1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Auto injector / Auto Sampler  Autosampler  capacity  min. 150 of the vial 2 ml / 1.5 ml;  Autoinjector for  liquid samples,  controlled by  software that is  done and control  devices  yes  The possibility of  changing the depth  of injection and  injection speed  yes  Injection volume  0.1μL - 8.0μL; 0.1μL increments with 10μL of   a micro syringe  Micro syringe of  10μL should be  included in the offer  Yes  Contamination  less than 10 $^{-4}$  Included vials  3000 units with volume capacity of 2ml or 1.5ml  Software for instrument control  Software for control of the device, acquisition data, processing of data recorded GC-MS/MS, the creation method, quantification, calibration, data archiving.  Automatic tuning and tuning option for creation of the files by the user.  Quantification of recorded samples with creation of calibration curves.  Creation of the reports according to user needs, while saving made reports and the possibility of subsequent use.  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 14 of 76 