{'fraction_non_alphanumeric': 0.024822695035460994, 'fraction_numerical': 0.01702127659574468, 'mean_word_length': 4.34469696969697, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Extension of CadPlan algorithm to model the dose distribution under a motorized wedge.

The CadPlan treatment planning system models the dose distribution in the non-wedge direction under a wedged field by converting the wedge thickness to an equivalent water thickness. The algorithm estimates the off-axis ratio (OAR) in the non-wedged direction using the open field OAR at a depth deeper by this equivalent water thickness. This model has been shown to work well for a Siemens Mevatron KD-2 Linac. However, the motorized wedge of the Elekta (formerly Philips) accelerators is tapered off-axis to give a flat dose profile in the non-wedged direction. The CadPlan model assumes that the wedge has a uniform thickness in the non-wedged direction and so cannot model the off-axis dose for the motorized wedge. For a 4 MV beam of a SL75/5 accelerator this leads to a 7% overestimate and a 9% underestimate of the OAR under the thin and thick edge of the wedge respectively. For 6 and 18 MV beams of a SL20 accelerator and a 6 MV beam of a SL75/5 accelerator, the model underestimates the OAR in the order of 10% under the thick end of the wedge. We have shown that by appropriate modification of the effective water thickness values at off-axis distances, the algorithm models the OAR in the non-wedged direction to within 2.5% of the measured values for the 4, 6 and 18 MV beams, for the Elekta motorized wedge.