{'fraction_non_alphanumeric': 0.03239202657807309, 'fraction_numerical': 0.023255813953488372, 'mean_word_length': 4.527522935779817, 'pattern_counts': {'":': 0, '<': 0, '<?xml version=': 0, '>': 0, 'https://': 0, 'lorem ipsum': 0, 'www.': 0, 'xml': 0}, 'pii_count': 0, 'substrings_counts': 0, 'word_list_counts': {'cursed_substrings.json': 0, 'profanity_word_list.json': 0, 'sexual_word_list.json': 0, 'zh_pornsignals.json': 0}}
TES Bolometers With High-Frequency Readout Circuit

In order to improve the frequency-division multiplexing (FDM) in transition edge sensor (TES) imaging arrays, it is suggested to replace commonly used SQUID amplifiers with a semiconductor high-frequency cooled amplifier. This would result in a single 10-GHz bandwidth amplifier serving the array of more than 1000 detectors. The basic idea is to implement an antenna-coupled TES as a load for a high-Q resonator, weakly coupled to a microwave transmission line. This high-frequency scheme substitutes the traditional wire connections to the TES. The NEP as low as 2×10-19 W/Hz0.5 is estimated at ambient temperature of 300 mK for submicron-size TES absorber made of Ti; the NEP is limited by 3 K noise temperature of the amplifier. To verify the new concept, prototype TES devices made of Nb are developed and tested above 4 K. The NEP of about 1.5×10-15 W/Hz0.5 is estimated for the experimental micron-size prototype devices made of Nb at 4.5 K. The IV -curves of the TES at different temperatures are recovered using the RF and heat balance models along with the experimental R(T) data; presence of the negative electrothermal feedback is verified.