1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Type  Single Pump: Analytical - Gradient  with Degas  Flow Accuracy  <0.1%  Flow Precision  <0.1%  Pressure Range  0 to 5000 psi  Pressure Ripple  <1%  Gradient  Proportioning  Accuracy  ±0.5% (at 2 mL/min)  DETECTOR/CHROMATOGRAP HY COMPARTMENT  Dual Temperature Zones,  Two Injection Valves  Temperature  Accuracy  ±0.5°C  Temperature Range  Upper Section Range: 18 – 40 °C  Lower Section Range: 10 – 70 °C  Conductivity Detector  Cell Temperature  Stability  <0.001°C  Cell Body  Chemically inert polymeric material  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 2 of 76 