TS-JRC/PTT/2021/CNP/0763  6  F13  10 additional inputs for analog signals for temperature measurement  devices PT100 RTD.  F14  One spare Ethernet interface and one spare USB for data loggings of  external devices.  F15  Equipment software must allow use of EIS (electrochemical impedance  spectroscopy) and CV (cyclic voltammetry) devices from other  manufaturers (Solartron, Biologic, Zahner, Gamry) in addition to the  Contractors’ own brand. G1  The Equipment must be programmable with a communication interface  to allow automated testing unattended and controllable from remote via  protected internet connection. All input and output parameters must be  monitored and recorded at a sufficient sampling rate of minimum 1 Hz  with the exception of current and voltage where sampling rate shall be  minimum 100 Hz.  G2  The Equipment must be equipped with a sufficient number of appropriate  safety devices (such as sensors, interlocks, etc.) including a pushable  emergency stop button to allow automated response with predefined  actions for all cases of potential hazards. The Equipment must foresee an  interface signal (potential free contact) to be connected to the alarm &  safety system of the building.  H1  A preliminary process & instrumentation diagram (P&ID) along with a  part list for the Equipment must be provided with the offer. The parts list  should specify for the instrumentation and measurement components at  least range, accuracy and precision. For the remaining components/parts,  type and brand will suffice. 