1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Cache, 2.90 GHz), Memory: 4GB RAM, Storage: 256GB, Input: 1GB  Ethernet port, USB: 2 USB 3.0 ports, 4 USB 2.0 ports  Compensation: full matrix compensation, automatic or manual, the  software has a compensation library that stores the "spillover" values of  fluorescent colors, which makes it easy to determine the compensation  matrix with new input settings  Quality control: automatic daily quality control  Performance  Sensitivity: MESF (molecules of equivalent soluble fluorochrome):  FITC: <30, PE: <10  Fluorescence resolution: rCV <3%  Acquisition rate: 30,000 particles per second with 15 parameters LOT 8  HYDROGRAPHIC MULTI-PARAMETER PROBE  8.1  HYDROGRAPHIC MULTI-PARAMETER PROBE, 1 PIECE  CTD multiparameter probe for autonomous profiling in a water column  of basic hydrographic parameters T, conductivity and depth, with  accompanying sensors for measuring: pH, oxygen dissolution (optical  sensor), turbidity and chlorophyll. The probe (in the plastic housing) and  the sensors are housed on a protective support that protects the system  from impact. The system has the ability to save data, as well as the  option to download data in real time (online system).  CE Declaration of conformity required  SBE 25plus Sealogger CTD: SEALOGGERplus CTD – 16 Hz sampling  rate  With modular temperature (SBE 3F) and conductivity (SBE 4C) sensors  and TC Duct, submersible pump (SBE 5), internal strain gauge pressure  sensor, 2 GB memory, eight A/D input channels (0 - 5 volt input range),  two RS-232 data input channels. Includes data I/O and pump Y-cable, 20  meter data I/O cable, protective cage, Sea soft software, and complete  documentation  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 54 of 76 