1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes The vacuum system  Turbomolecular pump to achieve ultra vacuum  and external vacuum pump for establishing and maintaining the desired vacuum  EI sensitivity MRM  100 fg OFN S / N> 30 000:1, or higher for a transition 272/222  EI MRM detection  limit of the  instrument  OFN 2 fg m/z 272/222 or better IDL≤0.5 fg  OFN  Scan speed  20,000 u / sec  Resolution  adjustable  Collision energy  0-60eV  Electronic energy  10 to 200eV  Scan speed in MRM  mode  800 MRM transitions / sec;  Minimum time in  the MRM mode  (dwell time)  <0.5 ms  Ion source  2 filament  Ion source  temperature  up to 350 ᵒC  Transfer line  temperature  (interface)  up to 350 ᵒC  Page 13 of 76  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx