{'fraction_non_alphanumeric': 0.022245762711864406, 'fraction_numerical': 0.025423728813559324, 'mean_word_length': 5.217105263157895, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
On-orbit experience with the Spitzer far infrared detectors

The far-infrared detectors on the Multiband Imaging Photometer for Spitzer (MIPS) represent a significant advancement in both format and sensitivity. We describe some of the operational experience since launch in August 2003. MIPS has three infrared detector arrays, a 128x128 format Si:As impurity band conduction detector operating at 24 μm, a 32x32 format Ge:Ga array operating at 70 μm and a 2x20 format stressed Ge:Ga array operating at 160 μm. Since both germanium detectors utilize conventional bulk photoconductors, they are subject to a number of non-ideal behaviors that are inherent in these types of devices when operated in ultra-low backgrounds. The principal problems are nonlinear time response, changing responsivity in a radiation environment, and flux non-linearities. We describe observing strategies that are used on MIPS to minimize the impact of these effects.