{'fraction_non_alphanumeric': 0.041666666666666664, 'fraction_numerical': 0.007575757575757576, 'mean_word_length': 4.5359281437125745, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Electrical interface characteristics (I-V), optical time of flight measurements, and the x-ray (20 keV) signal response of amorphous-selenium/crystalline-silicon heterojunction structures

We have investigated the dark current, optical TOF (time of flight) properties, and the X-ray response of amorphousselenium (a-Se)/crystalline-silicon (c-Si) heterostructures for application in digital radiography. The structures have been studied to determine if an x-ray generated electron signal, created in an a-Se layer, could be directly transferred to a c-Si based readout device such as a back-thinned CCD (charge coupled device). A simple first order band-theory of the structure indicates that x-ray generated electrons should transfer from the a-Se to the c-Si, while hole transfer from p-doped c-Si to the a-Se should be blocked, permitting a low dark signal as required. The structures we have tested have a thin metal bias electrode on the x-ray facing side of the a-Se which is deposited on the c-Si substrate. The heterostructures made with pure a-Se deposited on epitaxial p-doped (5×10 14 cm-3) c-Si exhibited very low dark current of 15 pA cm-2 at a negative bias field of 10 V μm-1 applied to the a-Se. The optical TOF (time of flight) measurements show that the applied bias drops almost entirely across the a-Se layer and that the a-Se hole and electron mobilities are within the range of commonly accepted values. The x-ray signal measurements demonstrate the structure has the expected x-ray quantum efficiency. We have made a back-thinned CCD coated with a-Se and although most areas of the device show a poor x-ray response, it does contain small regions which do work properly with the expected x-ray sensitivity. Improved understanding of the a-Se/c-Si interface and preparation methods should lead to properly functioning devices.