1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Silver (Ag) 0.011 ± 0.005  Tin (Sn) 0.096 ± 0.039  LOT 7  FLOW CITOMETER  7.1  FLOW CITOMETER, 1 PIECE  Dimensions of the instrument (L x W x H): 42.5 x 42.5 x 34 cm  Dimensions of liquid container with holder: 14 x 35.6 x 35.6 cm  Instrument weight: 23.4 kg  Acoustic noise: <65 dBA  Voltage: 100-240 VAC, 50/60 Hz, 250 VA  Excitation optics  The optical system is free of comparison  The laser delay is automatically adjusted using the daily QC system  (quality control system) if it is needed  No user intervention is required to ensure optimum instrument effects  Emission optics  Quartz flow cell with> 1.3 NA internal change size 420 x 180 µm  Lasers  Blue laser: 488 nm wavelength, 50 mW, laser beam spot size (beam  spot): 5 x 80 μm  Upgradeable instrument with two additional lasers - red (638 nm) and  purple (405 nm) with up to 13 fluorescence detectors  Blue laser: 488 nm wavelength, 50 mW, laser beam spot size (beam  spot): 5 x 80 μm  Forward scatter detection  Si photodiode with integrated 488/8 band-pass filter  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 52 of 76 