{'fraction_non_alphanumeric': 0.06117908787541713, 'fraction_numerical': 0.030033370411568408, 'mean_word_length': 4.232558139534884, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Neutral carrier electrode for continuous measurement of blood Ca2+ in the extracorporeal circulation.

A neutral-carrier-based Ca2+-selective electrode exposed to whole blood exhibited an EMF-stability better than 0.13 mV (approximately 1% change of the initial activity) during 3.7 to 6.6 h. It has been used successfully to monitor ionic Ca2+ in the extracorporeal circulation during administration of calcitonin to dogs. Data simulation shows that the apparent concentration of ionic Ca2+ is affected by the concentration of Na+. A variation in [Na+] by +/- 5 mmol/L from a value of 143 mmol/L used for the calibration leads to uncertainties of +/- 0.8% in the Ca2+ concentration and +/- 0.2% in the Ca2+ activity if 3 mol/L KCl is used as the reference electrode solution. When an isotonic solution was used as reference electrolyte, errors of +/- 0.8% and +/- 1.7%, respectively, were observed.