1.  Item number  2.  Specifications required  3.  Specifications offered  4. Notes, remarks, ref to documentation  5.  Evaluation  committee’s notes Fluorescence and side scatter (light scattering under angle of 90 °)  detection  Side scatter resolution: <300 nm  4 fluorescent channels for blue (488 nm) laser  The instrument can be upgraded with a purple (405 nm) laser device, and  the detector can be adjusted to collect side scatter (small scattering  angle) to allow nanoparticles to be distinguished from noise  Flow system  Sampling Speed - Default Speed: Slow Flow: 10 μL / min, Medium  Flow: 30 μL / min and Fast Flow: 60 μL / min, customizable user flow  rate in the range 10-240 μL / min  Container capacity: 2 x 4L sheath fluid and waste  Automated Instrument Maintenance Cycle: Instrument Launch, Software  Launch, Sample Mixing, backflush, prime (Bubble Ejection), Quenching  (Daily Cleaning), Deep Cleaning  Sampling tube formats: 5 ml (12x75 mm) of polystyrene or  polypropylene, microcentrifuge tube 1.5 and 2 ml  Upgradeable instrument with plate loader supporting standard 96-well  plate, U and V bottom well plates  Electronics  Signal processing: 7-decadal data display  Digital Sampling Speed: 25 Mhz  Signal: surface pulse, height for each channel, width for one selected  channel  Data management  Data analysis and processing software in English  FCS format: FCS 3.0  Workstation / Computer Minimum Requirements: Operating System:  Windows 7 Professional 64-bit, Processor: Intel Core i3 4th Gen (3MB  August 2020 EN-c4f_annexiitechspeciiitechoffer_en.docx  EN-c4f_annexiitechspeciiitechoffer_en.docx Page 53 of 76 