{'fraction_non_alphanumeric': 0.023890784982935155, 'fraction_numerical': 0.008532423208191127, 'mean_word_length': 5.522222222222222, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
LA-13671-MS Comparisons of the DSPEC and DSPEC Plus Spectrometer Systems

A previous experimental evaluation’ of nine commercial gamma-ray spectroscopy systems demonstrated significant benefits of digital signal processing for improvements in the performance of high-resolution gamma-ray spectroscopy systems. Spectacular improvements in the energy resolution and throughput of germanium detectors were demonstrated for the DSPEC. Recently, a new, improved version of DSPEC, the DSPEC Plus, was developed. Results of the performance of the DSPEC Plus are compared to those of the DSPEC.