Publication Ref: EuropeAid/140639/IH/SUP/TR  Subject: Supply of Equipment for Advanced Prototype Station (APS) Q28  Item 3.1 “3D Scanner”  Quote:  “Device shall include a notebook workstation with 16gb ram, 512gb ssd, 15 inch ips  screen, cpu with at least 12000 passmark score, minimum 9 Megabayt cache and  minimum 6 cores, with minimum 4.3ghz turbo/boosted frequency, a graphics card  with 4gb ddr5 vram, 128 bit memory bus, error correcting memory, and 4000 or  higher passmark g3d mark score.”  The definition “a graphics card with 4gb ddr5 vram” is not correctly described.  GDDR5 uses for graphic card description, while DDR5 (Data Double Rate) can also  be related to non-graphical memory. Currently, the fifth version of non-graphics  memory has not been introduced to the market, and therefore the requirement for  notebook should be change as follows:  “Device shall include a notebook workstation with 16gb ram, 512gb ssd, 15 inch ips  screen, cpu with at least 12000 passmark score, minimum 9 Megabayt cache and  minimum 6 cores, with minimum 4.3ghz turbo/boosted frequency, a graphics card  with 4GB GDDR5, 128 bit memory bus, error correcting memory, and 4000 or  higher passmark g3d mark score.”  Please refer to Changes No:1 to the Tender Dossier, article  3.1.14.  Q29  3.1.7 Scanning accuracy of arm with provided probe shall be 0,030 mm or more  accurate.  By definition it is not correctly described. According to ISO10360-12 norms; the  EUNI  value is the  maximum  permissible  error  for  unidirectional  length  measurements. It therefore most closely reflects most measurement needs. The  PSIZE value is the maximum permissible error for measuring the diameter of a  sphere. It therefore signifies the accuracy of feature measurements. The PFORM  value is the maximum permissible error for the form of a sphere. This is a value that  defines the dispersion accuracy of the arm. The LDIA value is the maximum  Please refer to Changes No:1 to the Tender Dossier, article 3.1.7.  Sayfa  14  /  23 CLARIFICATION No: 1