{'fraction_non_alphanumeric': 0.027636363636363636, 'fraction_numerical': 0.0014545454545454545, 'mean_word_length': 5.088495575221239, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
MISR prelaunch instrument calibration and characterization results

Each of the nine cameras that compose the Multi-angle Imaging SpectroRadiometer (MISR) has been rigorously tested, characterized, and calibrated. Requirements on these tests include a 3% (1/spl sigma/) radiometric calibration requirement, spectral response function determination of both the in- and out-of-band regions, and distortion mapping. The latter test determines the relative look-angle to the ground corresponding to each focal plane detector element. This is established to within one-tenth of the instantaneous field-of-view. Most of the performance testing was done on the cameras as they completed assembly. This was done to take advantage of the serial delivery of the hardware, minimize the required size of the thermal-vacuum facilities, and allow testing to occur early in the schedule allocated for the hardware build. This proved to be an effective strategy, as each of the test objectives was met. Additional testing as an integrated instrument included verification of the data packetization, camera pointing, and clearances of the fields-of-view. Results of these studies have shown that the MISR cameras are of high quality and will meet the needs of the MISR science community. Highly accurate calibration data are on-hand and available for conversion of camera output to radiances.