{'fraction_non_alphanumeric': 0.0380794701986755, 'fraction_numerical': 0.041390728476821195, 'mean_word_length': 4.170940170940171, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Dosimeter for detection of hydrogen fluoride].

The passive dosimeter for hydrogen fluoride monitoring consists of a plastic box with 40 holes of 1 mm in diameter in the front side containing a Na2CO3-impregnated filter paper as the collection medium. The sampling rate of this dosimeter is independent of the exposure dose and temperature but depends on the relative humidity. At 20% relative humidity the sampling rate is 1.69 l.h-1, however at 70% the sampling rate is 0.68 l.h-1. The dependence of the sampling rate on the relative humidity phi is expressed by the equation AR = - 0.0198.phi + 2.04.