{'fraction_non_alphanumeric': 0.022106631989596878, 'fraction_numerical': 0.007802340702210663, 'mean_word_length': 5.754385964912281, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Preprogrammed focusing of a micro-objective of an angle-measuring setup

The application of traditional methods for automatic focusing of micro-objectives in readout heads of measuring equipment used for control of metrological characteristics of precision optical angle-measuring structures (circular dials, rasters, and multibit graduated discs) poses significant difficulties. Application of the principle of preprogrammed focusing (prefocusing) of the micro-objective is proposed in such equipment. A functional scheme of the prefocusing subsystem of an AE.1686 circular measuring and diagnostic setup based on a Nano-OP65M linear positioning stage is considered. Experimental results obtained during a test operation of a setup with such a subsystem are presented.