to overcome the resistance encountered  during mixing of dense materials shall  be 400 watts or higher.  1.4.3 Amplitude-intensity value of ultrasonic  device shall be changeable between  Minimum rate: %20 or lower  Maximum rate: %100  Steps: %2  1.4.4 The device shall include a sonotrod- titanium probe with a diameter of  between 20-25 mm.  1.4.5 Sample capacity shall be;  Minimum sample capacity: 100ml or  lower  Maximum sample capacity: 3000ml- 4000ml  1.4.6 The probes provided with the device  shall be able to withstand sterilization in  an autoclave.  1.4.7 The ultrasonic device shall be capable of  integrating the flow cell. 1.4.8 The device shall have control panel with  display to show and control amplitude,  pulse level, time, energy input and  sample  temperature  values.  Control  panel shall support remote control via  LAN and/or RS232.  1.4.9 The device shall include three 1000 ml  double-walled glass beakers.  1.4.10 The device shall be delivered with a  sound isolating enclosure designed for  said device. The device shall be used  inside of the sound isolating enclosure,  August 2020 c4f_annexiitechspeciiitechoffer_en.doc  Page 20 of 49 