{'fraction_non_alphanumeric': 0.03768506056527591, 'fraction_numerical': 0.031628532974428, 'mean_word_length': 5.2217573221757325, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Reliability of a New Smartphone Accessory for Scoliosis Measurements


 BACKGROUND: Idiopathic scoliosis is the most common pathology of the children’s spine, identified as spine lateral curvature with Cobb angle greater than 10°. The rapid development of technology allows, even mobile devices, to perform a quick and cheap diagnosis ensuring an accuracy similar to the Bunnell scoliometer.
 OBJECTIVE: The study aimed to validate the accuracy of a newly developed accessory for measuring the angle of trunk rotation (ATR) using mobile devices.
 METHODS: ATR measurements were carried out on a group of 30 adults using 3 diagnostic tools: the Bunnell scoliometer, Scolioscreen, and mScolio designed by the authors. Scolioscreen and mScolio were used together with a smartphone and the Clinometer app. Three measurements were performed using each of the methods.
 RESULTS: The greatest data discrepancy was obtained between the Bunnell scoliometer and Scolioscreen (-3.7°÷4.4°) and was twice larger than between the Bunnell scoliometer and mScolio (-1.8°÷2.1°). The excellent agreement was obtained for the mScolio device, where the concordance correlation coefficient was 0.9381 (95% CI of 0.9076÷0.9588) and the interclass correlation coefficient was 0.9501 (95% CI of 0.9242÷0.9671).
 CONCLUSIONS: Tests showed excellent reliability and validity of ATR using the mScolio solution. Obtained results show excellent consistency between mScolio and the Brunnell scoliometer measurements.