Rn. To maintain this accuracy, the counting system consisting of a photomultiplier tube and associated electronics must be checked on a regular basis. A combination of an alpha source and a ZnS(Ag) scintillator is commonly used for these purposes. This paper compares the pulse height distributions of 4 alpha sources with the pulse height distribution from a 100 cm3 scintillation flask containing 222Rn. The source that most closely reproduced the distribution from an actual 222Rn sample in a 100 cm3 scintillation flask consisted of a sealed flask, of the same type, which contains a small piece of uranium-ore.