Ref. Ares(2020)7588492 - 14/12/2020 The following alterations and/or corrections are made to the Tender Dossier:  TECHNICAL SPECIFICATIONS-$_{ }$ LOT 1 - LABORATORY EQUIPMENT Page  1  of  4 Item Line Former Text New Text 1.1  1.1.9  Device shall have a memory depth of at least 20 Mpoints per  channel when operating in normal mode.  Device shall have a memory depth of at least 20 Mpoints for two  channel when operating in normal mode. 1.2  1.2.2  Homogenizing turbine shall have adjustable speed with  Minimum rpm value: 600 rpm or lower  Maximum rpm value: 5000 rpm or higher.  Homogenizing turbine shall have adjustable speed with  Minimum rpm value: 600 rpm or lower  Maximum rpm value: 3000 rpm or higher.  1.2  1.2.3  Rotating central blades agitator rpm values shall be; Minimum rpm value: 36 rpm or lower  Maximum rpm value: 70 rpm or higher.  External agitator rpm values shall be;  Minimum rpm value: 17 rpm or lower  Maximum rpm value: 95 rpm or higher  Rotating central blades agitator rpm values shall be; Minimum rpm value: 43 rpm or lower  Maximum rpm value: 70 rpm or higher.  External agitator rpm values shall be;  Minimum rpm value: 20 rpm or lower  Maximum rpm value: 90 rpm or higher  1.2  1.2.4  Scrapers  material  shall  be  made  from  chemically  coated  polytetrafluoroetilen (PTFE).  Scrapers  material  shall  be  made  from  chemically  coated  polytetrafluoroetilen (PTFE) or Proopiomelanokortin (POMc).  1.3  1.3.5  For feed water Total Organic Carbon (TOC) between 1500- 2000ppb, product water Total Organic Carbon (TOC) shall be  maximum 5ppb (parts per billion).  For feed water Total Organic Carbon (TOC) between 1000- 2000ppb, product water Total Organic Carbon (TOC) shall be  maximum 5ppb (parts per billion).  Contract title: Supply of Equipment for Advanced Prototype Station (APS)”  Publication reference: EuropeAid/140639/IH/SUP/TR Changes No:1 to Tender Dossier 