(2) P ceases to be required to self-isolate in accordance with these Regulations, and regulation  2A of the Self-Isolation Regulations applies in relation to P.  6. —(1) For the purposes of regulation 6(12)(a), a day 2 test complies with this paragraph  where—  (2) A test complies with this sub-paragraph where—  (a) it is a test provided by a public provider; or  (b) it is a test provided by a private provider—  (i) in respect of—  (aa) a non-Schedule 11 passenger, on or after 1st March 2021;  (bb) a Schedule 11 passenger, on 1st or 2nd March 2021,  (ii) where the test complies with sub-paragraph (2), and  (iii) where the private provider complies with paragraph 7.  (a) it is a semi-quantitative test for the detection of coronavirus which—  (i) targets a minimum of two distinguishable SARS-CoV-2 genes other than the S gene  and performance reference controls,  (ii) includes routine in silico assurance against every variant of concern, and  (iii) produces a test solution that provides extracted nucleic acid that is suitable for whole  genome sequencing using a specified method;  (b) it is, in relation to a Schedule 11 passenger, a test that can be self-administered;  (c) the manufacturer of any device used for the purposes of the test states that the device—  (i) uses an established molecular detection method,  (ii) has a specificity and a sensitivity greater than 99% (with a 95% two-sided  confidence interval entirely above 97%),  (iii) has a limit of detection of less than or equal to 1000 SARS-CoV-2 copies per  millilitre, and  (iv) is suitable for identifying every variant of concern; and  (d) any device used for the purposes of the test—  (i) can be put into service in accordance with Part 4 of the Medical Devices Regulations  2002, other than solely by virtue of regulation 39(2) of those Regulations, and  (ii) has been validated no more than 18 months before the test is administered or  provided to P.  (a) “specified method” means a targeted sequence method specific to SARS-CoV-2 or an  equivalent—  (i) amplicon method, or  (ii) sequence bait capture method;  (b) “validated”, in relation to a device, has the meaning given by paragraph 2(2) of Schedule  10;  (c) “variant of concern” means a variant of SARS-CoV-2 identified in a designation made by  the Secretary of State for the purposes of this paragraph and published in a manner as  appears to the Secretary of State to be appropriate.  (3) For the purposes of sub-paragraph (2)—  Day 2 tests: general test requirements  61