1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes 600 meter plastic housing for CTD and SBE 5P plastic pump  350 meter strain gauge pressure sensor  Wet-pluggable (MCBH) connectors on C & T sensors, pump, related  cables, and data I/O cable  Dissolved Oxygen Sensor: SBE 43 Dissolved Oxygen Sensor (Profiling  Configuration) with Wet-pluggable connector, 600 meter plastic (cable  and mount included; option 25-4 required)  pH/ORP Sensor: SBE27 pH/ORP Sensor, 1200 m depth  Fluorometer: 24245 - WetLabs ECO-FL(RT) Fluorometer for  Chlorophyll a, 600 m depth (Includes set up sensor cable p/n 801793).  Range options: 0-30 µg/l - 0-50 µg/l - 0-125 µg/l  Turbidity Meter: WET Labs ECO-NTUrt, Turbidity Meter, 125 NTU  range, 600 meter, setup cable not included (alternative sensing ranges  available: 0-125 NTU, 0-250 NTU and 0-1000 NTU --CM)  ECO & CTD integration for CTD with Wet-pluggable connectors,  includes cabling and mounting  Setup cable for WETLabs ECO sensors, MCIL-6FS to DB-9S with  red/black twisted wire leads (DN 33232)  USB to Serial Port Adapter, FTDI UC232R-10 (connects computers with  USB ports to RS-232 instruments) LOT 9  AUTONOMUS PASSIVE ACOUSTIC RECORDING SYSTEM  9.1  AUTONOMOUS PASSIVE ACOUSTIC RECORDING SYSTEM, 2  PIECES  Recording file format:  Waveform Audio File Format (.wav)  Sensitivity:  -165 to -185 dB re 1 V/μPa  Dynamic range: Minimum 16 bit  Frequency response: 10 Hz – 10 kHz flat within ±2 dB, 10 Hz –  20kHz flat within ±3 dB  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 55 of 76 