Page 40 of 49  August 2020 c4f_annexiitechspeciiitechoffer_en.doc  1x 6mm or 15mm diameter.  3.1.3 Laser scanner shall use blue laser for  laser scanning method.  3.1.4 Laser scan speed shall be 1.2 million  points per second or higher.  3.1.5 Laser Scan point repeatability shall be  maximum 27μm.  3.1.6 Scanning accuracy of arm with provided  laser shall be 0,065 mm or more  accurate.  3.1.7 Scanning accuracy of arm with provided  probe shall be 0,030 mm or more  accurate.  3.1.8  Device shall scan 2000 points per line.  3.1.9 Device shall have at least 300 Hz line  rate.  3.1.10 Device shall scan spherical diameter of  2500mm.  3.1.11 Device shall be able to export 3D data in  OBJ, STL data format.  3.1.12 Device shall have USB interface for  data transfer.  3.1.13 Reverse engineering and CAD/CAM  software licenses shall be included and  will be delivered for at least 1 user.  3.1.14 Device  shall  include  a  notebook  workstation with 16gb ram, 512gb ssd,  15 inch ips screen, cpu with at least  12000 passmark score, minimum 9  Megabayt cache  and minimum 6 cores 