1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Ø Trainer should be expert who has long time experience in  working /installing/ calibrating/dealing with problems /  with the offered equipment. CV of trainer to be attached  to the bid.  Ø Training shall be conducted in Montenegrin language. In  case the trainer doesn’t speak Montenegrin language the  cost of the translator will be on account of the contractor. Ø All trainer’s costs associated with the training will be on  the contractor account (international and national  transport, accommodation and allowance, communication  cost, etc.)  LOT 3  GROUNDWATER MONITORING EQUIPMENT 3.1  MICRO PHOTOMETER WITH  BLUETOOTH, 3  PIECES • Electronic water tester with LCD display.  To enable easy testing and direct reading of minimum 20 chemical  parameters, which among others must include the following: Nitrate,  Phosphate, Chloride, total Iron, Manganese, Copper, pH, total  Alkalinity, total Hardness, free Chlorine. • Required precision 0.01 ppm (mg/L). Minimum accuracy: in term of  Nitrate (as NO$_{3}$) 0.15 - 30.0 ppm, Phosphate: 0.2 - 3.0 ppm, total Iron  0.03 - 6.0 ppm. • Self-calibrates and automatically selects the optimal wavelength.  Automatic timer and memory for minimum 1000 tests.  • Instruction manual and reagents for minimum 500 tests.  Dimensions approx. 30 x 10 x 20 cm.  Bluetooth  connection, works with iPhone and Android mobile phone  apps  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 8 of 76 