{'fraction_non_alphanumeric': 0.04560260586319218, 'fraction_numerical': 0.016286644951140065, 'mean_word_length': 5.492957746478873, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Wide-Range Mass Calibration for Negative-Ion DART Mass Spectrometry

A method for accurate mass calibration covering a wide m/z range in negative-ion direct analysis in real time mass spectrometry (DART-MS) is described. It is based on a commercially available tuning solution (Agilent ES tune mix) normally applied in electrospray ionization. This acetonitrile solution is composed of ammonium trifluoroacetate, betaine, 2,4,6-tris(heptafluoropropyl)-1,3,5-triazine, and various symmetrical hexakis-(fluoroalkoxy)-phosphazenes. Upon addition of 2% (v/v) trifluoroacetic acid in negative-ion DART-MS, this mixture yields negative ions covering a range of m/z 69 to m/z 2834. Validity of the calibration procedure is demonstrated on a Fourier transform ion cyclotron resonance mass spectrometer by application to polyethylene glycol diacid and [60]fullerene. Experimental conditions and a mass reference list are provided.