1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Cell Electrodes  Passivated 316 stainless steel  Cell Temperature  5°C  Electronics Type  Microprocessor controlled digital  signal processing autoranging  Flow Cell Max.  Pressure  1500 psi  Flow Cell Volume  0.7 µL  Electronic Noise  <50 Pc.  Linearity  1%  Output Range  0 to 15000 µS/cm  AUTOSAMPLER  Injection Volume  Range  0.1 mL to 5.0 mL in 0.1 mL  increments  Sample Capacity  50 x 5 mL vials  Vials  PolyVial and 20 µm, Filter caps 250  pcs  CONSUMABLES  Column and  precolumn  Compliance monitoring of common  anions in drinking water as specified  in U.S. EPA Method 300.0 (A)  Suppressor  electrolytically regenerated  suppression  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 3 of 76 