{'fraction_non_alphanumeric': 0.03180212014134275, 'fraction_numerical': 0.019787985865724382, 'mean_word_length': 4.732793522267206, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
[Experimental studies on the role of intramedullary alignment rods in the etiology of fat embolisms in knee endoprostheses].

Fat embolism syndrome (FES) during implantation of an uncemented resurfacing type of total knee replacement is the leading cause of postoperative mortality. A possible explanation might be the insertion of a solid intramedullary alignment rod, resulting in an intrafemoral pressure increase. On 45 cadaveric human femora we measured the resulting intramedullary pressure during insertion of three different alignment rods: a solid rod and a hollow shaped guiding rod, both 10 mm in diameter, and a fluted alignment rod of 8.5 mm in diameter. The highest intramedullary pressures occurred after placement of a solid 10 mm alignment rod (3.77 +/- 0.44 bar), followed by a fluted 8.5 mm rod (2.53 +/- 0.28 bar). In both cases, the increased pressure led to an extrusion of bone marrow contents through the venous system of the proximal femur. The least increase occurred after insertion of a hollow shaped alignment rod (0.26 +/- 0.03 bar) without any fat extrusion. Our study confirms the hypothesis that insertion of solid intramedullary alignment rods during preparation of the femoral shaft results in a dangerous pressure increase, so that embolization can occur. To minimize the risk of a FES, instruments should be modified in a manner, that only hollow shaped alignment rods are used.