{'fraction_non_alphanumeric': 0.021766169154228857, 'fraction_numerical': 0.007462686567164179, 'mean_word_length': 4.725978647686833, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Automated and continuous redox potential measurements in soil.

Redox potential (Eh) describes the electrical state of a matrix. In soils, Eh is an important parameter controlling the persistence of many organic and inorganic compounds. A popular, but also criticized, manual measuring method makes use of a small tip of Pt placed on a copper wire that is placed in the soil; a reference electrode is placed in the same soil at a fixed distance. Fluctuations in redox potential values measured in the soil can be very large and depth-dependent. This will be overlooked when making single-point measurements. We developed the datalogger Hypnos 2.0 for continuous redox potential and temperature measurements at various depths in the soil and without disturbance of the site. Hypnos is field-deployable, relatively cheap, and runs on batteries. The datalogger can use a "sleep mode" between sampling events. In sleep mode, there is no constant voltage on the Pt wire or the reference electrode, but there is only a short pulse during sampling. We did not measure an effect of this short pulse on the measured redox potential. In sandy soils in mesocosms and in a salt marsh soil we measured changes in the Eh as large as from -400 to +100 mV within 4 d, and daily cycles of 200 mV. Both absolute redox potential values and their diurnal variations were depth-dependent. Because single redox measurements are insufficient in describing redox conditions in some soil systems, Hypnos can be a powerful tool when studying the effects of fluctuating redox conditions on metal availability and pollutant degradation.