{'fraction_non_alphanumeric': 0.03398926654740608, 'fraction_numerical': 0.02981514609421586, 'mean_word_length': 5.453846153846154, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Toward metrological traceability in the determination of prostate-specific antigen (PSA): calibrating Beckman Coulter Hybritech Access PSA assays to WHO standards compared with the traditional Hybritech standards

Abstract Background: The metrological traceability of prostate-specific antigen (PSA) assay calibration to WHO standards is desirable to potentially improve the comparability between PSA assays. A method comparison was performed between the traditionally standardized Beckman Coulter Hybritech Access PSA and free PSA (fPSA) assays and a new alternate calibration of assays aligned to the WHO standards 96/670 and 96/668, respectively. Methods: Sera from 641 men with and without prostate cancer, various control materials and mixtures of different proportions of the WHO standards were measured with both assay calibrations. Results: Excellent comparability between the corresponding assay calibrations was observed, with correlation coefficients of at least 0.996. The Passing-Bablok slopes were 0.747 for total PSA (tPSA), 0.776 for fPSA and 1.02 for the percentage ratio of fPSA to tPSA (%fPSA), while the corresponding percentages of the new WHO-aligned assay results related to the traditional assays were 76.2%, 77% and 102.2%. Receiver operating characteristics revealed no differences between the two PSA assay calibrations. Conclusions: The WHO calibration yields results approximately 25% lower for tPSA and fPSA values when compared with the conventional Hybritech calibration. Using the WHO-aligned PSA assay, a tPSA cut-off of 3 μg/L should be considered in clinical practice, while %fPSA cut-offs could be retained. Clin Chem Lab Med 2008;46:623–9.