{'fraction_non_alphanumeric': 0.04093816631130064, 'fraction_numerical': 0.019189765458422176, 'mean_word_length': 4.909319899244332, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
Does the Robotic Arm and Preoperative CT Planning Help with 3D Intraoperative Total Knee Arthroplasty Planning?

Although several studies highlight the advantages of robotic arm-assisted total knee arthroplasty (RA-TKA), few investigate its intraoperative outcome. Therefore, the purpose of this study was to analyze the RA-TKA's ability to assist with intraoperative correction of: (1) flexion and (2) extension gaps, as well as its ability to (3) accurately predict implant sizes. Additionally, in this RA-TKA cohort, length of stay, complications, and readmissions were assessed. A total of 335 patients who underwent RA-TKA were included. The robotic software virtually measured the intraoperative prebone cut extension and flexion gaps. Differences in medial versus lateral prebone cut extension and flexion gaps were calculated. A total of 155 patients (46%) had an extension gap difference of between -2 and 2 mm (mean, -0.3 mm), while 119 patients (36%) had a flexion gap difference of between -2 and 2 mm (mean, -0.6 mm). Postbone cut differences in medial versus lateral flexion and extension gaps were measured. Balanced knees were considered to have a medial and lateral flexion gap difference within 2 mm. The robot-predicted implant size was also compared with the final implant size. Additionally, lengths of stay, complications, and readmissions were assessed. All patients achieved a postbone cut extension gap difference between -1 and 1 mm (mean, -0.1 mm). A total of 332 patients (99%) achieved a postbone cut flexion gap difference of between -2 and 2 mm (mean, 0 mm). For 98% of prostheses, the robotic software predicted within 1 implant size the actual tibial or femoral implant size used.The mean length of stay was found to be 2 days. No patients suffered from superficial skin infection, pin site infections or fractures, soft tissue damage, and no robotic cases were converted to manual TKA due to intraoperative complications. A total of 8 patients (2.2%) were readmitted; however, none were directly related to robotic use. The robotic software and use of a preoperative computed tomography (CT) substantially helped with intraoperative planning and accurate prediction of implant sizes. Therefore, based on the results of this study, the RA-TKA device does, in fact, provide considerable intraoperative assistance.