1.1.6 Device  shall  have  integrated  touch  screen with a minimum of 8 inches size,  minimum 800 x 480 pixels to display  real time graphical analyses.  1.1.7 Device  shall  have  minimum  8-Bit  analog to digital Converter.  1.1.8 Device shall have a sampling rate of at  least 1.25 Gsample/s.  1.1.9 Device shall have a memory depth of at  least 20 Mpoints per channel when  operating in normal mode.  1.1.10 Device shall have waveform generator  with up to min. 25 MHz.  1.1.11 Device shall have logic analyzer.  1.1.12 4 passive probes shall be included in the  device.  1.1.13 Device shall have I2C, SPI, UART,  CAN, LIN trigger features.  1.1.14 Device shall have the ability to take a  screenshot of its screen.  1.1.15 Device shall be compliant with IEC  61010-1.  1.1.16 Device shall have min. 8-channel digital signal option supplied with the  device.  1.1.17 The oscilloscope shall have ethernet and  USB interface.  1.2 Vacuum Mixer  August 2020 c4f_annexiitechspeciiitechoffer_en.doc  Page 15 of 49 