2.5  2.5.5  Machine shall be able to print balancing results.  Machine shall have a printer to print balancing results. 3.1  3.1.2  3 touch probes shall be provided with below specifications:  1x 3mm diameter ruby,  1x 6mm diameter ruby,  1x 6mm or 15mm diameter.  3 touch probes shall be provided with below specifications:  1x 3mm diameter ruby probe,  1x 6mm diameter ruby probe,  1x ruby probe (6mm or 15mm diameter)  3.1  3.1.7  Scanning accuracy of arm with provided probe shall be 0,030 mm  or more accurate.  The maximum permissible error of the arm location shall be 0,030  mm or less.  3.1  3.1.13  Reverse engineering and CAD/CAM software licenses shall be  included and will be delivered for at least 1 user.  Quality control and reverse engineering software licenses shall be  included and will be delivered for at least 1 user.  3.1  3.1.14  Device shall include a notebook workstation with 16gb ram, 512gb  ssd, 15 inch ips screen, cpu with at least 12000 passmark score,  minimum 9 Megabayt cache and minimum 6 cores with minimum  4.3ghz turbo/boosted frequency, a graphics card with 4gb ddr5  vram, 128 bit memory bus, error correcting memory, and 4000 or  higher passmark g3d mark score.  Device shall include a notebook workstation with 16GB ram,  512GB SSD hard disk, 15 inch IPS screen, CPU with at least 12000  passmark score, minimum 9 Megabayt cache and minimum 6 cores  with minimum 4.3ghz turbo/boosted frequency, a graphics card  with 4GB GDDR5 VRAM, 128 bit memory bus, error correcting  memory, and 4000 or higher passmark G3D Mark score.  3.3  3.3.4  Heave values shall be minimum,  Heave displacement: ±130 mm  Heave velocity: 130 mm/s  Heave acceleration: 0.4G  Heave values shall be minimum,  Heave displacement: ±130 mm  Heave velocity: 130 mm/s  Heave acceleration: ±0.4G  3.3  3.3.13  Maximum payload floor area shall be at least 2000x2000 mm.  Payload floor area shall be minimum 2000x2000 mm.  3.3  3.3.14  Platform shall include required motion and simulation software and  Software Development Kit.  Platform shall include required motion and simulation software and  Software  Development  Kit  for  generic  use.  Software  and  Application Programming Interface (API) shall be ready for test and  simulation applications.  LOT 3 - PROTOTYPING & VISUAL PROCESSING EQUIPMENT LOT 4 - 3D PRINTER 4.1  4.1.3  Effective Building volume shall be at minimum 130 x 130 x 175  mm.  Effective Building volume shall be at minimum:  Surface area: 105mm x 140mm or 140mm x105mm  Page  3  of  4