{'fraction_non_alphanumeric': 0.03110047846889952, 'fraction_numerical': 0.02631578947368421, 'mean_word_length': 4.506578947368421, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Medical imaging with reverse geometry x-radiography

Reverse Geometry X-radiography imaging uses a unique configuration of x-ray source, patient, and x-ray detector to produce high resolution images. With the current system, produced by Digiray, one can obtain a resolution of 16 lp/mm and a contrast sensitivity of 0.2%. Using a lead focusing grid with 100 keV x-rays at 1 mA a dose of less than 20 mR/sec was measured. Typical acquisition times are 1/16 sec for 256 line resolution to 8 sec for 2048 line resolution. An optimized multi-detector system under development should reduce the dose rate to far less than 1 mR/sec and allow the possibility for fast volume CT applications providing laminographic view of the patient. Also `mini-probe' detectors have been developed which may be used to produce in-vitro unobstructured views.