sphere. It therefore signifies the accuracy of feature measurements.  The PFORM value is the maximum permissible error for the form of a sphere. This  is a value that defines the dispersion accuracy of the arm.  The LDIA value is the maximum permissible error for the articulation location. It  therefore represents the repeatability of the arm.  The most precise 3D scanner in the market has the following probing (measuring)  accuracy values:  EUNI:0.031 mm PSIZE:0.012mm LDIA: 0.048mm PFORM:0.025 mm  Kindly advise if these values meet the specification or specify in detail for each  measurement type defined by the ISO 10360-12.  Q32  Item 3.1.13  Reverse engineering and CAD/CAM software licenses shall be included and will be  delivered for at least 1 user.  Software for Quality Control and Reverse Engineering will be delivered. However,  the manufacturer of the 3D scanner claims that CAD/CAM cannot be used with 3 D  Scanner. CAD/CAM is used for applications on CNC machines.  If a CAD/CAM software is required for applications of other machines, please  provide technical specifications of the required CAD/CAM.  Please refer to Changes No:1 to the Tender Dossier, article  3.1.13.  Q33  Item 3.1.13  Reverse engineering and CAD/CAM software licenses shall be included and will be  No specific programing language is defined. For other changes in this item please refer to Changes No:1 to  Sayfa  16  /  23 CLARIFICATION No: 1 Publication Ref: EuropeAid/140639/IH/SUP/TR  Subject: Supply of Equipment for Advanced Prototype Station (APS)