1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes  Including  • Sonar Mounting Pole 1.85m (6’)  Short Antenna Mast 0.3m (1’)  • Long Antenna Mast 1.5m (4’11”)  LiDAR Mounting plate  • Draft Clamp  Wheeled Hard Case  LOT 15  TITANIUM - STAINLESS STEEL MULTIPARAMETER PROBE 15.1  TITANIUM - STAINLESS STEEL MULTIPARAMETER PROBE,  1 PIECE  Titanium/Stainless Steel multiparameter probe for directly sound speed,  temperature and pressure observations, and Conductivity, Salinity and  Density calculation with integral GPS module - depth rating up to 500m  rated titanium housing with  carbon composite time of flight sound velocity sensor with regards to  Permanent hydrographical changes can occur due to changes in the  thermal or salinity regimes, changes in the tidal regime, sediment and  freshwater transport, current or wave action and changes in turbidity  Sensor Specification : Turbidity  Linear Range: Nephelometer: 0 to >1 ,000 NTU - linear response1  OBS: 0 to >4,000 NTU - linear response 1 & 2 1 depending on  suspended material 2 >4,000 NTU has a non-linear monotonic response  that allows derivation of higher values using look-up tables\secondary  calibration  Linearity: 0.99 R2  Minimum Detection Level: 0.03 NTU (Nephelometer)  Conductivity  Range: 0 - 80 mS/cm  Resolution: 0.001 mS/cm  Accuracy: ±0.05 mS/cm  Temperature (Platinum Resistance Thermometer)  Range: -5°C - +35°C  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 73 of 76 