{'fraction_non_alphanumeric': 0.028118609406952964, 'fraction_numerical': 0.012269938650306749, 'mean_word_length': 4.841791044776119, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
SU-E-T-223: High-Energy Photon Standard Dosimetry Data: A Quality Assurance Tool.

PURPOSE
Describe the Radiological Physics Center's (RPC) extensive standard dosimetry data set determined from on-site audits measurements.


METHOD AND MATERIALS
Measurements were made during on-site audits to institutions participating in NCI funded cooperative clinical trials for 44 years using a 0.6cc cylindrical ionization chamber placed within the RPC's water tank. Measurements were made on Varian, Siemens, and Elekta/Philips accelerators for 11 different energies from 68 models of accelerators. We have measured percent depth dose, output factors, and off-axis factors for 123 different accelerator model/energy combinations for which we have 5 or more sets of measurements. The RPC analyzed these data and determined the 'standard data' for each model/energy combination. The RPC defines 'standard data' as the mean value of 5 or more sets of dosimetry data or agreement with published depth dose data (within 2%).


RESULTS
The analysis of these standard data indicates that for modern accelerator models, the dosimetry data for a particular model/energy are within ï,±2%. The RPC has always found accelerators of the same make/model/energy combination have the same dosimetric properties in terms of depth dose, field size dependence and off-axis factors. Because of this consistency, the RPC can assign standard data for percent depth dose, average output factors and off-axis factors for a given combination of energy and accelerator make and model.


CONCLUSIONS
The RPC standard data can be used as a redundant quality assurance tool to assist Medical Physicists to have confidence in their clinical data to within 2%. The next step is for the RPC to provide a way for institutions to submit data to the RPC to determine if their data agrees with the standard data as a redundant check. This work was supported by PHS grants CA10953 awarded by NCI, DHHS.