{'fraction_non_alphanumeric': 0.03827227993439038, 'fraction_numerical': 0.022963367960634227, 'mean_word_length': 4.683229813664596, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 1, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Effect of connector design on fracture resistance of zirconia all-ceramic fixed partial dentures.

The purpose of the present study was to determine the relationship between cross-sectional design and fracture load using a static load bearing test in yttria-stabilized tetragonal zirconia polycrystal ceramic frameworks on a molar fixed partial denture. The test framework was designed as a 3-unit bridge with two abutment teeth at the second premolar and second molar of the mandible. The cross-sectional area of the connector was 9.0, 7.0, or 5.0mm(2). In terms of shape, the cross-section was either circular or oval, with a height/width ratio of 1:1, 3:4, or 2:3. For each of the 9 combinations of cross-sectional area and shape, 5 frameworks were prepared (45 in total). Frameworks were cemented to a metallic test model with adhesive resin cement. After fracture load was measured, the percentage of fracture sites was determined and the fracture surfaces observed. In terms of cross-sectional area, there was a statistically significant difference in fracture load between 9.0, 7.0, and 5.0mm(2). No significant difference in fracture load was observed between any two shapes of connector (p>0.05). The fracture load of all frameworks with a cross-sectional area of 9.0 or 7.0mm(2) was over 880 N, which was recognized as parafunctional occlusal force. Fracture occurred at the distal connector in 82.2% of all frameworks on average. Fracture load decreased as cross-sectional area of the connector became smaller. The cross-sectional shape used in the present study was less influential on fracture load. It appears to be clinical possible to apply a connector with a cross-sectional area of 7.0mm(2). Fracture often occurred at the distal connector between the pontic and the abutment, corresponding to the second molar.