{'fraction_non_alphanumeric': 0.01267605633802817, 'fraction_numerical': 0.0, 'mean_word_length': 5.7075471698113205, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
A Case of Ventricular Undersensing in the DDI Mode: Cause and Correction

A case is presented that demonstrates a confusing problem of ventricular undersensing in the DDI pacing mode, Electrocardiographic monitoring of the patient after pacemaker implantation revealed intermittent ventricular channel outputs which appeared to be inappropriate. These occurred a period of time after the intrinsic R wave, equal to the programmed AV interval. This problem was caused by ventricular lead undersensing, which resulted when the patient's intrinsic rate was such that the intrinsic ventricular complex occurred during the ventricular blanking period. The problem was corrected by reprogramming the blanking period.