{'fraction_non_alphanumeric': 0.028604118993135013, 'fraction_numerical': 0.028604118993135013, 'mean_word_length': 5.0344827586206895, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Precise wavelength calibration in continuous-wave cavity ringdown spectroscopy based on the HITRAN database.

We describe the wavelength calibration method of a narrowband laser diode in continuous-wave (CW) cavity ringdown spectroscopy (CRDS). The method uses known spectral lines as wavelength markers to calibrate and refine the wavelength-current relation of laser diodes, and their spectral positions are taken directly from the HITRAN 2004 database. We built a compact CW CRDS apparatus with a 1.517 microm (approximately 6594 cm-1) distributed feedback (DFB) laser diode as the light source and a 25 cm long glass ceramic as the cavity in which to demonstrate the method. A wavelength precision of approximately 0.8 x 10(-3) cm-1 was obtained by comparing the HITRAN 2004 database, which was approximately four times more precise than that of the conventional method.